and Run Clicking Setup and Run will send the setup to the logger and immediately start logging data. Only Clicking Only will send the setup to the logger but it will not actually start logging data. You will need to select Run from the Logger context menu or go back into the Setup dialog. Clicking Close will close the Setup dialog without saving any of your configurations. Note: Running the logger clears the logger memory. All previously recorded data will be erased when you a new logging session and DaqLink Setup Tab A feature of the DataNet and DaqLink systems is the ability to configure alarm levels into the loggers so any breach of these levels as recorded on the data logger is monitored by the software, and the user is notified via Email/SMS/audible and visual indicators. The Alarm Setup tab lets you configure each sensor for the following alarms: Low Pre-Low Pre-High High Contact Open / Contact Close (when the Contact sensor is selected) Using the Pre Alarms gives the user ample warning of a possible breach of real alarm levels, so necessary measures can be taken to prevent such a breach. Simply select the check box for the type of alarm you wish to be notified. You can select all alarms too. Enter the alarm value in the text box. Figure 89: DataNet and DaqLink Alarm Setup tab Delay and Pre-alarm Delay The time the alarm (or pre-alarm) is activated. You may not want to have the alarm sound immediately after the alarm level is breached as you are only interested in a condition where the alarm level lasted a certain amount of time. For example, if monitoring change of temperature in a freezer, you would only be interested in a High alarm which lasted more than one minute as anything else could just be caused by the freezer door opening for a few seconds, as opposed to a power failure leading to rise in temperature. Duration The duration of the alarm that is set off by one or more sensors breaching the alarm level.
and Run Clicking Setup and Run will send the setup to the logger and immediately start logging data. Only Clicking Only will send the setup to the logger but it will not actually start logging data. You will need to select Run from the Logger context menu or go back into the Setup dialog. Clicking Close will close the Setup dialog without saving any of your configurations. Note: Running the logger clears the logger memory. All previously recorded data will be erased when you a new logging session and DaqLink Setup Tab A feature of the DataNet and DaqLink systems is the ability to configure alarm levels into the loggers so any breach of these levels as recorded on the data logger is monitored by the software, and the user is notified via Email/SMS/audible and visual indicators. The Alarm Setup tab lets you configure each sensor for the following alarms: Low Pre-Low Pre-High High Contact Open / Contact Close (when the Contact sensor is selected) Using the Pre Alarms gives the user ample warning of a possible breach of real alarm levels, so necessary measures can be taken to prevent such a breach. Simply select the check box for the type of alarm you wish to be notified. You can select all alarms too. Enter the alarm value in the text box. Figure 89: DataNet and DaqLink Alarm Setup tab Delay and Pre-alarm Delay The time the alarm (or pre-alarm) is activated. You may not want to have the alarm sound immediately after the alarm level is breached as you are only interested in a condition where the alarm level lasted a certain amount of time. For example, if monitoring change of temperature in a freezer, you would only be interested in a High alarm which lasted more than one minute as anything else could just be caused by the freezer door opening for a few seconds, as opposed to a power failure leading to rise in temperature. Duration The duration of the alarm that is set off by one or more sensors breaching the alarm level.
DataSuite%20User%20Guide_v2.5.0.0_Final.docx