Range: -200 to 1,000 Range: -200 to 1,000 -200 to 400 ˚C ˚C ˚C Resolution: 0.1 Resolution: 0.1 Resolution: 0.1 ˚C ˚C ˚C Accuracy: Accuracy: Accuracy: -200 to 60 ±0.5 % -200 to 60 ±0.5 % -200 to 60 ±0.5 % – ˚C – ˚C – ˚C -60 to 60 ±0.5 -60 to 60 ±0.5 -60 to 60 ±0.5 ˚C ˚C ˚C ˚C ˚C ˚C 60 to 1,000 ±0.5 % 60 to 1,000 ±0.5 % 60 to 400 ±0.5 % ˚C ˚C ˚C Cold junction Cold junction Cold junction compensation: ±0.3 compensation: ±0.3 compensation: ±0.3 ˚C ˚C ˚C Temperature PT-100 Counter Frequency (2-wire) (Input 4 only) 4 only) Range: -200 to 400 1 to 65,536 counts Range: 20 Hz to 4 KHz ˚C Resolution: 0.1 Resolution: 1 count Input signal: 0 to 5 V ˚C) Accuracy: Frequency range: 0 to 4 Input impedance: 470 Ω KHz -200 to - 60 ±0.5 % ˚C Input signal: 0 to 5 V -60 to 60 ±0.3 60 to ˚C ˚C 400 ±0.5 % impedance: 470 ˚C Ω 4 to 20 mA 0 to 1 V 0 to 50 mV 4 to 20 mA 0 to 1 V 0 to 50 mV Resolution: 47.6 Resolution: 200 Resolution: 37 μA μV μV Accuracy: ±0.5 % Accuracy: ±0.5 % Accuracy: ±0.5 % Loop impedance: 21 impedance: 25 MΩ Input impedance: 25 MΩ Ω Maximum voltage: 5.2 V Maximum voltage: 5.2 V Contact Open/Close General Specifications DaqLink Logger Sampling Features Memory capacity: ~38,000 samples in Online Mode ~59,000 samples in Optimized Memory Mode (depending on logger configuration) Sampling rate: Once per second to once every 18 hours Channel separation: 80 dB DBSA710 and DBSA720 Hardware Specifications Display 2-row LCD 16 character display Power Supply : Internal rechargeable 4.8 V, 800 mAh NiMH battery (2 batteries in series) Battery Life: Up to six months depending on logger configuration AC Power:
Range: -200 to 1,000 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 1,000 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C Range: -200 to 1,000 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 1,000 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C -200 to 400 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 400 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C Temperature PT-100 (2-wire) Counter (Input 4 only) Frequency 4 only) Range: -200 to 400 ˚C Resolution: 0.1 ˚C) Accuracy: -200 to - 60 ˚C ±0.5 % -60 to 60 ˚C ±0.3 ˚C 60 to 400 ˚C ±0.5 % 1 to 65,536 counts Resolution: 1 count Frequency range: 0 to 4 KHz Input signal: 0 to 5 V impedance: 470 Ω Range: 20 Hz to 4 KHz Input signal: 0 to 5 V Input impedance: 470 Ω 4 to 20 mA 0 to 1 V 0 to 50 mV 4 to 20 mA Resolution: 47.6 μA Accuracy: ±0.5 % Loop impedance: 21 Ω 0 to 1 V Resolution: 200 μV Accuracy: ±0.5 % impedance: 25 MΩ Maximum voltage: 5.2 V 0 to 50 mV Resolution: 37 μV Accuracy: ±0.5 % Input impedance: 25 MΩ Maximum voltage: 5.2 V Contact Open/Close General Specifications DaqLink Logger Sampling Features Memory capacity: ~38,000 samples in Online Mode ~59,000 samples in Optimized Memory Mode (depending on logger configuration) Sampling rate: Once per second to once every 18 hours Channel separation: 80 dB DBSA710 and DBSA720 Hardware Specifications Display 2-row LCD 16 character display Power Supply : Internal rechargeable 4.8 V, 800 mAh NiMH battery (2 batteries in series) Battery Life: Up to six months depending on logger configuration AC Power:
DataSuite%20User%20Guide_v2.5.0.0_Final.docx