The X actuator is mounted directly over the isocenter point, and is X Actuator: referred to Maxx-1100 unit 3. 4.1.1 LASER LINE ADJUSTMENT Proper alignment of the laser lines requires precise adjustment of the laser lines within the confines of the treatment room. Two adjustments are possible for each laser diode, namely, tilt adjustment and rotation adjustment. (Figure 4.2) 4.1.1.1 TILT ADJUSTMENT Tilt adjustment allows the laser line to pivot from end to end for the moving laser, adjusting the positioning relative to the moving axis of the Maxx-1100 unit, and side to side for the fixed laser relative to the moving axis of the Maxx-1100 unit. (Figure 4.2) 4.1.1.2 ROTATION ADJUSTMENT For the moving laser the rotation adjustment alters the angle of the laser line to be parallel to the moving axis of the Maxx-1100 unit by rotating the laser diode. The rotation adjustment alters the fixed laser line angle so that it is perpendicular to the moving laser. (Figure 4.2) 4.2 SYSTEM CONTROLLER PC. (Optional) The system controller (PC) can control multiple-axis configurations, allowing the movement of up to three Maxx-1100 units. All controllers include a Monitor, a router, a USB to network adapter and all necessary cabling. (Figure 7.1) 4.2.1 CONTROLLER CONNECTIONS The system controller has a multiple output router; this is used to communicate with each Maxx-1100 unit which are connected to the router with a Cat6 cable. (Figure 7.1)
X Actuator: The X actuator is mounted directly over the isocenter point, and is referred to Maxx-1100 unit 3. 4.1.1 LASER LINE ADJUSTMENT Proper alignment of the laser lines requires precise adjustment of the laser lines within the confines of the treatment room. Two adjustments are possible for each laser diode, namely, tilt adjustment and rotation adjustment. (Figure 4.2) 4.1.1.1 TILT ADJUSTMENT Tilt adjustment allows the laser line to pivot from end to end for the moving laser, adjusting the positioning relative to the moving axis of the Maxx-1100 unit, and side to side for the fixed laser relative to the moving axis of the Maxx-1100 unit. (Figure 4.2) 4.1.1.2 ROTATION ADJUSTMENT For the moving laser the rotation adjustment alters the angle of the laser line to be parallel to the moving axis of the Maxx-1100 unit by rotating the laser diode. The rotation adjustment alters the fixed laser line angle so that it is perpendicular to the moving laser. (Figure 4.2) 4.2 SYSTEM CONTROLLER PC. (Optional) The system controller (PC) can control multiple-axis configurations, allowing the movement of up to three Maxx-1100 units. All controllers include a Monitor, a router, a USB to network adapter and all necessary cabling. (Figure 7.1) 4.2.1 CONTROLLER CONNECTIONS The system controller has a multiple output router; this is used to communicate with each Maxx-1100 unit which are connected to the router with a Cat6 cable. (Figure 7.1)
CemarElectro-Maxx1100-Manual-V-lw-1.1.5.25817July2018.docx