MicroLite Specifications General Specifications LITE5008 LITE5008P LITE5032P-RH LITE5016 LITE5032P Internal Temp Internal Temp Internal Temp/RH 4-20 mA 0-10 V External NTC Temp: -40 to 80 °C Range -40 to 80 °C -40 to 80 °C 4-20 mA 0-10 V -50 to 150 °C RH: 5-95 % ±0.3˚C @ 0 to 60 ±0.3 ˚C ˚C (after Temp: ±0.3 ˚C Accuracy ± 0.3 ±20 µA ±5 mV ˚C ±0.3 @ -40 to 0 calibration ˚C RH: ± 2 %RH and 60 to 80 with Fourtec ˚C ˚C after calibration NTC sensor) only 1 µA 1 mV Temp: 0.1 °C (display) (display) (display) 0.02 °C 0.01 °C Resolution 0.03 °C or better or better 0.01 °C (software) 5 µA 4 mV RH: 0.05 %RH (software) (software) Maximum N/A N/A N/A 25 mA, 3 V 12 V N/A Load Dew X X N/A N/A N/A √ LCD √ √ √ √ √ √ LED Alarm X √ √ √ √ √ Indicator Sample 32,000 8,000 and 16,000 8,000 and 32,000 32,000 Memory (16K per channel) Once per second Sampling / once every 2 Once per second / once every 18 hours Rate hours CR2032 3V 1.5 years, CR2032 3V Battery Type depending on / Battery Life sampling rate 2 years, depending on sampling rate and operating conditions and operating conditions Operating -40 to 80 °C Environment Dimensions 11 x 3.9 x 2.6 cm 11 x 3.9 x 3 cm 14 x 3.9 x 2.6 cm (with cap) Weight 45.5 g
MicroLite Specifications General Specifications LITE5008 LITE5016 LITE5008P LITE5032P LITE5032P-RH Internal Temp Internal Temp Internal Temp/RH 4-20 mA 0-10 V External NTC Range -40 to 80 °C -40 to 80 °C Temp: -40 to 80 °C RH: 5-95 % 4-20 mA 0-10 V -50 to 150 °C Accuracy ± 0.3 ˚C ±0.3˚C @ 0 to 60 ˚C ±0.3 ˚C @ -40 to 0 ˚C and 60 to 80 ˚C after calibration only Temp: ±0.3 ˚C RH: ± 2 %RH ±20 µA ±5 mV ±0.3 ˚C (after calibration with Fourtec NTC sensor) Resolution 0.1 °C (display) 0.01 °C (software) 0.03 °C or better Temp: 0.01 °C RH: 0.05 %RH 1 µA (display) 5 µA (software) 1 mV (display) 4 mV (software) 0.02 °C or better Maximum Load N/A N/A N/A 25 mA, 3 V 12 V N/A Dew X X √ N/A N/A N/A LCD √ √ √ √ √ √ LED Alarm Indicator X √ √ √ √ √ Sample Memory 8,000 and 16,000 8,000 and 32,000 32,000 (16K per channel) 32,000 Sampling Rate Once per second / once every 2 hours Once per second / once every 18 hours Battery Type / Battery Life CR2032 3V 1.5 years, depending on sampling rate and operating conditions CR2032 3V 2 years, depending on sampling rate and operating conditions Operating Environment -40 to 80 °C Dimensions (with cap) 11 x 3.9 x 2.6 cm 11 x 3.9 x 3 cm 14 x 3.9 x 2.6 cm Weight 45.5 g
DataSuite%20User%20Guide_v2.5.0.0_Final.docx