P1000 F123602 200 ±3 0.6 ±8 3.0 ≤ ≤ 500 ±4 1.0 ±8 3.0 ≤ ≤ 1000 ±8 1.5 ±8 3.0 ≤ ≤ Control Shaking Ii Specification 01 Incubator Screen Master 3000, Iii 01 Screen Master 3000, MODEL REAGENTS USED Liquid, Biochemical Systems POWER NEEDED, VAC (ENERGY) 120, 200, Watts (GENERATOR) 50, DIMENSIONS (HXWXD) CM, (IN) (DISPLAY) 40 x 40 x 17 (15.7 x 15.7 x 6.7), WEIGHT, kg (lb) (DISPLAY) 13 (28.7), TECHNIQUE Photometry, Incubator Yes, CONFIGURATION Integral 25°C to 37°C, Tube capacity (TEST CAPABILITY) 10, SAMPLE REQUIREMENTS, Whole blood, plasma, serum, Size/range, Âµl (SAMPLE) From 5 Iv Refrigerated Eppendorf Microcentrifuge 5418 R (Refrigerated), With 01 Rooter Centrifuge includes aerosol-tight QuickLock FA-45-18-11 and lid, 230 V/50 Hz V Z338818 SIGMA ncludes base unit, safety lid, one standard casting tray for 01 13 × 15 cm gels, Midi horizontal and two combs (1.0 mm × 12 sample wells and 1.0 mm × electrophoresis unit 16 sample wells).
P1000 F123602 200 ±3 ≤ 0.6 ±8 ≤ 3.0 500 ±4 ≤ 1.0 ±8 ≤ 3.0 1000 ±8 ≤ 1.5 ±8 ≤ 3.0 Ii Control Shaking Incubator Specification 01 Iii Screen Master 3000, Biochemical Systems MODEL Screen Master 3000, REAGENTS USED Liquid, POWER NEEDED, VAC (ENERGY) 120, 200, Watts (GENERATOR) 50, DIMENSIONS (HXWXD) CM, (IN) (DISPLAY) 40 x 40 x 17 (15.7 x 15.7 x 6.7), WEIGHT, kg (lb) (DISPLAY) 13 (28.7), TECHNIQUE Photometry, Incubator Yes, CONFIGURATION Integral 25°C to 37°C, Tube capacity (TEST CAPABILITY) 10, SAMPLE REQUIREMENTS, Whole blood, plasma, serum, Size/range, Âµl (SAMPLE) From 5 01 Iv Refrigerated Centrifuge Eppendorf Microcentrifuge 5418 R (Refrigerated), With Rooter includes aerosol-tight QuickLock FA-45-18-11 and lid, 230 V/50 Hz 01 V Z338818 SIGMA Midi horizontal electrophoresis unit ncludes base unit, safety lid, one standard casting tray for 13 × 15 cm gels, and two combs (1.0 mm × 12 sample wells and 1.0 mm × 16 sample wells). 01
Tender%20Document%202016-17%20%20F.docx