# embeddings_queue

created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 950
encoding: FLOAT32
metadata:
{"chroma:document":"Messages Missing Wash Vial Error ID: The wash vial programmed in the method was not found. When the gripper arm picks a vial out of the sample tray, the processor monitors the gripper motor encoder. If a vial is present, the closing of the gripper fingers is limited by the vial. However, if no vial is present, the gripper fingers close too far. This is sensed by the processor (encoder position), causing the error message to be generated. Invalid Vial Position Error ID: The vial position defined","source_document_id":3,"chunk_index":172}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 951
encoding: FLOAT32
metadata:
{"chunk_index":173,"source_document_id":3,"chroma:document":"the error message to be generated. Invalid Vial Position Error ID: The vial position defined in the method or sequence does not exist. The reflection sensors on the transpor t assembly flex board are used to automatically check which sample trays are installed (coding on tray). If the vial position does not exist in the current sample tray configuration, the error message is generated. Probable cause Suggested actions 1 No wash vial in the position defined in the method. Install the wash vial"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 952
encoding: FLOAT32
metadata:
{"chroma:document":"cause Suggested actions 1 No wash vial in the position defined in the method. Install the wash vial in the correct position, or edit the method accordingly. Probable cause Suggested actions 1 Incorrect tray or trays installed. Install the correct trays, or edit the method or sequence accordingly. 2 Incorrect vial positions defined in the method or sequence. Check method settings vs. installed tray. 3 Tray recognition defective (dirty sample tray or defective transport assembly flex board).","source_document_id":3,"chunk_index":174}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 953
encoding: FLOAT32
metadata:
{"chroma:document":"tray. 3 Tray recognition defective (dirty sample tray or defective transport assembly flex board). Ensure the coding surfaces of the sample tray are clean (located at the rear of the sample tray). If this does not solve the problem, contact your Agilent service representative. Agilent InfinityLab LC Series Vialsamplers User Manual Error Information Sampler Error Messages Valve switch not found. Error ID: The injection valve has failed to switch into the position. The valve position sensor","chunk_index":175,"source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 954
encoding: FLOAT32
metadata:
{"chroma:document":"Error ID: The injection valve has failed to switch into the position. The valve position sensor monitors the position of the valve. If the valve fails to switch into the position (mainpass (main path) or bypass), or if the sensor fails to recognize the valve position, the error message is generated. Metering Initialization Failed Error ID: The metereing device has failed to do the initialization. If the piston fails to move to the home position, or if the metering device is not recognized by","source_document_id":3,"chunk_index":176}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 955
encoding: FLOAT32
metadata:
{"source_document_id":3,"chroma:document":"If the piston fails to move to the home position, or if the metering device is not recognized by the instrument. Probable cause Suggested actions 1 Defective rotor seal/stator head. Exchange the rotor seal/stator head. 2 Defective injection valve. Please contact your Agilent service representative. Probable cause Suggested actions 1 Metering device is not recognized. Configure the metering device in LabAdvisor 2 Loose belt. Please contact your Agilent service representative. 3 Rear wheel in the","chunk_index":177}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 956
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":178,"chroma:document":"in LabAdvisor 2 Loose belt. Please contact your Agilent service representative. 3 Rear wheel in the hydraulic unit is loose. Please contact your Agilent service representative. 4 Defective metering-drive motor. Please contact your Agilent service representative. 5 Defective metering device Please contact your Agilent service representative. 180 Agilent InfinityLab LC Series Vialsamplers User Manual 8 Error Information Sample Cooler Error Messages Sample Cooler Error Messages Cooler Voltage"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 957
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":179,"chroma:document":"Manual 8 Error Information Sample Cooler Error Messages Sample Cooler Error Messages Cooler Voltage Failure Error ID: The sample cooler board has detected a voltage failure. Cooler PCB Error State Error ID: 30724, The sample cooler PCB board seems to be defective. Probable cause Suggested actions 1 Blown fuses, defective cable  Check the power cable from the ALS to the sample cooler.  Please contact your Agilent service representative. 2 Voltage issue in the sample cooler Please contact your"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 958
encoding: FLOAT32
metadata:
{"chunk_index":180,"chroma:document":"your Agilent service representative. 2 Voltage issue in the sample cooler Please contact your Agilent service representative. Probable cause Suggested actions 1 Blown fuses, defective cable  Check the power cable from the ALS to the sample cooler.  Please contact your Agilent service representative. 2 Defective PCB boards Replace the sample cooler. Agilent InfinityLab LC Series Vialsamplers User Manual Error Information Sample Cooler Error Messages Cooler temperature below or above target","source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 959
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":181,"chroma:document":"User Manual Error Information Sample Cooler Error Messages Cooler temperature below or above target temperatur. Error ID: 30705, The sample cooler temperature is above or below the target temperature. Cooler Overpressure Failure Error ID: The sample cooler pressure sensor has detected an overpressure. Probable cause Suggested actions 1 Condensate inside of the chiller Check if the drainage is ok. 2 Defective cable Check the signal cable from the ALS to the sample cooler. 3 Defective temperature"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 960
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":182,"chroma:document":"2 Defective cable Check the signal cable from the ALS to the sample cooler. 3 Defective temperature sensor Replace the sample cooler. Probable cause Suggested actions 1 Overpressure in the chiller Switch off the instrument and wait for 15 min. Restart the instrument and check the signals in Lab Advisor. 2 Defective compressor Replace the sample cooler. 182 Agilent InfinityLab LC Series Vialsamplers User Manual 8 Error Information Sample Cooler Error Messages Cooler Condensate Error Error ID:"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 961
encoding: FLOAT32
metadata:
{"chroma:document":"User Manual 8 Error Information Sample Cooler Error Messages Cooler Condensate Error Error ID: 30702, 30709, The sample cooler has detected condensate inside of the leak pane. Probable cause Suggested actions 1 Blocked drainage Switch off the instrument and remove the condensate tubings. Place a beaker underneath the waste outlet. Check the waste container. There should be no kinked and sealed tubing. 2 Defective condensate sensor Switch the sample cooler off/on and check if after 8 min the","source_document_id":3,"chunk_index":183}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 962
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":184,"chroma:document":"tubing. 2 Defective condensate sensor Switch the sample cooler off/on and check if after 8 min the error shows up again. 3 Defective leak sensor Replace the sample cooler. Agilent InfinityLab LC Series Vialsamplers User Manual Error Information Integrated Column Compartment (ICC) Heater Error Messages Integrated Column Compartment (ICC) Heater Error Messages ICC Temp Wait Timeout Error Error ID: The Integrated Column Compartment (ICC) Heater has detected that the left heater control has reached"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 963
encoding: FLOAT32
metadata:
{"chunk_index":185,"source_document_id":3,"chroma:document":"Integrated Column Compartment (ICC) Heater has detected that the left heater control has reached a time limit. ICC Heater Overcurrent Failure Error ID: The Integrated Column Compartment (ICC) Heater has detected an overcurrent in the heater circuits. For safety reason heater over-current protection became active. Probable cause Suggested actions 1 Defective cable or connector Please contact your Agilent service representative. 2 Defective ICC heater Replace the ICC heater. Probable cause"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 964
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":186,"chroma:document":"your Agilent service representative. 2 Defective ICC heater Replace the ICC heater. Probable cause Suggested actions 1 Defective cable or connector Please contact your Agilent service representative. 2 Defective ICC heater Replace the ICC heater. 184 Agilent InfinityLab LC Series Vialsamplers User Manual 8 Error Information Integrated Column Compartment (ICC) Heater Error Messages ICC Heater Temp Sensor Failure Error ID: 30821, 30823,30825 The Integrated Column Compartment (ICC) Heater has"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 965
encoding: FLOAT32
metadata:
{"chunk_index":187,"chroma:document":"Temp Sensor Failure Error ID: 30821, 30823,30825 The Integrated Column Compartment (ICC) Heater has detected a reading issue with the heater sensors in the heating element. ICC Heater Over Temp Failure Error ID: The Integrated Column Compartment (ICC) Heater has detected a temperature reading above 100 °C. This exceeds the limit of the heating element. Probable cause Suggested actions 1 Defective cable or connector Please contact your Agilent service representative. 2 Defective ICC heater","source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 966
encoding: FLOAT32
metadata:
{"chunk_index":188,"source_document_id":3,"chroma:document":"cable or connector Please contact your Agilent service representative. 2 Defective ICC heater sensor Replace the ICC heater. Probable cause Suggested actions 1 Defective cable or connector Please contact your Agilent service representative. 2 Defective ICC heater Replace the ICC heater. Agilent InfinityLab LC Series Vialsamplers User Manual Agilent Technologies Maintenance and Repair Introduction to Maintenance Warnings and Cautions Overview of Maintenance Cleaning the Module Remove and Install"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 967
encoding: FLOAT32
metadata:
{"chroma:document":"to Maintenance Warnings and Cautions Overview of Maintenance Cleaning the Module Remove and Install Doors Exchange the Needle Assembly Exchange the Needle Seat Assembly Exchange the Sample Loop Assembly Exchange the Rotor Seal Exchange the Metering Seal and Piston Exchange the Gripper Arm Replace the Peristaltic Pump Cartridge Exchange the Wash Port Assembly Replace the Module Firmware This chapter describes the maintenance and the repair of the module. 186 Agilent InfinityLab LC Series","chunk_index":189,"source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 968
encoding: FLOAT32
metadata:
{"source_document_id":3,"chroma:document":"chapter describes the maintenance and the repair of the module. 186 Agilent InfinityLab LC Series Vialsamplers User Manual 9 Maintenance and Repair Introduction to Maintenance Introduction to Maintenance The module is designed for easy maintenance. Maintenance can be done from the front with module in place in the system stack. Figure 25 Overview of the Vialsampler \u00036WDWXV\u0003LQGLFDWRU\u0003\u0003\u0003\u0003 \u00031HHGOH\u0003DVVHPEO\\\u0003\u0003\u0003\u0003 \u0003'UDZHU\u0003\u0003\u0003\u0003 \u0003&ROXPQ\u0003VKHOI\u0003\u0003\u0003\u0003 \u0003/HDN\u0003GUDLQ\u0003\u0003\u0003\u0003 \u00033RZHU\u0003VZLWFK\u0003\u0003\u0003\u0003 NOTE There are no","chunk_index":190}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 969
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":191,"chroma:document":"\u0003'UDZHU\u0003\u0003\u0003\u0003 \u0003&ROXPQ\u0003VKHOI\u0003\u0003\u0003\u0003 \u0003/HDN\u0003GUDLQ\u0003\u0003\u0003\u0003 \u00033RZHU\u0003VZLWFK\u0003\u0003\u0003\u0003 NOTE There are no serviceable parts inside. Do not open the module. Agilent InfinityLab LC Series Vialsamplers User Manual Maintenance and Repair Warnings and Cautions Warnings and Cautions WARNING Personal injury or damage to the product Agilent is not responsible for any damages caused, in whole or in part, by improper use of the products, unauthorized alterations, adjustments or modifications to the products, failure to comply"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 970
encoding: FLOAT32
metadata:
{"chroma:document":"products, unauthorized alterations, adjustments or modifications to the products, failure to comply with procedures in Agilent product user guides, or use of the products in violation of applicable laws, rules or regulations. ➔ Use your Agilent products only in the manner described in the Agilent product user guides. WARNING Electrical shock Repair work at the module can lead to personal injuries, e.g. shock hazard, when the cover is opened. ➔ Do not remove the cover of the module. ➔ Only","chunk_index":192,"source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 971
encoding: FLOAT32
metadata:
{"chroma:document":"e.g. shock hazard, when the cover is opened. ➔ Do not remove the cover of the module. ➔ Only certified persons are authorized to carry out repairs inside the module. WARNING Sharp metal edges Sharp-edged parts of the equipment may cause injuries. ➔ To prevent personal injury, be careful when getting in contact with sharp metal areas. 188 Agilent InfinityLab LC Series Vialsamplers User Manual 9 Maintenance and Repair Warnings and Cautions WARNING Toxic, flammable and hazardous solvents, samples","source_document_id":3,"chunk_index":193}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 972
encoding: FLOAT32
metadata:
{"chunk_index":194,"chroma:document":"and Repair Warnings and Cautions WARNING Toxic, flammable and hazardous solvents, samples and reagents The handling of solvents, samples and reagents can hold health and safety risks. ➔ When working with these substances observe appropriate safety procedures (for example by wearing goggles, safety gloves and protective clothing) as described in the material handling and safety data sheet supplied by the vendor, and follow good laboratory practice. ➔ The volume of substances should be reduced to","source_document_id":3}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 973
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":195,"chroma:document":"by the vendor, and follow good laboratory practice. ➔ The volume of substances should be reduced to the minimum required for the analysis. ➔ Do not operate the instrument in an explosive atmosphere. CAUTION Safety standards for external equipment ➔ If you connect external equipment to the instrument, make sure that you only use accessory units tested and approved according to the safety standards appropriate for the type of external equipment. WARNING Heavy weight The module is heavy. ➔ Carry"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 974
encoding: FLOAT32
metadata:
{"source_document_id":3,"chunk_index":196,"chroma:document":"appropriate for the type of external equipment. WARNING Heavy weight The module is heavy. ➔ Carry the module at least with 2 people. ➔ Avoid back strain or injury by following all precautions for lifting heavy objects. ➔ Ensure that the load is as close to your body as possible. ➔ Ensure that you can cope with the weight of your load. Agilent InfinityLab LC Series Vialsamplers User Manual Maintenance and Repair Overview of Maintenance Overview of Maintenance It is necessary to perform periodic"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 975
encoding: FLOAT32
metadata:
{"source_document_id":3,"chroma:document":"and Repair Overview of Maintenance Overview of Maintenance It is necessary to perform periodic inspec tion of this instrument to ensure its safe use. It is possible to have these periodic inspections performed by Agilent service representatives on a contractual basis. For information regarding the maintenance inspection contract, contact your Agilent representative. The following pages describe the maintenance (simple repairs) of the module that can be carried out without opening the main","chunk_index":197}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 976
encoding: FLOAT32
metadata:
{"chunk_index":198,"source_document_id":3,"chroma:document":"the maintenance (simple repairs) of the module that can be carried out without opening the main cover. Table 17 Overview of maintenance Procedure Typical interval (minimum) Notes Change needle/needle seat 30000 needle into seat movements Change peristaltic pump cartridge 3000 min on time Change rotor seal 30000 injections 190 Agilent InfinityLab LC Series Vialsamplers User Manual 9 Maintenance and Repair Cleaning the Module Cleaning the Module To keep the module case clean, use a soft cloth"}


created_at: 2025-06-19 07:06:04
topic: persistent://default/default/3cd34793-a006-4ebf-a237-6efa1014ebb2
id: 977
encoding: FLOAT32
metadata:
{"chroma:document":"and Repair Cleaning the Module Cleaning the Module To keep the module case clean, use a soft cloth slightly dampened with water, or a solution of water and mild detergent. WARNING Liquid dripping into the electronic compartment of your module can cause shock hazard and damage the module ➔ Do not use an excessively damp cloth during cleaning. ➔ Drain all solvent lines before opening any connections in the flow path.","chunk_index":199,"source_document_id":3}


