Modify Units (not Pool or VBECS truncates the The short name of the None Provided. This is Split) KDA CR 1698 product name. product displays in full. The unchanged system full product name is behavior. available for display when the unit is selected and the user hovers over the name to view the tool tip. Modify Pool Units VBECS allows a user to Dual assignment is not None Provided. add a unit to a pool and created and the unit is KDA CR 2387 transfer assignment to the available for Issue Blood pooled unit when both units Component. are already assigned to the same patient. This creates a unit with dual assignment and the unit cannot be issued. Modify Units: Pool Units When a user chooses to Dual assignment is not None Provided. add or remove units from a created and the unit is KDA CR 2650 pooled unit, the patient available for Issue Blood assignment is duplicated Component. on the pool. No system error occurs. Leading to additional complications: Users are also prevented from selecting additional units for a patient that has a duplicate assignment until the duplicate is removed. A VBECS system error occurs when a user attempts to issue this unit with duplicate assignment. Modify Units: Split a Unit The system does not The system does not allow None Provided. split modification for any modifications on these KDA CR 3248 product types E065 = product types. This is not an Washed Thawed Apheresis anomaly. PLATELETS and E067 = Reconstituted FRESH FROZEN PLASMA Patient Testing, ABO/Rh Test Appendix B: Patient Appendix B, rows 12 and 14 Appendix A, Test ABO/Rh Test System row validation no Scenario Group Five CR 3277 Response only rows override is displayed and the DR 4592 incorrectly allowed invalid results and interpretations test result and cannot be saved. interpretation patterns to be See Appendix B: Patient saved for patient with a ABO/Rh Test System processed override. Response for all conditions. Patient Testing: Pending VBECS does not display A patient’s middle initial None Provided. Task List the patient’s middle initial displays. on the printed Pending KDA CR 1775 Task List.
Modify Units (not Pool or Split) KDA CR 1698 VBECS truncates the product name. The short name of the product displays in full. The full product name is available for display when the unit is selected and the user hovers over the name to view the tool tip. None Provided. This is unchanged system behavior. Modify Pool Units KDA CR 2387 VBECS allows a user to add a unit to a pool and transfer assignment to the pooled unit when both units are already assigned to the same patient. This creates a unit with dual assignment and the unit cannot be issued. Dual assignment is not created and the unit is available for Issue Blood Component. None Provided. Modify Units: Pool Units KDA CR 2650 When a user chooses to add or remove units from a pooled unit, the patient assignment is duplicated on the pool. Leading to additional complications: Users are also prevented from selecting additional units for a patient that has a duplicate assignment until the duplicate is removed. A VBECS system error occurs when a user attempts to issue this unit with duplicate assignment. Dual assignment is not created and the unit is available for Issue Blood Component. No system error occurs. None Provided. Modify Units: Split a Unit KDA CR 3248 The system does not split modification for product types E065 = Washed Thawed Apheresis PLATELETS and E067 = Reconstituted FRESH FROZEN PLASMA The system does not allow any modifications on these product types. This is not an anomaly. None Provided. Patient Testing, ABO/Rh Test CR 3277 DR 4592 Appendix B: Patient ABO/Rh Test System Response only rows incorrectly allowed invalid test result and interpretation patterns to be saved for patient with a processed override. Appendix B, rows 12 and 14 row validation no override is displayed and the results and interpretations cannot be saved. See Appendix B: Patient ABO/Rh Test System Response for all conditions. Appendix A, Test Scenario Group Five Patient Testing: Pending Task List KDA CR 1775 VBECS does not display the patient’s middle initial on the printed Pending Task List. A patient’s middle initial displays. None Provided.
vbecs_2_0_0_release_notes.docx