008 GC-ECNI-LRMS 2 105 (1min), 34/min to 200 0.9 Internal, Castells et al.[3] 1000 190 (1min), 8/min to 250, 40/min to 290 (8min) 009 GC-ECNI-LRMS 1 100 (10min), 10/min to 230 NA Internal, linear 1000 HP5-MS 260 (30min) 012 GC-EI-MS/MS 2 100 (2min), 50/min to 250 1.5 External, linear 300 (2min) GC-ECNI-LRMS 1 100 (3min), 20/min to 200 1.2 External, linear 1000 DB-5MS 140, 5/min to 300 GC×GC-µECD 1 65 (2min), 20/min to 2706 F1 0.4 External, linear NA DB1-MS 140, 3/min to 270 F2 30 Innowax 3×0.25×0.1 016 GC-ECNI-LRMS 2 50 (1min), 70/min to 150 1.3 MLR ISO 12010 1000 DB-5MS 100 (1min), 60/min to 290 (4min) 017 Magnetic 2 110, 10/min to 310 120 NA, Internal, DB-57 Sector Hydrogen own standards Tomy et al. [2] 018 GC-EI-MS/MS 5 60 (1min), 20/min to 150 1.4 Internal, linear, HP-5MS 300 (10min) provided standards [a] own standards [b] 019 GC-EI-MS/MS 2 130 (4min), 40/min to 230 1.4 Internal, linear ZB-XLBHT 300 023 GC-ECNI-LRMS 2 110 (0.5min), 25/min to 150 2.5 External 1000 DB1-MS 325 (3min) R Resolving power RF Response factor NA Not available MLR Multiple linear regression 1 Approximately Length m × i.d. mm × film thickness µm 2 3 Hold time in brackets 4 carrier gas was helium and reagent gas methane, unless otherwise stated 4 When using participants’ own standards
008 GC-ECNI-LRMS 1000 2 105 (1min), 34/min to 190 (1min), 8/min to 250, 40/min to 290 (8min) 200 0.9 Internal, Castells et al.[3] 009 GC-ECNI-LRMS 1000 HP5-MS 1 100 (10min), 10/min to 260 (30min) 230 NA Internal, linear 012 GC-EI-MS/MS 2 100 (2min), 50/min to 300 (2min) 250 1.5 External, linear GC-ECNI-LRMS 1000 DB-5MS 1 100 (3min), 20/min to 140, 5/min to 300 200 1.2 External, linear GC×GC-µECD NA DB1-MS Innowax 3×0.25×0.1 1 65 (2min), 20/min to 140, 3/min to 270 2706 F1 0.4 F2 30 External, linear 016 GC-ECNI-LRMS 1000 DB-5MS 2 50 (1min), 70/min to 100 (1min), 60/min to 290 (4min) 150 1.3 MLR ISO 12010 017 Magnetic Sector DB-57 2 110, 10/min to 310 120 NA, Hydrogen Internal, own standards Tomy et al. [2] 018 GC-EI-MS/MS HP-5MS 5 60 (1min), 20/min to 300 (10min) 150 1.4 Internal, linear, provided standards [a] own standards [b] 019 GC-EI-MS/MS ZB-XLBHT 2 130 (4min), 40/min to 300 230 1.4 Internal, linear 023 GC-ECNI-LRMS 1000 DB1-MS 2 110 (0.5min), 25/min to 325 (3min) 150 2.5 External R Resolving power RF Response factor NA Not available MLR Multiple linear regression 1 Approximately 2 Length m × i.d. mm × film thickness µm 3 Hold time in brackets 4 carrier gas was helium and reagent gas methane, unless otherwise stated 4 When using participants’ own standards
1-s2.0-S0165993617303308-mmc1.docx