IEC 60904-7 Photovoltaic devices Part 7: Computation of spectral mismatch error introduced in the testing of a – – photovoltaic device IEC 60904-8 Photovoltaic devices Part 8: Measurement of spectral response of a photovoltaic (PV) device – – IEC 60904-9 Photovoltaic devices Part 9: Solar simulator performance requirements – – IEC 61646 Thin-film terrestrial photovoltaic (PV) modules Design qualification and type approval – – Unless otherwise indicated, all documents cited shall be the latest published versions. NOTICE: Terminology Abbreviations and Acronyms organic photovoltaic OPV— dye-sensitized solar cell DSSC— device under test DUT— device RD—reference monitor TM—temperature measurement unit SMU—source test plane ITP—isothermal interface CI—control simulator SS—solar wavelength (nm) — spectrum response SR— response SR1—spectral responsivity (A·W–1 or A·m2·W–1) SR2—spectral test conditions STC—standard normal test conditions NTC — reporting conditions SRC—standard Definitions the temperature (°C) of test sample cell temperature — a rotating blade or other device used to modulate a light source chopper— incident photon-electron conversion efficiency IPCE— the isothermal plane intended to contain DUT at the reference irradiance level isothermal test plane— a source of radiant energy to simulate natural sunlight and used for cell performance light source— measurement a photo detector incorporated into the optical system to monitor the amount of light monitor photo-detector— reaching the device under test, enabling adjustments to be made to accommodate varying light intensity chopped light from a monochromatic source reaching the reference photo-detector monochromatic beam—a or device under test an optical device that allows a selected wavelength of light to pass while blocking other monochromator— wavelengths number of collected electrons per incident photon at a specific wavelength in percent quantum efficiency—the units
IEC 60904-7 – Photovoltaic devices – Part 7: Computation of spectral mismatch error introduced in the testing of a photovoltaic device IEC 60904-8 – Photovoltaic devices – Part 8: Measurement of spectral response of a photovoltaic (PV) device IEC 60904-9 – Photovoltaic devices – Part 9: Solar simulator performance requirements IEC 61646 – Thin-film terrestrial photovoltaic (PV) modules – Design qualification and type approval NOTICE: Unless otherwise indicated, all documents cited shall be the latest published versions. Terminology Abbreviations and Acronyms OPV— organic photovoltaic DSSC— dye-sensitized solar cell DUT— device under test RD—reference device TM—temperature monitor SMU—source measurement unit ITP—isothermal test plane CI—control interface SS—solar simulator — wavelength (nm) SR— spectrum response SR1—spectral response SR2—spectral responsivity (A·W–1 or A·m2·W–1) STC—standard test conditions NTC — normal test conditions SRC—standard reporting conditions Definitions cell temperature — the temperature (°C) of test sample chopper— a rotating blade or other device used to modulate a light source IPCE— incident photon-electron conversion efficiency isothermal test plane— the isothermal plane intended to contain DUT at the reference irradiance level light source— a source of radiant energy to simulate natural sunlight and used for cell performance measurement monitor photo-detector— a photo detector incorporated into the optical system to monitor the amount of light reaching the device under test, enabling adjustments to be made to accommodate varying light intensity monochromatic beam—a chopped light from a monochromatic source reaching the reference photo-detector or device under test monochromator— an optical device that allows a selected wavelength of light to pass while blocking other wavelengths quantum efficiency—the number of collected electrons per incident photon at a specific wavelength in percent units
5647.docx