Due to external causes(Temperature, wind, and dust), there are small weight difference, indicator will ignore the weight difference and 0 display Zero. F04 5 For this compensation function, indicator will estimate the weight ∫ 9 difference is over the set range during fixed time period. If there is large weight difference over set range within fixed time period, the is breaking and will find new zero point. “Zero” Auto Zero Range setting Within the Zero” weighing part is steady, indicator will “Auto 00 display current weight as “Zero” F05 00 If the weighing part is not indicator will display current ∫ “Steady”, 99 weight. ( : ± Set + weight unit) Digital Filter setting 00 set value for weak vibration F06 15 set value more sensitive ∫ Large set value for strong vibration 49 Zero /Tare key Operation mode selection 0 Activate when condition, only ● “Steady” F07 1 Always activated Zero key selection 0 Activated within 2% of Max. Capacity 1 Activated within 5% of Max. Capacity F08 2 Activated within 10% of Max. Capacity 3 Activated within 20% of Max. Capacity ● 4 Activated within 100% of Max. Capacity Tare key selection 0 Activated within 10% of Max. Capacity 1 Activated within 20% of Max. Capacity F09 2 Activated within 50% of Max. Capacity 3 Activated within 100% of Max. Capacity ● Mode selection “Hold” 0 Peak Hold : Measure Max. weight and hold on display. ● 1 Sample Hold : Hold current weight until Reset”. “Hold F10 2 Average Hold : Make average during 3sec, and hold display 3 Average Hold : Make average during 5sec, and hold display 4 Average Hold : Make average during 8sec, and hold display
F04 5 0 ∫ 9 Due to external causes(Temperature, wind, and dust), there are small weight difference, indicator will ignore the weight difference and display Zero. For this compensation function, indicator will estimate the weight difference is over the set range during fixed time period. If there is large weight difference over set range within fixed time period, the “Zero” is breaking and will find new zero point. Auto Zero Range setting F05 00 00 ∫ 99 Within the “Auto Zero” weighing part is steady, indicator will display current weight as “Zero” If the weighing part is not “Steady”, indicator will display current weight. ( : ± Set + weight unit) Digital Filter setting F06 15 00 ∫ 49 set value for weak vibration Large set value for strong vibration set value more sensitive Zero /Tare key Operation mode selection F07 ● 0 Activate when “Steady” condition, only 1 Always activated Zero key selection F08 0 Activated within 2% of Max. Capacity 1 Activated within 5% of Max. Capacity 2 Activated within 10% of Max. Capacity ● 3 Activated within 20% of Max. Capacity 4 Activated within 100% of Max. Capacity Tare key selection F09 0 Activated within 10% of Max. Capacity 1 Activated within 20% of Max. Capacity 2 Activated within 50% of Max. Capacity ● 3 Activated within 100% of Max. Capacity “Hold” Mode selection F10 ● 0 Peak Hold : Measure Max. weight and hold on display. 1 Sample Hold : Hold current weight until “Hold Reset”. 2 Average Hold : Make average during 3sec, and hold display 3 Average Hold : Make average during 5sec, and hold display 4 Average Hold : Make average during 8sec, and hold display
CI-5100AENG-ver.docx