3. Issues reading cold cathode pressure and state, so Sample Transfer procedure will perform Intro Pumpdown when cold cathode pressure readback is questionable. 4. Add delay when switching 20kVC60 mode from Off -> Standby to avoid issue of target temperature not being set correctly while controller is turning on high voltage. 5. Fix C60 pressure display on vacuum glyph. It was not updating the C60 fields with the correct pressure readback. 6. Fix blanking status in 20kV C60 diagnostics menu, change of logic. 7. Fix 20kVC60 Raster Calibration values in Properties view not being set correctly. 8. Change default Max Stage Z to 20 mm. 9. When running acquisition with UPS Neutralization to be On. 10. Disable meters 2&3 in diagnostics. Hardware supports 3 meters, but is only using 1 currently. 11. Fix issue where bend voltage is clipped to -350 when profile setting is loaded. 12. Fix bug where Read Beam Power would fail to move anode and Beam Power button is then stuck. 13. Fix bug positions could be lost on SmartSoft crash or force quit. 14. Fix issue with SXI gamma value having incorrect default value and not reading/writing from/to property file correctly. HELPFUL HINTS 1. Entry fields must be 'terminated' either by selecting the <Enter> key or by moving to another parameter field. If the application is exited before the value is terminated the change will not be saved.
3. Issues reading cold cathode pressure and state, so Sample Transfer procedure will perform Intro Pumpdown when cold cathode pressure readback is questionable. 4. Add delay when switching 20kVC60 mode from Off -> Standby to avoid issue of target temperature not being set correctly while controller is turning on high voltage. 5. Fix C60 pressure display on vacuum glyph. It was not updating the C60 fields with the correct pressure readback. 6. Fix blanking status in 20kV C60 diagnostics menu, change of logic. 7. Fix 20kVC60 Raster Calibration values in Properties view not being set correctly. 8. Change default Max Stage Z to 20 mm. 9. When running acquisition with UPS Neutralization to be On. 10. Disable meters 2&3 in diagnostics. Hardware supports 3 meters, but is only using 1 currently. 11. Fix issue where bend voltage is clipped to -350 when profile setting is loaded. 12. Fix bug where Read Beam Power would fail to move anode and Beam Power button is then stuck. 13. Fix bug positions could be lost on SmartSoft crash or force quit. 14. Fix issue with SXI gamma value having incorrect default value and not reading/writing from/to property file correctly. HELPFUL HINTS 1. Entry fields must be 'terminated' either by selecting the <Enter> key or by moving to another parameter field. If the application is exited before the value is terminated the change will not be saved.
readme-vpv2.8.0.docx