Weight Value is 1). Check Load cell 1). Load cell Error increased regular connection 2). Load cell rate, but not return 2). Measure Load cell connection Error to Resistance “Zero” Weight Value is Load cell Output wire increased to under (SIG+, SIG-) is Make wire correction Zero switched Load cell broken or Load cell Indicator connection Load cell connection Check PASS” display “UN Power was “ON” Remove weight on the when some weight is Load cell on the load cell? or display 1). Load cell broken 1). Load cell “OL” “UL” or Indicator 2). Load cell connection connection Error 2). Loading over than 3). Remove over Max. Capacity loaded weight 2-2. Calibration Process Cause Treatment When Max.capacity/digit Re-input the Max. Capacity, less than 20.00 Err 01 value is over 20.00 (Max. Capacity / Digit) Standard weight value is Re-input Standard weight value with Number Err 04 over than Max. Capacity keys, under Max. Capacity Standard weight value is Re-input Standard weight value with Number Err 05 less than 10% of Max. keys, more than 10% of Max. Capacity Capacity 1. Amp. Gain is too big standard weight’s weight with set value. 2. Sig+ and Sig- wire If there is difference between set value and real Err 06 connection error weight, please re-input the value 3. Test weight is not loaded (set value is too small)
Weight Value is increased regular rate, but not return to “Zero” 1). Load cell Error 2). Load cell connection Error 1). Check Load cell connection 2). Measure Load cell Resistance Weight Value is increased to under Zero Load cell Output wire (SIG+, SIG-) is switched Make wire correction “UN PASS” display Load cell broken or Indicator connection Load cell Load cell connection Check Power was “ON” when some weight is on the load cell? Remove weight on the Load cell “OL” or “UL” display 1). Load cell broken or Indicator connection Error 2). Loading over than Max. Capacity 1). Load cell 2). Load cell connection 3). Remove over loaded weight 2-2. Calibration Process Cause Treatment Err 01 When Max.capacity/digit value is over 20.00 Re-input the Max. Capacity, less than 20.00 (Max. Capacity / Digit) Err 04 Standard weight value is over than Max. Capacity Re-input Standard weight value with Number keys, under Max. Capacity Err 05 Standard weight value is less than 10% of Max. Capacity Re-input Standard weight value with Number keys, more than 10% of Max. Capacity Err 06 1. Amp. Gain is too big 2. Sig+ and Sig- wire connection error 3. Test weight is not loaded standard weight’s weight with set value. If there is difference between set value and real weight, please re-input the value (set value is too small)
CI-5100AENG-ver.docx