-If computer recognizes the card normally, you can see the sign of the window as follows. The following figure shows the correct parameters for setting up a working PCL812 card for the current UTI MS computer. Then click the test button and you can see the following windows. In the input” tap, you can see the voltage input of all the channels. Give a “Analogue known voltage using a battery or a power supply, than you can check the voltage reading in that channel. Click the output” tap, then you can apply the “Analogue voltage to output channel 0 and output channel 1. 3) Copy the TPD program and RGA program file to the desktop 4) Copy the MassCalibration.txt to c:/ folder. For the current UTI MS computer, a Labview version of MS and TPD program is installed. To reinstall the software, install and files “LVRunTimeEng.exe” “Install” included in the installation files folder, then copy Mass Spectrometer” folder to “UTI the directory C:/Program Files. 3. The way going back to the former computer At the back panel of mass spectrometer controller, there is a D-connector (50 pin) port which was written as Take out the current cable and put the old connecter “AUX”. back. (It is easy to find. Old cable is connected to the Keithley Series 500 terminal) TPD program Ⅱ. 1. Click a TPD program icon.
-If computer recognizes the card normally, you can see the sign of the window as follows. The following figure shows the correct parameters for setting up a working PCL812 card for the current UTI MS computer. Then click the test button and you can see the following windows. In the “Analogue input” tap, you can see the voltage input of all the channels. Give a known voltage using a battery or a power supply, than you can check the voltage reading in that channel. Click the “Analogue output” tap, then you can apply the voltage to output channel 0 and output channel 1. 3) Copy the TPD program and RGA program file to the desktop 4) Copy the MassCalibration.txt to c:/ folder. For the current UTI MS computer, a Labview version of MS and TPD program is installed. To reinstall the software, install “LVRunTimeEng.exe” and “Install” files included in the installation files folder, then copy “UTI Mass Spectrometer” folder to the directory C:/Program Files. 3. The way going back to the former computer At the back panel of mass spectrometer controller, there is a D-connector (50 pin) port which was written as “AUX”. Take out the current cable and put the old connecter back. (It is easy to find. Old cable is connected to the Keithley Series 500 terminal) Ⅱ. TPD program 1. Click a TPD program icon.
uhv1_victor_2015-10.docx