output that external input is normal or TEST 6. External Input)TEST Mode not the pure analogue value when TEST 7. Un-Calibrated Analogue TEST Mode not calibration you installed Analogue Option card, you can test Analogue output test with 2” mode. “TEST (Please check detailed information) Enter to TEST Mode Turn on the power with pressing key. Then, display will show , then , and then press No. key and move to the certain TEST mode. Exit from TEST Mode Press key to exit from each TEST mode. Then, display will show Under TEST 3. mode, press key to exit . 2. Error and Treatment 2-1. Load Cell Installation Error Cause Treatment Remark 1).Input Resistance 1). Load cell broken of and “EX+” “EX-“ 2). Load cell 1). Measure is about isolation input/output resistance 350Ω~450Ω. resistance error of Load cell. 3). Weighing part 2). Output Weight Value is touches other 2). Measure Load cell Resistance of “EX-“ unstable devices or some isolation resistance and is about “EX+” weight is on the 350Ω. weighing part 3) Check attach point 4). Summing with other devices. 3). Isolate Error Resistance is more than 100Ω
output TEST 6. External Input)TEST Mode that external input is normal or not TEST 7. Un-Calibrated Analogue TEST Mode the pure analogue value when not calibration you installed Analogue Option card, you can test Analogue output test with “TEST 2” mode. (Please check detailed information) Enter to TEST Mode Turn on the power with pressing key. Then, display will show , then , and then press No. key and move to the certain TEST mode. Exit from TEST Mode Press key to exit from each TEST mode. Then, display will show Under TEST 3. mode, press key to exit . 2. Error and Treatment 2-1. Load Cell Installation Error Cause Treatment Remark Weight Value is unstable 1). Load cell broken 2). Load cell isolation resistance error 3). Weighing part touches other devices or some weight is on the weighing part 4). Summing Error 1). Measure input/output resistance of Load cell. 2). Measure Load cell isolation resistance 3) Check attach point with other devices. 1).Input Resistance of “EX+” and “EX-“ is about 350Ω~450Ω. 2). Output Resistance of “EX-“ and “EX+” is about 350Ω. 3). Isolate Resistance is more than 100Ω
CI-5100AENG-ver.docx