8-3 software changes Version Date New functionality Other changes Bug fixes 8-3m Text typed in Password at logon screen now not shown 8-3L Feed to LIMS did not work the text files – were not recognised on some sites. Code to mark calibrants as drained gave an error message. 8-3k 6 Nov 18 Get_support_files now gives the option to Split the history file into 2 within 6 months and – collect 7d, 6m, or all. Also it includes a list over 6 months. This allows the files collected for of files and folders in the BV200 directory. forwarding to support to be much smaller. Main program - PV results are now show as measured, rather than adjusted according to CHECK drift. Main program - Removed Daily Usage report Main program - Results shown with coloured highlighting, with patient samples are red text when rejected or voided by subsequent CHECK, and failed CHECKs highlighted better. Main program CHECK, LOW, HIGH barcodes – now indicate whether they are used as a calibrant or a control, or neither. Main program new Dips and Drift columns. Dip – number moved from Barcode to the new column. Drift moved from System Notes to the new column. Main program results get refreshed in total – rather than just adding a new row. Configure system removed Barcode to LIMS – option can only send the barcode listed in the – database Configure system removed Hold back results to – LIMS if CHECK fails once now always on – Plasma-D-viewer has more options to – hide/unhide some columns. Also has coloured text and background depending on the type of result displayed. 8-3j 17 Jan 18 Added tickboxes for CONTROL to Removed automatic display of UNKNOWN tab Fix to allow 13th dip for CHECK if retest ‘Over during running required on 12th dip and drain on 13th day results’ tab Coloured the tickboxes in tab ‘7-day results’
8-3 software changes Version Date New functionality Other changes Bug fixes 8-3m Text typed in Password at logon screen now not shown 8-3L Feed to LIMS did not work – the text files were not recognised on some sites. Code to mark calibrants as drained gave an error message. 8-3k 6 Nov 18 Get_support_files now gives the option to collect 7d, 6m, or all. Also it includes a list of files and folders in the BV200 directory. Split the history file into 2 – within 6 months and over 6 months. This allows the files collected for forwarding to support to be much smaller. Main program - PV results are now show as measured, rather than adjusted according to CHECK drift. Main program - Removed Daily Usage report Main program - Results shown with coloured highlighting, with patient samples are red text when rejected or voided by subsequent CHECK, and failed CHECKs highlighted better. Main program – CHECK, LOW, HIGH barcodes now indicate whether they are used as a calibrant or a control, or neither. Main program – new Dips and Drift columns. Dip number moved from Barcode to the new column. Drift moved from System Notes to the new column. Main program – results get refreshed in total rather than just adding a new row. Configure system – removed Barcode to LIMS option – can only send the barcode listed in the database Configure system – removed Hold back results to LIMS if CHECK fails once – now always on Plasma-D-viewer – has more options to hide/unhide some columns. Also has coloured text and background depending on the type of result displayed. 8-3j 17 Jan 18 Added tickboxes for CONTROL to ‘Over results’ tab Removed automatic display of UNKNOWN tab during running Coloured the tickboxes in ‘7-day results’ tab Fix to allow 13th dip for CHECK if retest required on 12th dip and drain on 13th day
8-3m_software_versions.docx