c. When approved, TVE-005OP3B and SM-005OP2B will provide the spare equipment requirements. d. When other wireless components are approved, provide ten (10) components with one spare components for each 10 issued.// F. TIP Cable Systems: Connect the system to the TIP system provided as a part of Speciation Section 27 15 00. Provide additional TIP equipment, interfaces and connections as required by System design. Provide secured pathway(s) and lockable cabinet/rack(s) as required. G. Interface Equipment: 1. TCR: a. Code Blue Annunciation Station: 1) The Code Blue Remote Annunciation Station shall be located in the Telephone Operators Room, // or _____ //. 2) The Annunciation Station shall be connected to the System via hard wire connection(s) that shall contain all the electrical supervisory tone signals, visual bulbs, read out panel to indicate the location of the Code and system troubles. 3) The System shall not be connected to the Telephone system unless specifically APPROVED BY VA HEADQUARTERS (005OP3B) and (005OP2B) PRIOR TO CONTRACT BID. 4) The Annunciation Station shall be installed in a location directly viewable and the readout is completely readable from the Public Address Microphone Control Console. 5) Provide one (1) spare panel. b. Electrical Supervision Trouble Annunciator Panel: 1) The Electrical Supervision Trouble Annunciation Panel shall be located in the Telephone Operators Room, Police Control Center, associate Nurses Station(s) // and _____ //. 2) The panel(s) shall be compatible with the generated electrical and/or electronic supervising signals to continuously monitor the operating condition for the System head-end processing equipment, master stations, staff stations, patient stations, duty stations, audio power amplifier(s), UPS, power supplies, dome lights and interconnecting trunks. The panels shall generate an audible and visual signal when the System’s supervising system detects a system and equipment trouble or trunk-line is malfunctioning.
c. When approved, TVE-005OP3B and SM-005OP2B will provide the spare equipment requirements. d. When other wireless components are approved, provide ten (10) components with one spare components for each 10 issued.// F. TIP Cable Systems: Connect the system to the TIP system provided as a part of Speciation Section 27 15 00. Provide additional TIP equipment, interfaces and connections as required by System design. Provide secured pathway(s) and lockable cabinet/rack(s) as required. G. Interface Equipment: 1. TCR: a. Code Blue Annunciation Station: 1) The Code Blue Remote Annunciation Station shall be located in the Telephone Operators Room, // or _____ //. 2) The Annunciation Station shall be connected to the System via hard wire connection(s) that shall contain all the electrical supervisory tone signals, visual bulbs, read out panel to indicate the location of the Code and system troubles. 3) The System shall not be connected to the Telephone system unless specifically APPROVED BY VA HEADQUARTERS (005OP3B) and (005OP2B) PRIOR TO CONTRACT BID. 4) The Annunciation Station shall be installed in a location directly viewable and the readout is completely readable from the Public Address Microphone Control Console. 5) Provide one (1) spare panel. b. Electrical Supervision Trouble Annunciator Panel: 1) The Electrical Supervision Trouble Annunciation Panel shall be located in the Telephone Operators Room, Police Control Center, associate Nurses Station(s) // and _____ //. 2) The panel(s) shall be compatible with the generated electrical and/or electronic supervising signals to continuously monitor the operating condition for the System head-end processing equipment, master stations, staff stations, patient stations, duty stations, audio power amplifier(s), UPS, power supplies, dome lights and interconnecting trunks. The panels shall generate an audible and visual signal when the System’s supervising system detects a system and equipment trouble or trunk-line is malfunctioning.
275223.docx