Analolg output (standard): 4 bis 20 mA, aktiv / passiv, selectable Digital output (standard): 2 Can be configured as pulse, frequency or binary output. – Multiple output options, ability to have 3 current outputs Free selectable I/O Selectable/Retrofitable I/O cards: Digital input (DRN), Digital output (DRG), Analog output passive 4 ... 20 mA) (DRA), 24 V DC transmitter loop power supply (DRT), Selectable I/O card Slot 1 Approval – Digital input [DRN] Digital output [DRG] Analog output passive (4 ... 20 mA) [DRA] 24 V DC transmitter loop power supply [DRT] Selectable I/O card Slot 2 Approval – Digital [DRN] Digital output [DRG] Analog output passive (4 ... 20 mA) [DRA] Communication: HART-Protocol Serviceport: Infrared adapter FZA100 Measuring medium temperature: -25 150 °C [-13 302 °F] … … -25 300 °C [-13 572 °F] … … Ambient temperature: -20 +70°C [-4 158 °F] … … -40 +70°C [-40 158 °F] … … -50 +70 °C [-58 158 °F] … … Power supply: 11 30VDC … 110 230VAC … Protection class: IP65/67/NEMA 4X Diagnostic: Acc. Namur NE107, Status messages including helptext available on display and by digital communication Verification: Verification (VeriMass or equivalent) SensyMaster FMT450 or equivalent Device name: Producer: ABB Automation Products GmbH Additional technical information: Internet
Analolg output (standard): 4 bis 20 mA, aktiv / passiv, selectable Digital output (standard): 2 – Can be configured as pulse, frequency or binary output. Free selectable I/O Multiple output options, ability to have 3 current outputs Selectable/Retrofitable I/O cards: Digital input (DRN), Digital output (DRG), Analog output passive 4 ... 20 mA) (DRA), 24 V DC transmitter loop power supply (DRT), Selectable I/O card – Slot 1 Approval Digital input [DRN] Digital output [DRG] Analog output passive (4 ... 20 mA) [DRA] 24 V DC transmitter loop power supply [DRT] Selectable I/O card – Slot 2 Approval Digital [DRN] Digital output [DRG] Analog output passive (4 ... 20 mA) [DRA] Communication: HART-Protocol Serviceport: Infrared adapter FZA100 Measuring medium temperature: -25 … 150 °C [-13 … 302 °F] -25 … 300 °C [-13 … 572 °F] Ambient temperature: -20 … +70°C [-4 … 158 °F] -40 … +70°C [-40 … 158 °F] -50 … +70 °C [-58 … 158 °F] Power supply: 11 … 30VDC 110 … 230VAC Protection class: IP65/67/NEMA 4X Diagnostic: Acc. Namur NE107, Status messages including helptext available on display and by digital communication Verification: Verification (VeriMass or equivalent) Device name: SensyMaster FMT450 or equivalent Producer: ABB Automation Products GmbH Additional technical information: Internet
TES_FMT450_101-EN_A.docx