British OK No update No update No update No update Jansen Penn Penn Jason Penn Virgin informed that reported that the reported that Islands the BVI Tide tide gauge was Tide Gauge is Gauge is logging but not working fine the functioning as it sending a data 19th and 20th was off for a for a few days to displayed some while due to a a week a few spikes however faulty Satlink months ago but will monitor and which was the link was see if this replaced. repaired. A rain continues. Their gauge was added plan is to do to that site so some now they have a maintenance on weather the Tide Gauge “full station”. They in March if not have some issues then the second with the quarter. programming of the rainfall collection. They are requesting assistance with this matter. Colombia Problems No update Gabriel Herrera Wilder Alvarez Problems Marta Gabriel Herrera – decoding the data DIMAR: they report that decoding, they station needs to Díaz, at OTT. need to review DIMAR stations will contact change the DIMAR, reported the transmission are exporting directly the batteries already that they made date through vendor OTT. sent by UHSLC the second formats for the FTP. They are They are asking if and received by maintainance for data of new SLSs confronting is possible to DIMAR. In July the Caribbean problems change the DIMAR will be stations (San decodifying using platform system working on visits Andrés y Santa GOES. Sebastien or need to of the stations Marta). Pressure suggested try contact UHSLC to for maintenance sensor of St. with OTT to get try to resolve the including Santa Marta station better results software Marta. UHSLC will be replaced with the data. problems. Santa will be visiting by Jason Klem
British Virgin Islands OK No update No update No update No update Jansen Penn informed that the BVI Tide Gauge is functioning as it was off for a while due to a faulty Satlink which was replaced. Penn reported that the tide gauge was logging but not sending a data for a few days to a week a few months ago but the link was repaired. A rain gauge was added to that site so now they have a “full weather station”. They have some issues with the programming of the rainfall collection. They are requesting assistance with this matter. Jason Penn reported that Tide Gauge is working fine the 19th and 20th displayed some spikes however will monitor and see if this continues. Their plan is to do some maintenance on the Tide Gauge in March if not then the second quarter. Colombia Problems decoding the data at OTT. No update Gabriel Herrera – DIMAR: they need to review the transmission formats for the data of new SLSs Wilder Alvarez report that DIMAR stations are exporting date through FTP. They are confronting problems decodifying using GOES. Sebastien suggested try with OTT to get better results with the data. Problems decoding, they will contact directly the vendor OTT. They are asking if is possible to change the platform system or need to contact UHSLC to try to resolve the software problems. Santa Marta station needs to change the batteries already sent by UHSLC and received by DIMAR. In July DIMAR will be working on visits of the stations for maintenance including Santa Marta. UHSLC will be visiting Gabriel Herrera Díaz, DIMAR, reported that they made the second maintainance for the Caribbean stations (San Andrés y Santa Marta). Pressure sensor of St. Marta station will be replaced by Jason Klem
Sea%20Level%20Station%202017-01-31Meeting_V2.docx