8-1 software changes for the last year Version Date New functionality Other changes Bug fixes 8-1bp 14 June 18 processing of CONTROL 2.5 samples 8-1bo 13 June 18 RFC029 - ability to process CONTROL 3, Calibrant barcodes is now Type + barcode Fixed bug with retesting of CHECK not 4, 6, and 10 calibrants (e.g. CHECK 12345), and does not include draining if failed after 13th dip – User can determine whether to retest the Dip number. CONTROL samples (since they are over Get_Support_Files changed to gather more 2.0) information to send to the support team Removed processing of 22 digit calibrants Changed order of items on Maintenance Menu, so items most used are at the top Changed names on Configuration Menu 8-1bn 23 Apr 18 New columns displayed for Dip number Altered PV calculation so it is consistent Reset of Belt number when HIGH and Drift. across all reports (specifically calibrant detected has been reworked User notes information is now written and Uncertainty). This means the auto-adjust Standardised the format of colours out along with all the other information of results based on CHECK drift is no longer when displaying different types of output to spreadsheet (customer used. CHECK results will be shown as records (e.g. CHECK, LOW, HIGH, requested change) measured, and not compensated to 1.60 CONTROL 2.5, etc) User can specify the format for the (PV25). This does not affect how accurate the barcodes for CHECK, LOW, HIGH, and Viscometer is. CONTROL 2.5 (customer requested Changed format of the barcode for change) LOW, HIGH, and CONTROL 2.5, removing the Dip number. This was requested by some sites so they can compare the results from these types of fluid outside of the Benson software Drift percentage is show in all reports for CHECK, LOW, HIGH, and CONTROL 2.5 The History data file has been split into 2 – results within 6 months and results over 6 months. This allows the file generated by Get_Support_Files to be a reasonable size
8-1 software changes for the last year Version Date New functionality Other changes Bug fixes 8-1bp 14 June 18 processing of CONTROL 2.5 samples 8-1bo 13 June 18 RFC029 - ability to process CONTROL 3, 4, 6, and 10 calibrants User can determine whether to retest CONTROL samples (since they are over 2.0) Calibrant barcodes is now Type + barcode (e.g. CHECK – 12345), and does not include the Dip number. Get_Support_Files changed to gather more information to send to the support team Removed processing of 22 digit calibrants Changed order of items on Maintenance Menu, so items most used are at the top Changed names on Configuration Menu Fixed bug with retesting of CHECK not draining if failed after 13th dip 8-1bn 23 Apr 18 New columns displayed for Dip number and Drift. User notes information is now written out along with all the other information output to spreadsheet (customer requested change) User can specify the format for the barcodes for CHECK, LOW, HIGH, and CONTROL 2.5 (customer requested change) Altered PV calculation so it is consistent across all reports (specifically and Uncertainty). This means the auto-adjust of results based on CHECK drift is no longer used. CHECK results will be shown as measured, and not compensated to 1.60 (PV25). This does not affect how accurate the Viscometer is. Changed format of the barcode for LOW, HIGH, and CONTROL 2.5, removing the Dip number. This was requested by some sites so they can compare the results from these types of fluid outside of the Benson software Drift percentage is show in all reports for CHECK, LOW, HIGH, and CONTROL 2.5 The History data file has been split into 2 – results within 6 months and results over 6 months. This allows the file generated by Get_Support_Files to be a reasonable size Reset of Belt number when HIGH calibrant detected has been reworked Standardised the format of colours when displaying different types of records (e.g. CHECK, LOW, HIGH, CONTROL 2.5, etc)
bv200_user_software_versions_to_8-1bp.docx