Wave form trace speed : 25 and 50mm/sec ECG Leads selection: I, II, III, aVR, aVF, aVL, V, V1-V6, TEST Selectable neonatal & pediatric operation Ability to detect neonatal & pediatric heart rate range, qrs widths and amplitudes. Numeric heart rate Real time wave form Variable gain: 0.5,1,2&4 Heart beat indication NIBP Pressure ,oscillometric principal. Systolic , diastolic and mean pressure Rising cuff/continuous pressure display Temperature: Numeric temperature selectable Temperature with probe (external and rectal) Ability to measure neonate and pediatric temperature range Pulse Oximeter Number oxygen saturation measurement Range: 0-100% Sensor for neonate/infant and pediatrics Respiration impedance method using ECG leads for sensing Breathing rate range, sweep speed Selectable apnea alarm Built-in battery with charger for at least 2 hours or more ETCO2 Mainstream ETCO2 Adult and Neonate Measurement Range: 0 ~ 99mmHg Sampling Rate: 50 ml/min Respiration Rate: 0 ~ 150 breaths/min The system must be complete with all sensors, probes and cables and accessories. Or equivalent FDA/CE/JIS Marked 9 Central Monitor 1
Wave form trace speed : 25 and 50mm/sec ECG Leads selection: I, II, III, aVR, aVF, aVL, V, V1-V6, TEST Selectable neonatal & pediatric operation Ability to detect neonatal & pediatric heart rate range, qrs widths and amplitudes. Numeric heart rate Real time wave form Variable gain: 0.5,1,2&4 Heart beat indication NIBP Pressure ,oscillometric principal. Systolic , diastolic and mean pressure Rising cuff/continuous pressure display Temperature: Numeric temperature selectable Temperature with probe (external and rectal) Ability to measure neonate and pediatric temperature range Pulse Oximeter Number oxygen saturation measurement Range: 0-100% Sensor for neonate/infant and pediatrics Respiration impedance method using ECG leads for sensing Breathing rate range, sweep speed Selectable apnea alarm Built-in battery with charger for at least 2 hours or more ETCO2 Mainstream ETCO2 Adult and Neonate Measurement Range: 0 ~ 99mmHg Sampling Rate: 50 ml/min Respiration Rate: 0 ~ 150 breaths/min The system must be complete with all sensors, probes and cables and accessories. Or equivalent FDA/CE/JIS Marked 9 Central Monitor 1
document-PD_300_BEDDED_521_2017_1514292975180.docx