of the App to the clinician so s/he can use it accordingly. The ICE system shall be Device alarms, 6.4.9 CR_0000000039 able to display all the alarm audio and visual settings from each device. Device may be The ICE system shall be capable of sending aware of the required data at various frequency / accuracy / be aware of the frequencies and reliability of the incoming required data for each parameter settings. The ICE frequency / based on clinical 6.4.10 CR_0000000081 system shall be accuracy / significance, and shall able to help reliability of the choose the closest available clinician decide frequency / accuracy / incoming data what frequency is reliability on the device and provide this information to most useful for a the clinician for review. given scenario. This helps the clinician monitor the patient's The ICE System shall status continuously, Fixed patient display basic patient and ensure the correct information (patient name, 6.4.11 CR_0000000089 information patient is being MRN, vital signs, allergies, display monitored, irrespective etc.) at all times on the of what other screen is screen. being displayed at any given time. When there are multiple There are different sources of data, the ICE levels of device detail - system shall provide the clinician should be able clinician with an option of to choose level based choosing which device or on specific use case to 6.4.12 CR_0000000090 Data sources devices s/he would like to prevent information use as the source for each overload and avoid data element for each inaccurate data being application. used for decision support. The ICE system shall warn the clinician when s/he Required turns off a set of 'required 6.4.13 CR_0000000091 parameter parameters' from any warning device that contributes to "necessary data" The Infusion Patient Management App shall monitoring provide a simple, 6.4.14 CR_0000000216 through IV drug convenient process for infusions. clinicians to set up and monitor patients
of the App to the clinician so s/he can use it accordingly. 6.4.9 CR_0000000039 Device alarms, audio and visual The ICE system shall be able to display all the alarm settings from each device. 6.4.10 CR_0000000081 be aware of the required frequency / accuracy / reliability of the incoming data The ICE system shall be aware of the required frequency / accuracy / reliability of the incoming data for each parameter based on clinical significance, and shall choose the closest available frequency / accuracy / reliability on the device and provide this information to the clinician for review. Device may be capable of sending data at various frequencies and settings. The ICE system shall be able to help clinician decide what frequency is most useful for a given scenario. 6.4.11 CR_0000000089 Fixed patient information display The ICE System shall display basic patient information (patient name, MRN, vital signs, allergies, etc.) at all times on the screen. This helps the clinician monitor the patient's status continuously, and ensure the correct patient is being monitored, irrespective of what other screen is being displayed at any given time. 6.4.12 CR_0000000090 Data sources When there are multiple sources of data, the ICE system shall provide the clinician with an option of choosing which device or devices s/he would like to use as the source for each data element for each application. There are different levels of device detail - clinician should be able to choose level based on specific use case to prevent information overload and avoid inaccurate data being used for decision support. 6.4.13 CR_0000000091 Required parameter warning The ICE system shall warn the clinician when s/he turns off a set of 'required parameters' from any device that contributes to "necessary data" 6.4.14 CR_0000000216 Patient monitoring through IV drug infusions. The Infusion Management App shall provide a simple, convenient process for clinicians to set up and monitor patients
ICE-PCA-MSIS-Mock-Vendor-Submission-v0.11.docx