08 Solvent Distillation Specifications should be: Assembly 5 Liters Comprise of Capacity: 5000 ml Round-bottom flasks with Socket(NS): 29/32 conical ground joint, Flask outer dia. (mm): 223 Effective length (mm): 600 Columns acc. to Vigreux with Cone and socket (NS): 29/32 fixed vacuum jacket, Range (°C): 0-360 cone 14/23 oC, Distillation thermometers Scale (°C): 1 Built-in length* (mm): 70 (NS): 29/32 Distilling links with socket, Socket (NS): 14/23 (For thermometer) Length (mm): 350 Spiral condensers, with Socket and cone (NS): 29/32 olives, Jacket length (mm): 400 Socket and cone (NS): 29/32 Receiver adapters, straight, with olive & vacuum connection, Capacity(ml): 3000. Round-bottom flasks with Socket(NS): 29/32 conical ground joint, 29/32 & 14/23 Fork clamps for ground joints (10 each) Capacity ml: 5000 Heating Power W: 860 Mantle Heating zone: 2 Flask Ø mm: 223 flasks, round bottom. For glass yarn heating element in plastic coated, Flexible casing with built-in mains- and heating zone switch. Max. heating element temperature 450 °C. 1.5 m power supply (earthed) cable and residual-current circuit-breaker. supply: 230 V AC. 09 Pigment free septa for reactions CHEMICALS:
08 Solvent Distillation Assembly 5 Liters Comprise of Round-bottom flasks with conical ground joint, Columns acc. to Vigreux with fixed vacuum jacket, Distillation thermometers Distilling links with socket, (For thermometer) Spiral condensers, with olives, Receiver adapters, straight, with olive & vacuum connection, Round-bottom flasks with conical ground joint, Fork clamps for ground joints (10 each) Heating Mantle Specifications should be: Capacity: 5000 ml Socket(NS): 29/32 Flask outer dia. (mm): 223 Effective length (mm): 600 Cone and socket (NS): 29/32 Range (°C): 0-360 oC, cone 14/23 Scale (°C): 1 Built-in length* (mm): 70 (NS): 29/32 Socket (NS): 14/23 Length (mm): 350 Socket and cone (NS): 29/32 Jacket length (mm): 400 Socket and cone (NS): 29/32 Capacity(ml): 3000. Socket(NS): 29/32 29/32 & 14/23 Capacity ml: 5000 Power W: 860 Heating zone: 2 Flask Ø mm: 223 For flasks, round bottom. Flexible glass yarn heating element in plastic coated, casing with built-in mains- and heating zone switch. Max. heating element temperature 450 °C. 1.5 m power supply (earthed) cable and residual-current circuit-breaker. supply: 230 V AC. 09 Pigment free septa for reactions CHEMICALS:
tender%20documents%20of%20chemistry17%20Final.docx