GET_OPR Get Operator {Oprerator Text string. See Section 3.4.1 Info Msg} ERR=0020 No operator logged on REC=nnnn Set Recipe {Recipe Auto Recipe was set Msg} Number, with Only 'nnnn' is a digit number in the range 0000 to 9999 ERR=0100 Incorrectly formatted recipe number ERR=0101 Requested recipe is not defined ERR=0202 Cannot perform action because system is not online ERR=0104 Recipe Failed to Load ERR=0105 Obsolete Recipe ERR=0206 Cannot Change Recipe while glass is present GET_RECIPE Get Recipe REC=nnnn number in the range 0000 to 9999 Number ERR=0202 Cannot perform action because system is not online ERR=0102 No receipe is loaded GET_UTILITY_DATA Get Utility Data {Utility Text string. See Section 3.4.1 Message} ERR=0202 Cannot perform action because CGM is not online ERR=0500 Unable to communicate with Sensor Suite GET_TRACE_DATA Get Trace Data {Trace Text string. See Section 3.4.1 Message} ERR=0202 Cannot perform action because CGM is not online ERR=0500 Unable to communicate with Sensor Command Description Response Comment GET_STATUS Get Instrument {Status Text string. See Section 3.4.1 Status Message} GET_STATE Get Instrument STATE=bbbbbb 5 binary number corresponding to the system state, outlined in Section 5 Perform OK Auto Response delayed until gauging complete Gauging Only ERR=0300 Gauging failed ERR=0202 Cannot perform action because System is not online MEASURE Perform Text string returns after measurement is complete. If Auto an {Measurement Measurement Message} error occurs at any time during the measurement, an ERR Only is returned instead. Filename is indexed by a 8 digit number 00000000.txt to 99999999.txt ERR=0200 Instrument status will not measurement ERR=0102 No receipe is loaded
GET_OPR Get Operator Info {Oprerator Msg} Text string. See Section 3.4.1 ERR=0020 No operator logged on REC=nnnn Set Recipe Number, with 'nnnn' is a number in the range 0000 to 9999 {Recipe Msg} Auto Only Recipe was set ERR=0100 Incorrectly formatted recipe number ERR=0101 Requested recipe is not defined ERR=0202 Cannot perform action because system is not online ERR=0104 Recipe Failed to Load ERR=0105 Obsolete Recipe ERR=0206 Cannot Change Recipe while glass is present GET_RECIPE Get Recipe Number REC=nnnn number in the range 0000 to 9999 ERR=0202 Cannot perform action because system is not online ERR=0102 No receipe is loaded GET_UTILITY_DATA Get Utility Data {Utility Message} Text string. See Section 3.4.1 ERR=0202 Cannot perform action because CGM is not online ERR=0500 Unable to communicate with Sensor Suite GET_TRACE_DATA Get Trace Data {Trace Message} Text string. See Section 3.4.1 ERR=0202 Cannot perform action because CGM is not online ERR=0500 Unable to communicate with Sensor Command Description Response Comment GET_STATUS Get Instrument Status {Status Message} Text string. See Section 3.4.1 GET_STATE Get Instrument STATE=bbbbbb 5 digit binary number corresponding to the system state, outlined in Section 5 Perform Gauging OK Auto Only Response delayed until gauging complete ERR=0300 Gauging failed ERR=0202 Cannot perform action because System is not online MEASURE Perform Measurement {Measurement Message} Auto Only Text string returns after measurement is complete. If an error occurs at any time during the measurement, an ERR is returned instead. Filename is indexed by a 8 digit number 00000000.txt to 99999999.txt ERR=0200 Instrument status will not measurement ERR=0102 No receipe is loaded
AxoScan%20CIM%20Comm%20Spec%202.0%20-%20June%2025%202014.docx