Anesthesia Instrument Trolley (Qty: 1) Entirely made of AISI304 stainless steel Anesthesia instruments connection system 8 wheels Recognition through magnet Should be from Washer Disinfector manufacturer Air-Water Gun (Qty: 1) Spray gun made of stainless steel AISI304, suitable for water and compressed air, supplied with 8 nozzles Tube with respective connection to the power supply of water or air under the tub or at floor Handle thermally insulated Accessories contained in a structure suitable for wall mounting The length of the connecting hose: 2m Heat Sealer (Qty: 1) Standards: CE / CEM / EN 868-5 / DIN 58953-7 Sealing: 15 mm / flat and full weld for better checking, unlimited sealing length. Display: 1 line LCD Temperature range: 20-230C° with a 2% regulation tolerance Adjustable pouches guide (0-30 mm from edge). Welds counter with reset. Should seals every kind of pouches (steam, Tyvek, OE, alu, Ultra, etc). Starts automatically when the set temperature is reached. Environmentally responsible: automatic start and stop by photocell. Safety: critical parameters control complying to ISO 11607-2 standard (temperature & pressure) Automatic stop if critical parameters exceed limits
Anesthesia Instrument Trolley (Qty: 1) Entirely made of AISI304 stainless steel Anesthesia instruments connection system 8 wheels Recognition through magnet Should be from Washer Disinfector manufacturer Air-Water Gun (Qty: 1) Spray gun made of stainless steel AISI304, suitable for water and compressed air, supplied with 8 nozzles Tube with respective connection to the power supply of water or air under the tub or at floor Handle thermally insulated Accessories contained in a structure suitable for wall mounting The length of the connecting hose: 2m Heat Sealer (Qty: 1) Standards: CE / CEM / EN 868-5 / DIN 58953-7 Sealing: 15 mm / flat and full weld for better checking, unlimited sealing length. Display: 1 line LCD Temperature range: 20-230C° with a 2% regulation tolerance Adjustable pouches guide (0-30 mm from edge). Welds counter with reset. Should seals every kind of pouches (steam, Tyvek, OE, alu, Ultra, etc). Starts automatically when the set temperature is reached. Environmentally responsible: automatic start and stop by photocell. Safety: critical parameters control complying to ISO 11607-2 standard (temperature & pressure) Automatic stop if critical parameters exceed limits
document-HD_P_E_3_8_33__2017_1515155159116.docx