Switching main menus of measuring data: set. U→I, long press Left. I→U, long press Right. 9.2.1 Setting of Wire Switching secondary menus of measuring dat- Click Up/Down to select Wire, and Click Enter to set. a:U→UTH2, click Down. UTH2→U, click Up. Long press Left/Right to switch displacement, click Up/Down to change Menu display and quit: click Menu/Quit, value of figures, and Click Enter to set. display menu. Click Menu/Quit again, exit menu, and enter the interface of measuring select the wiring connect mod: “Mode” data. phase 4 wire system “3PH4W”:3 phase 3 wire system “3PH3W”:3 phase 2 wire L_L system ”1PH2W_LL”:1 9. Instructions of function and Setting phase 2 wire L_N system ”1PH2W_LN”1 Switching main menu: phase 2 wire L_L_N system ”1PH3W_LLN”1 Up/Down to select main menu, and select Direct voltage connect or VT “Vcon” Click Enter to go to secondary menu. conect: Please input voltage connect “DirectCon”:Directly Password to entry into secondary menu, Voltage sensor connect when Password of Set is 1000 and Password of “3VT”:3 3phase system Cal. need to apply. select Rogowski coil or CTs. 9.1 Secondary menu Select and set “Icon” :CT connection Secondary menu of Set or Cal. is as the “CTCon” following pictures: Click Up/Down to select coil connection “RoCon”:Rogowski third menu, Click Enter to enter third menu. Secondary menu of Cal. rated frequency. “Freq”select The calibrate setting only use on conditions: A.Change Rated Value B.Change ratio rogowski coil connection 9.2.2 Setting of Ratio Click Up/Down to select Ratio, and Click Enter to set. 9.2 Third menu of set Settings of Ct,Addr,Baud, Harmonic, Password, Time,Threshold: Long press Left/Right to switch displacement or items, click Up/Down to change value of figures, and Click Enter to If select RcoilCon,setting Rcoil Sec and
Switching main menus of measuring data: U→I, long press Left. I→U, long press Right. Switching secondary menus of measuring dat- a:U→UTH2, click Down. UTH2→U, click Up. Menu display and quit: click Menu/Quit, display menu. Click Menu/Quit again, exit menu, and enter the interface of measuring data. 9. Instructions of function and Setting Switching main menu: Up/Down to select main menu, and Click Enter to go to secondary menu. Please input Password to entry into secondary menu, Password of Set is 1000 and Password of Cal. need to apply. 9.1 Secondary menu Select and set Secondary menu of Set or Cal. is as the following pictures: Click Up/Down to select third menu, Click Enter to enter third menu. Secondary menu of Cal. The calibrate setting only use on conditions: A.Change Rated Value B.Change ratio rogowski coil connection 9.2 Third menu of set Settings of Ct,Addr,Baud, Harmonic, Password, Time,Threshold: Long press Left/Right to switch displacement or items, click Up/Down to change value of figures, and Click Enter to set. 9.2.1 Setting of Wire Click Up/Down to select Wire, and Click Enter to set. Long press Left/Right to switch displacement, click Up/Down to change value of figures, and Click Enter to set. “Mode” select the wiring connect mod: “3PH4W”:3 phase 4 wire system “3PH3W”:3 phase 3 wire system ”1PH2W_LL”:1 phase 2 wire L_L system ”1PH2W_LN”1 phase 2 wire L_N system ”1PH3W_LLN”1 phase 2 wire L_L_N system “Vcon” select Direct voltage connect or VT conect: “DirectCon”:Directly voltage connect “3VT”:3 Voltage sensor connect when 3phase system “Icon” select Rogowski coil or CTs. “CTCon” :CT connection “RoCon”:Rogowski coil connection “Freq”select rated frequency. 9.2.2 Setting of Ratio Click Up/Down to select Ratio, and Click Enter to set. If select RcoilCon,setting Rcoil Sec and
2017612115848.docx