Issue Blood Component Missing a between the This message did not and None Provided sentences: unit had a does not appear in the “This KDA CR 2085 crossmatch result of updated application. A unit incompatible or the with an incompatible crossmatch was not crossmatch is not available completed. This unit cannot for selection in this option. be issued.” Blood Component When the patient’s ABO/Rh Override requirement is no None Provided. was performed in a longer displayed in Full KDA CR 2011 Transfusion Only (TO) Service division for patient division, the ABO/Rh was whose previous ABO/Rh not properly read by a Full was performed in Service division at issue. Transfusion Only division. Blood Component When a user has a unit The on the unit’s None Provided. “lock” selected for and record no longer allows CR 3471 another user selects and selection of the same unit at INC 1024778 attempts to release the the same time in Issue same blood unit from Blood Component and assignment; the unit Release Unit from Patient records are unavailable for Assignment. further processing. Maintain Specimen A VBECS system error No system error occurs after None Provided. occurs when a user allows system timeouts, single or KDA CR 875 two consecutive system multiple. timeouts. Modify Units (not Pool or When a user enters a Target Unit Volume value None Provided. Split) outside the allowable range returns to an acceptable in the Target Unit Volume value at the upper or lower KDA CR 1920 field; VBECS accepts the end prior to saving. entry but disables the OK button. Modify Units (not Pool or The Volume Reduce Unit The OK button enables as None Provided. Split) form contains fields for expected. discarded plasma volume KDA CR 1921 and original unit volume. When either field contains data entry errors, the OK button is enabled, which allows the user to save the form. Modify Units (not Pool or The Volume Reduce Unit The discarded plasma None Provided. Split) form contains fields for volume and original unit discarded plasma volume volume values return to an KDA CR 1924 and original unit volume. acceptable value at the When either field contains upper or lower end prior to data entry errors, the OK saving. button is enabled, which allows the user to save the form.
Issue Blood Component KDA CR 2085 Missing a between the sentences: “This unit had a crossmatch result of incompatible or the crossmatch was not completed. This unit cannot be issued.” This message did not and does not appear in the updated application. A unit with an incompatible crossmatch is not available for selection in this option. None Provided Blood Component KDA CR 2011 When the patient’s ABO/Rh was performed in a Transfusion Only (TO) division, the ABO/Rh was not properly read by a Full Service division at issue. Override requirement is no longer displayed in Full Service division for patient whose previous ABO/Rh was performed in Transfusion Only division. None Provided. Blood Component CR 3471 INC 1024778 When a user has a unit selected for and another user selects and attempts to release the same blood unit from assignment; the unit records are unavailable for further processing. The “lock” on the unit’s record no longer allows selection of the same unit at the same time in Issue Blood Component and Release Unit from Patient Assignment. None Provided. Maintain Specimen KDA CR 875 A VBECS system error occurs when a user allows two consecutive system timeouts. No system error occurs after system timeouts, single or multiple. None Provided. Modify Units (not Pool or Split) KDA CR 1920 When a user enters a outside the allowable range in the Target Unit Volume field; VBECS accepts the entry but disables the OK button. Target Unit Volume value returns to an acceptable value at the upper or lower end prior to saving. None Provided. Modify Units (not Pool or Split) KDA CR 1921 The Volume Reduce Unit form contains fields for discarded plasma volume and original unit volume. When either field contains data entry errors, the OK button is enabled, which allows the user to save the form. The OK button enables as expected. None Provided. Modify Units (not Pool or Split) KDA CR 1924 The Volume Reduce Unit form contains fields for discarded plasma volume and original unit volume. When either field contains data entry errors, the OK button is enabled, which allows the user to save the form. The discarded plasma volume and original unit volume values return to an acceptable value at the upper or lower end prior to saving. None Provided.
vbecs_2_0_0_release_notes.docx