Optical Interface Connector : FC/PC 6.3.4 Optical Power Meter The proposed light source shall be of JDSU OLP-34 model or equivalent model of world renowned North American/European Union/Japanese brand. The proposed equipment shall satisfy following minimum requirements:- Detector type :Germanium/InGaAs Wavelength Range :800 to 1600 nm Calibrated wavelengths :850, 980, 1300, 1310, 1490 & 1550 nm Power Range : -60 to +5 dBM Display Range :-60 to +10 dBm Maximum input power :+10 dBm or more Measurement units :dB/ dBm/ W Absolute Uncertainty : ± 0.2 dB Linearity : ± 0.06 dB (-50 to +5 dBm) Optical Interface/ Connector: Universal 2.5 mm/1.25 mm 6.3.5 E1 Tester JDSU SmartClass E1 or equivalent or better model of other renowned brand from EU/North America or Japan Required Features Both 120 ohm Balanced and 75 ohm unbalanced interfaces for E1 measurement Performance measurement, G.821 and G.826 Power: Battery and A/C line with adapter Line code: AMI, HDB3 Framing: Unframed, PCM30, PCM31 Clock: Internal, Recovered 6.3.6 Special Tool Kit Special tools packed in a transportable tool case for jointing the optical fiber cable shall be provided. Detailed list of tools shall be submitted with the bid include the following items as a minimum:- Transit Case. Measuring Tape. Fiber Stripper (250μ). Buffer Tube Stripper (250μ). Automatic Screwdriver (Slotted and Star). Pliers and Cutter.
Optical Interface Connector : FC/PC 6.3.4 Optical Power Meter The proposed light source shall be of JDSU OLP-34 model or equivalent model of world renowned North American/European Union/Japanese brand. The proposed equipment shall satisfy following minimum requirements:- Detector type :Germanium/InGaAs Wavelength Range :800 to 1600 nm Calibrated wavelengths :850, 980, 1300, 1310, 1490 & 1550 nm Power Range : -60 to +5 dBM Display Range :-60 to +10 dBm Maximum input power :+10 dBm or more Measurement units :dB/ dBm/ W Absolute Uncertainty : ± 0.2 dB Linearity : ± 0.06 dB (-50 to +5 dBm) Optical Interface/ Connector: Universal 2.5 mm/1.25 mm 6.3.5 E1 Tester JDSU SmartClass E1 or equivalent or better model of other renowned brand from EU/North America or Japan Required Features Both 120 ohm Balanced and 75 ohm unbalanced interfaces for E1 measurement Performance measurement, G.821 and G.826 Power: Battery and A/C line with adapter Line code: AMI, HDB3 Framing: Unframed, PCM30, PCM31 Clock: Internal, Recovered 6.3.6 Special Tool Kit Special tools packed in a transportable tool case for jointing the optical fiber cable shall be provided. Detailed list of tools shall be submitted with the bid include the following items as a minimum:- Transit Case. Measuring Tape. Fiber Stripper (250μ). Buffer Tube Stripper (250μ). Automatic Screwdriver (Slotted and Star). Pliers and Cutter.
14423140427969__1000_UP_Package3_MSPP_Book-2_02122014.docx