F83 Analogue output setting Under option installed F89 Span Value check F90 Date check / change F91 Time / change 5-6. Function List detailed information. Set-Up / Calibration Mode Selection Clear Set-Up mode F00 Enter Calibration Mode Decimal Point Setting 0 No Decimal point ● 1 1st place under Zero (0.0) F01 2 2nd place under Zero (0.00) 3 3rd place under Zero (0.000) Back up mode selection 0 Normal mode ● F02 1 Back up mode setting This is set acceptable range of weighing part. “Steady” If there is vibration on weighing part, you can set this function and 0 reduce the vibration effect on weighing process. F03 5 ∫ 0 : Weak vibration 9 ∫ 9 : Strong Vibration Zero setting
F83 Analogue output setting Under option installed F89 Span Value check F90 Date check / change F91 Time / change 5-6. Function List detailed information. Set-Up / Calibration Mode Selection F00 Clear Set-Up mode Enter Calibration Mode Decimal Point Setting F01 ● 0 No Decimal point 1 1st place under Zero (0.0) 2 2nd place under Zero (0.00) 3 3rd place under Zero (0.000) Back up mode selection F02 ● 0 Normal mode 1 Back up mode setting F03 5 0 ∫ 9 This is set “Steady” acceptable range of weighing part. If there is vibration on weighing part, you can set this function and reduce the vibration effect on weighing process. 0 : Weak vibration ∫ 9 : Strong Vibration Zero setting
CI-5100AENG-ver.docx