DaqLink Specifications Data Logger Input Types DBSA710 and DBSA720 External Inputs Internal Inputs Four external inputs with selectable sensor DBSA710 type for each input: PT-100 Temperature 4 to 20 mA DBSA720 0 to 1 V Digital Temperature 0 to 50 mV Digital Relative Humidity PT-100 (2-wire) Contact (Open/Closed) Thermocouple J, K or T counter (Input 4 only) Frequency (Input 4 only) User-defined sensors DBSA710 and DBSA720 Outputs Alarm Output (Output 1) External Power Excitation (transducers usage) 12 VDC @ 300 mA Open collector Reset fuse overload protection Close position resistance: 50 Ω Max. Load: 50 mA, 3 V DC Logger Input Specifications DBSA710 and DBSA720 Internal PT-100 Internal Digital Temperature Internal Digital Temperature (DBSA720) Relative Humidity (DBSA710) (DBSA720) Type: PT-100 Type: Digital sensor Range: 5 to 95 % RH Range: -20 to 50 Range: -20 to 50 Resolution: 0.5% RH ˚C ˚C Resolution: 0.1 Resolution: 0.1 Accuracy: ˚C ˚C Accuracy: ±0.3 Accuracy: ±0.5 3% RH (DataSuite software) ˚C ˚C 4% RH (Logger LCD) Temperature Thermocouple Temperature Thermocouple Temperature Thermocouple J K T
DaqLink Specifications Data Logger Input Types DBSA710 and DBSA720 External Inputs Four external inputs with selectable sensor type for each input: 4 to 20 mA 0 to 1 V 0 to 50 mV PT-100 (2-wire) Contact (Open/Closed) Thermocouple J, K or T counter (Input 4 only) Frequency (Input 4 only) User-defined sensors Internal Inputs DBSA710 PT-100 Temperature DBSA720 Digital Temperature Digital Relative Humidity DBSA710 and DBSA720 Outputs Alarm Output (Output 1) External Power Excitation (transducers usage) Open collector Close position resistance: 50 Ω Max. Load: 50 mA, 3 V DC 12 VDC @ 300 mA Reset fuse overload protection Logger Input Specifications DBSA710 and DBSA720 Internal PT-100 Temperature (DBSA710) Internal Digital Temperature (DBSA720) Internal Digital Relative Humidity (DBSA720) Type: PT-100 Range: -20 to 50 ˚C Resolution: 0.1 ˚C Accuracy: ±0.3 ˚C Type: Digital sensor Range: -20 to 50 ˚C Resolution: 0.1 ˚C Accuracy: ±0.5 ˚C Range: 5 to 95 % RH Resolution: 0.5% RH Accuracy: 3% RH (DataSuite software) 4% RH (Logger LCD) Temperature Thermocouple J Temperature Thermocouple K Temperature Thermocouple T
DataSuite%20User%20Guide_v2.5.0.0_Final.docx