Power supply control any thro’ RS485/Ethernet/LXI/ other. Vendor may give available options and effect on the performance in system Heater harness with connectors Total heater extension wire shall be about 20m length. Due to test set-up constraints the extension wire is divided into two parts (15m & 5m). Vendor shall provide these with 50-pins D-sub connectors and extension wire length. Harness bunch shall be suitably tied. Extension of 15m: This shall be extension of temperature sensor from power supply to intermediate 50-pins female D-sub connectors. Connection of this harness with the Power Supply unit shall be the responsibility of the vendor. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Extension of 5m: This shall be extension of heater wire from heater element to intermediate 50-pins male D-sub connectors. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Heater element: SAC shall supply heater element during installation at SAC. Vendor should provide 50-pins female D-sub connector extension with harness length of 15m. 50-pins male D-sub connector extension with harness length of 5m. 50-pins Male and Female Vacuum Compatible/Space grade D-sub Crimpable connectors Make: Standard Make like Amphenol, Positronics, etc. The wires, connectors, sensors and crimping should be compatible in vacuum chambers (<1x10-5torr). All extension cable shall be flexible extension wire, silver plated, multi-stranded copper conductors, individually PTFE insulated, and Twisted. Typical Heater Harness Diagram: Server Server shall be used for controlling and monitoring of real time test data, storage etc. Processor: Intel Xeon Processor E5-2637 V3 Hard Disk and Controller: 1.2TB SAS HDD 10K Qty: 3 Nos. The internal storage should configure in RAID 5.
Power supply control thro’ RS485/Ethernet/LXI/ any other. Vendor may give available options and effect on the performance in system Heater harness with connectors Total heater extension wire shall be about 20m length. Due to test set-up constraints the extension wire is divided into two parts (15m & 5m). Vendor shall provide these with 50-pins D-sub connectors and extension wire length. Harness bunch shall be suitably tied. Extension of 15m: This shall be extension of temperature sensor from power supply to intermediate 50-pins female D-sub connectors. Connection of this harness with the Power Supply unit shall be the responsibility of the vendor. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Extension of 5m: This shall be extension of heater wire from heater element to intermediate 50-pins male D-sub connectors. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Heater element: SAC shall supply heater element during installation at SAC. Vendor should provide 50-pins female D-sub connector extension with harness length of 15m. 50-pins male D-sub connector extension with harness length of 5m. 50-pins Male and Female Vacuum Compatible/Space grade D-sub Crimpable connectors Make: Standard Make like Amphenol, Positronics, etc. The wires, connectors, sensors and crimping should be compatible in vacuum chambers (<1x10-5torr). All extension cable shall be flexible extension wire, silver plated, multi-stranded copper conductors, individually PTFE insulated, and Twisted. Typical Heater Harness Diagram: Server Server shall be used for controlling and monitoring of real time test data, storage etc. Processor: Intel Xeon Processor E5-2637 V3 Hard Disk and Controller: 1.2TB SAS HDD 10K Qty: 3 Nos. The internal storage should configure in RAID 5.
IDT00834000000000000isro05401.docx