10: Troubleshooting Rejected VBECS HL7 Messages Error Code Description of Problem 100 Segment Sequence 101 Required Missing 102 Data Type Error 103 Table Value Not Found 200 Unsupported Message Type 201 Unsupported Event Code 202 Unsupported Processing ID Unsupported Version Id 203 See 11: VBECS HL7 Versions 204 Unknown Key Identifier 205 Duplicate Key Identifier 206 Application Record Locked 207 Application Internal Error 208 Conflicting Processing Id Table 11: VBECS HL7 Versions HL7 Interface HL7 Version VistA CPRS- Order Update CPRS OERR 2.4 – VistA PIMS Patient ADT Update VAFC ADT 2.3 – VistA MPI/PD PatientMerge MPI TRIGGER 2.4 – BCE COTS Patient Blood Product – Transfusion Verification 2.5 Table 12: Troubleshooting VBECS Application Interfaces Source Description of Problem Possible Cause Solution The Logical Link Start the OERR-VBECS is not running on the VistA Logical Link. system. VBECS: Order New orders or The VBECS <Prod or Test> Alerts and cancellations of existing HL7 Listener Windows Start or restart the VBECS Pending Order orders in CPRS are not Service is not running or is <Prod or Test> HL7 Listener List showing up in locked on the application Windows Service. server. Network connectivity issue Contact local system support.
10: Troubleshooting Rejected VBECS HL7 Messages Error Code Description of Problem 100 Segment Sequence 101 Required Missing 102 Data Type Error 103 Table Value Not Found 200 Unsupported Message Type 201 Unsupported Event Code 202 Unsupported Processing ID 203 Unsupported Version Id See 11: VBECS HL7 Versions 204 Unknown Key Identifier 205 Duplicate Key Identifier 206 Application Record Locked 207 Application Internal Error 208 Conflicting Processing Id Table 11: VBECS HL7 Versions HL7 Interface HL7 Version VistA CPRS- Order Update – CPRS OERR 2.4 VistA PIMS Patient ADT Update – VAFC ADT 2.3 VistA MPI/PD PatientMerge – MPI TRIGGER 2.4 BCE COTS – Patient Blood Product Transfusion Verification 2.5 Table 12: Troubleshooting VBECS Application Interfaces Source Description of Problem Possible Cause Solution VBECS: Order Alerts and Pending Order List New orders or cancellations of existing orders in CPRS are not showing up in The Logical Link is not running on the VistA system. Start the OERR-VBECS Logical Link. The VBECS <Prod or Test> HL7 Listener Windows Service is not running or is locked on the application server. Start or restart the VBECS <Prod or Test> HL7 Listener Windows Service. Network connectivity issue Contact local system support.
vbecs_2_1_0_tech_security_guide.docx