The tests shall be carried out in a room that has an air speed close to the product under test of ≤0,5 m/s, according to the standard EN 50564:2011; 4.2 Test room, or equivalent. The ambient temperature shall be maintained at (23 ± 2) °C throughout the test. The power measurement device shall be calibrated with traceability document, i.e. a document which describes the method of calibration which shows that the measuring device is calibrated according to prevailing standards and that the calibration can be traced. Customised use scenario, test conditions The power measurement device shall be calibrated with traceability document, i.e. a document which describes the method of calibration which shows that the measuring device is calibrated according to prevailing standards and that the calibration can be traced. Other conditions During the tests the temperature of incoming water shall be 15 degrees according to EN 285:2006 or equivalent. The sterilization/ disinfection result shall comply with prevailing standards. Appendix 5 Computed tomography
The tests shall be carried out in a room that has an air speed close to the product under test of ≤0,5 m/s, according to the standard EN 50564:2011; 4.2 Test room, or equivalent. The ambient temperature shall be maintained at (23 ± 2) °C throughout the test. The power measurement device shall be calibrated with traceability document, i.e. a document which describes the method of calibration which shows that the measuring device is calibrated according to prevailing standards and that the calibration can be traced. Customised use scenario, test conditions The power measurement device shall be calibrated with traceability document, i.e. a document which describes the method of calibration which shows that the measuring device is calibrated according to prevailing standards and that the calibration can be traced. Other conditions During the tests the temperature of incoming water shall be 15 degrees according to EN 285:2006 or equivalent. The sterilization/ disinfection result shall comply with prevailing standards. Appendix 5 Computed tomography
Final_Draft_EU_GPP_Criteria_for_Health_Care_EEE_for_GPP_AG_final.docx