remaining motionless, agitation in subjects with (0010,4000 □ Not feasible decreased levels of consciousness, subjects with chronic pain syndromes, etc.). Data Collection □ Routinely do already Acquisition Diameter □ Feasible, will do Field of View Shall set Consistent with baseline. (0018,0090) □ Feasible, will not do (FOV) □ Not feasible Image Data Reconstruction (section 3.7) Shall select a protocol that has been previously □ Routinely do already prepared and validated for this purpose (See □ Feasible, will do Reconstruction section 3.4.2 "Protocol Design Specification"). □ Feasible, will not do Protocol Shall report if any parameters are modified beyond □ Not feasible those specifications. Shall either □ Routinely do already □ Feasible, will do select the same protocol as used for the □ Feasible, will not do baseline scan, or In-plane □ Not feasible select a protocol with a recorded f50 value Spatial within 0.2 mm-1 of the f50 value recorded Resolution for the baseline scan protocol. See section 3.4.2 for further details. Shall either □ Routinely do already □ Feasible, will do select the same protocol as used for the □ Feasible, will not do baseline scan, or □ Not feasible select a protocol with a recorded standard Voxel Noise deviation within 5HU of the standard deviation recorded for the baseline scan protocol. See section 3.4.2 for further details. □ Routinely do already Reconstructed Shall set to between 0.5mm and 2.5mm (inclusive) □ Feasible, will do Image if not set in the protocol. □ Feasible, will not do Thickness □ Not feasible □ Routinely do already Shall set to less than or equal to the Reconstructed Reconstructed □ Feasible, will do Image Thickness (i.e. no gap, may have overlap) Image Interval □ Feasible, will not do and consistent with baseline. □ Not feasible Shall set the reconstruction kernel and parameters Convolution Kernel □ Routinely do already consistent with baseline (i.e. the same kernel and Group (0018,9316), □ Feasible, will do Reconstruction parameters if available, otherwise the kernel most Convolution Kernel □ Feasible, will not do Characteristics closely matching the kernel response of the (0018,1210) □ Not feasible baseline). Reconstruction Shall ensure the Field of View spans at least the Reconstruction □ Routinely do already Field of View full extent of the thoracic and abdominal cavity, Field of View □ Feasible, will do
remaining motionless, agitation in subjects with decreased levels of consciousness, subjects with chronic pain syndromes, etc.). (0010,4000 □ Not feasible Acquisition Field of View (FOV) Shall set Consistent with baseline. Data Collection Diameter (0018,0090) □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Image Data Reconstruction (section 3.7) Reconstruction Protocol Shall select a protocol that has been previously prepared and validated for this purpose (See section 3.4.2 "Protocol Design Specification"). Shall report if any parameters are modified beyond those specifications. □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible In-plane Spatial Resolution Shall either select the same protocol as used for the baseline scan, or select a protocol with a recorded f50 value within 0.2 mm-1 of the f50 value recorded for the baseline scan protocol. See section 3.4.2 for further details. □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Voxel Noise Shall either select the same protocol as used for the baseline scan, or select a protocol with a recorded standard deviation within 5HU of the standard deviation recorded for the baseline scan protocol. See section 3.4.2 for further details. □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Reconstructed Image Thickness Shall set to between 0.5mm and 2.5mm (inclusive) if not set in the protocol. □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Reconstructed Image Interval Shall set to less than or equal to the Reconstructed Image Thickness (i.e. no gap, may have overlap) and consistent with baseline. □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Reconstruction Characteristics Shall set the reconstruction kernel and parameters consistent with baseline (i.e. the same kernel and parameters if available, otherwise the kernel most closely matching the kernel response of the baseline). Convolution Kernel Group (0018,9316), Convolution Kernel (0018,1210) □ Routinely do already □ Feasible, will do □ Feasible, will not do □ Not feasible Reconstruction Field of View Shall ensure the Field of View spans at least the full extent of the thoracic and abdominal cavity, Reconstruction Field of View □ Routinely do already □ Feasible, will do
QIBA_Profile_Template-2017.07.26.docx