Ongoing derived Beginning and end of Project Geophysicist/ Library Match to initial RCA/CA polarizabilities precision each day as part of IVS tracking summary/QC polarizabilities metric (IVS) testing Geophysicist ≥ 0.9 for each set of three inverted polarizabilities Ongoing derived target Beginning and end of Project Geophysicist/ All IVS items fit RCA/CA position precision (IVS) each day as part of IVS tracking summary/QC locations within 0.25m testing Geophysicist of average of derived fit locations Initial measurement of Once per background Field Team Leader/ All decay amplitudes CA: reject BG location and find production area location background location lower than project alternate background locations report/Project threshold (five background Geophysicist measurements: one centered at the flag and one offset at least 35cm in each cardinal direction) Ongoing production area Background data Field Team All decay amplitudes CA: BG measurement rejected background collected a minimum Leader/failures noted lower than project and recollected measurements of every two hours in field log and tracking threshold during production summary/Project Geophysicist Ongoing instrument Each time instrument Field Team Response (mean static CA: make necessary repairs and function test is restarted Leader/tracking spike minus mean re-verify (TEMTADS) summary/Project static background) Geophysicist within 20% of predicted response for all monostatic Tx/Rx combinations
Ongoing derived polarizabilities precision (IVS) Beginning and end of each day as part of IVS testing Project Geophysicist/ tracking summary/QC Geophysicist Library Match to initial polarizabilities metric ≥ 0.9 for each set of three inverted polarizabilities RCA/CA Ongoing derived target position precision (IVS) Beginning and end of each day as part of IVS testing Project Geophysicist/ tracking summary/QC Geophysicist All IVS items fit locations within 0.25m of average of derived fit locations RCA/CA Initial measurement of production area background locations (five background measurements: one centered at the flag and one offset at least 35cm in each cardinal direction) Once per background location Field Team Leader/ background location report/Project Geophysicist All decay amplitudes lower than project threshold CA: reject BG location and find alternate Ongoing production area background measurements Background data collected a minimum of every two hours during production Field Team Leader/failures noted in field log and tracking summary/Project Geophysicist All decay amplitudes lower than project threshold CA: BG measurement rejected and recollected Ongoing instrument function test (TEMTADS) Each time instrument is restarted Field Team Leader/tracking summary/Project Geophysicist Response (mean static spike minus mean static background) within 20% of predicted response for all monostatic Tx/Rx combinations CA: make necessary repairs and re-verify
final_draft_gcmr-qapp_121515_worksheets.docx