103 Volts, dc, 2mV range 104 Volts, dc, 10mV range 310-314 Thermocouple type E The third digit sets the range: 320-324 Thermocouple type J 0 = auto-ranging, 330-334 Thermocouple type K 4 = least sensitive range (10V) 340-344 Thermocouple type R 350-354 Thermocouple type S *Type B and N thermocouples may be used only 360-364 Thermocouple type T with an IMP marked with product status C3 370-374 Thermocouple type B* onwards 380-384 Thermocouple type N* 500 Current, dc, auto-ranging 501 Current, dc, 200µA Channels used for current measurements require 502 Current, dc, 2mA the fitting of a (See IMP Installation 100Ω shunt. 503 Current, dc, 20mA Guide) 504 Current, dc, 100mA Table 2.4: Mode codes for IMP type 1B (Strain) Code, m Mode Set Comments 000 Skip 100 Volts, dc, auto-ranging 101 Volts, dc, 20mV range There is no ‘10V range’ 102 Volts, dc, 200mV range 103 Volts dc, 2V range Resistance, 4-terminal 200 0.8mA drive, auto-ranging 201 0.8mA drive, range 25Ω 202 0.8mA drive, range 250Ω 203 0.8mA drive, range 2k5Ω Resistance, 3-terminal 210 0.8mA drive, auto-ranging *Only an IMP with a product status of 211 0.8mA drive, range C6 (or higher) are able to comply with 25Ω 212 0.8mA drive, range an auto-ranging command 250Ω 213 0.8mA drive, range 2k5Ω 4-terminal RTD/PRT, 100Ω, 400 “auto-ranging 411 range “20mV 412 range “200mV 413 range “2V RTD/PRT, 100Ω, 3-terminal 410 * Only an IMP with a product status of “auto-ranging 411 range C6 (or higher) are able to comply with “20mV 412 range an auto-ranging command “200mV 413 range “2V
103 Volts, dc, 2mV range 104 Volts, dc, 10mV range 310-314 Thermocouple type E The third digit sets the range: 0 = auto-ranging, 4 = least sensitive range (10V) *Type B and N thermocouples may be used only with an IMP marked with product status C3 onwards 320-324 Thermocouple type J 330-334 Thermocouple type K 340-344 Thermocouple type R 350-354 Thermocouple type S 360-364 Thermocouple type T 370-374 Thermocouple type B* 380-384 Thermocouple type N* 500 Current, dc, auto-ranging Channels used for current measurements require the fitting of a 100Ω shunt. (See IMP Installation Guide) 501 Current, dc, 200µA 502 Current, dc, 2mA 503 Current, dc, 20mA 504 Current, dc, 100mA Table 2.4: Mode codes for IMP type 1B (Strain) Code, m Mode Set Comments 000 Skip There is no ‘10V range’ 100 Volts, dc, auto-ranging 101 Volts, dc, 20mV range 102 Volts, dc, 200mV range 103 Volts dc, 2V range Resistance, 4-terminal 200 0.8mA drive, auto-ranging 201 0.8mA drive, 25Ω range 202 0.8mA drive, 250Ω range 203 0.8mA drive, 2k5Ω range Resistance, 3-terminal *Only an IMP with a product status of C6 (or higher) are able to comply with an auto-ranging command 210 0.8mA drive, auto-ranging 211 0.8mA drive, 25Ω range 212 0.8mA drive, 250Ω range 213 0.8mA drive, 2k5Ω range RTD/PRT, 100Ω, 4-terminal 400 “auto-ranging 411 “20mV range 412 “200mV range 413 “2V range RTD/PRT, 100Ω, 3-terminal 410 “auto-ranging * Only an IMP with a product status of C6 (or higher) are able to comply with an auto-ranging command 411 “20mV range 412 “200mV range 413 “2V range
3595-4U%20User%20Manual.docx