XSTAT 30 Indications For Use 30 is a temporary device for use up to four hours until surgical care is acquired. It should only be used for patients at high risk for immediate life-threatening XSTAT 30 is appropriate for wounds in the axilla, bleeding from hemodynamically significant, below the clavicles outside the rib cage, and in the groin junctional wounds when definitive care at distal to the inguinal ligaments when limb tourniquets, an emergency care facility cannot be achieved within junctional tourniquets, and hemostatic dressings cannot minutes. be effectively applied. XSTAT 30 is NOT indicated for use in: the thorax; the pleural cavity; the mediastinum; the abdomen; the retroperitoneal space; the sacral space above the inguinal ligament; or tissues above the clavicle. 30’s Technical Characteristics 30 is composed of compressed mini-sponges coated with chitosan a compound designed to stop – bleeding. Read the text. Upon contact with blood, the mini-sponges absorb blood and, expand to 10 - 12 times their compressed volume within approximately 20 seconds. A radiopaque marker is embedded into each of the to make them detectable by X-ray. XSTAT 30 Applicator Main body approximately 92 mini-sponges. The XSTAT 30 applicator is six inches long (10.5 inches with the handle extended) and 30 mm in Telescoping Handle diameter. Bifurcated silicone tip allows to exit.
XSTAT 30 Indications For Use 30 is a temporary device for use up to four hours until surgical care is acquired. It should only be used for patients at high risk for immediate life-threatening bleeding from hemodynamically significant, junctional wounds when definitive care at an emergency care facility cannot be achieved within minutes. XSTAT 30 is NOT indicated for use in: the thorax; the pleural cavity; the mediastinum; the abdomen; the retroperitoneal space; the sacral space above the inguinal ligament; or tissues above the clavicle. XSTAT 30 is appropriate for wounds in the axilla, below the clavicles outside the rib cage, and in the groin distal to the inguinal ligaments when limb tourniquets, junctional tourniquets, and hemostatic dressings cannot be effectively applied. 30’s Technical Characteristics 30 is composed of compressed mini-sponges coated with chitosan – a compound designed to stop bleeding. Upon contact with blood, the mini-sponges absorb blood and, expand to 10 - 12 times their compressed volume within approximately 20 seconds. A radiopaque marker is embedded into each of the to make them detectable by X-ray. Read the text. XSTAT 30 Applicator Main body approximately 92 mini-sponges. Telescoping Handle Bifurcated silicone tip allows to exit. The XSTAT 30 applicator is six inches long (10.5 inches with the handle extended) and 30 mm in diameter.
010303A%20TCCC-MP%20Tactical%20Field%20Care%20%231%20IG%20160603.docx