If calibration fails, the message “Cal Failed” is displayed. Check the integrity of the circuit, connections, and test lung, and recalibrate When calibration is finished, remove the test lung and press Enter or Cancel to exit Ventilation Set-up To turn on the ventilator: Press the On/Off button for 3 sec To adjust control values: Select the parameters by taping the relevant control button Adjust the numeric value using the Up/Down button Accept the value by taping the relevant control button again, or by pressing Enter. New value will be accepted if another control button is taped or after 5 seconds with no activity To cancel parameter adjustment: Press the CANCEL button within 5 seconds To turn off the ventilator: Press the On/Off button and then press and hold the On/Off button again for 3 seconds Control Parameters Ptrig (sensitivity) (-9.9 to 0 cmH20) - The Ptrig LED indicator illuminates each time the airway pressure reaches the set Ptrig level. Ptrig value is adjusted using the Up/Down button PEEP / CPAP (3 to 30 cmH20) – Sets the baseline pressure during the exhalation phase. In Pressure Control ventilation PEEP is limited to no more than 5 cmH2O below the Pressure Control setting. PEEP value is adjusted using the Up/Down button
If calibration fails, the message “Cal Failed” is displayed. Check the integrity of the circuit, connections, and test lung, and recalibrate When calibration is finished, remove the test lung and press Enter or Cancel to exit Ventilation Set-up To turn on the ventilator: Press the On/Off button for 3 sec To adjust control values: Select the parameters by taping the relevant control button Adjust the numeric value using the Up/Down button Accept the value by taping the relevant control button again, or by pressing Enter. New value will be accepted if another control button is taped or after 5 seconds with no activity To cancel parameter adjustment: Press the CANCEL button within 5 seconds To turn off the ventilator: Press the On/Off button and then press and hold the On/Off button again for 3 seconds Control Parameters Ptrig (sensitivity) (-9.9 to 0 cmH20) - The Ptrig LED indicator illuminates each time the airway pressure reaches the set Ptrig level. Ptrig value is adjusted using the Up/Down button PEEP / CPAP (3 to 30 cmH20) – Sets the baseline pressure during the exhalation phase. In Pressure Control ventilation PEEP is limited to no more than 5 cmH2O below the Pressure Control setting. PEEP value is adjusted using the Up/Down button
Flight-60-user-kit_v3.docx