Return to the main view. Select the channel 1’, right-click on it and select ‘TG186, ‘Properties’. The Properties’ window Copy all the source position parameters to the ‘Applicator Properties’ of the channel 2’ by selecting it from the list on the left. Note ‘Applicator ‘TG187, that when you switch between the applicator channels from the list on the left, the properties window also switches between the two obviating the need to close and re-open the properties window. Click the button and confirm that both applicators have the same points (as shown below in the Set Editor’ window). ‘Point
Return to the main view. Select the ‘TG186, channel 1’, right-click on it and select ‘Properties’. The ‘Applicator Properties’ window Copy all the source position parameters to the ‘Applicator Properties’ of the ‘TG187, channel 2’ by selecting it from the list on the left. Note that when you switch between the applicator channels from the list on the left, the properties window also switches between the two obviating the need to close and re-open the properties window. Click the button and confirm that both applicators have the same points (as shown below in the ‘Point Set Editor’ window).
User%20Guide%20for%20BV%20Testing.docx