Initial Instrument Once following Field Geophysicist/ Response (mean static CA: Make necessary Function Test assembly Initial IVS spike minus mean adjustments, and re-verify (MetalMapper) Memorandum/ Project static background) Geophysicist within 20% of predicted response for all monostatic Tx/Rx combinations Initial Instrument Once following Field Geophysicist/ Response (mean static CA: Make necessary Function Test assembly Initial IVS spike minus mean adjustments, and re-verify (EM61) Memorandum/ Project static background) Geophysicist within 20% of predicted response for all channels Initial dynamic Once prior to start of Project Geophysicist/ Derived positions of CA: Make necessary positioning accuracy dynamic data IVS Memorandum/QC IVS target(s) are adjustments, and re-verify (IVS) acquisition Geophysicist within 25cm of the ground truth locations Ongoing Instrument Beginning and end of Field Team Leader/ Response (mean static CA: Make necessary repairs and Function Test each day and each running QC summary spike minus mean re-verify (Instrument response time instrument is (Excel/Geosoft) static background) amplitudes) turned on /Project or QC within 20% of (TEMTADS) Geophysicist predicted response for all monostatic Tx/Rx combinations Ongoing Instrument Beginning and end of Field Team Leader/ Response (mean static CA: Make necessary repairs and Function Test each day and each running QC spike minus mean re-verify (MetalMapper) time instrument is summary/Project or static background) turned on QC Geophysicist within 20% of predicted response for all monostatic Tx/Rx combinations
Initial Instrument Function Test (MetalMapper) Once following assembly Field Geophysicist/ Initial IVS Memorandum/ Project Geophysicist Response (mean static spike minus mean static background) within 20% of predicted response for all monostatic Tx/Rx combinations CA: Make necessary adjustments, and re-verify Initial Instrument Function Test (EM61) Once following assembly Field Geophysicist/ Initial IVS Memorandum/ Project Geophysicist Response (mean static spike minus mean static background) within 20% of predicted response for all channels CA: Make necessary adjustments, and re-verify Initial dynamic positioning accuracy (IVS) Once prior to start of dynamic data acquisition Project Geophysicist/ IVS Memorandum/QC Geophysicist Derived positions of IVS target(s) are within 25cm of the ground truth locations CA: Make necessary adjustments, and re-verify Ongoing Instrument Function Test (Instrument response amplitudes) (TEMTADS) Beginning and end of each day and each time instrument is turned on Field Team Leader/ running QC summary (Excel/Geosoft) /Project or QC Geophysicist Response (mean static spike minus mean static background) within 20% of predicted response for all monostatic Tx/Rx combinations CA: Make necessary repairs and re-verify Ongoing Instrument Function Test (MetalMapper) Beginning and end of each day and each time instrument is turned on Field Team Leader/ running QC summary/Project or QC Geophysicist Response (mean static spike minus mean static background) within 20% of predicted response for all monostatic Tx/Rx combinations CA: Make necessary repairs and re-verify
final_draft_gcmr-qapp_121515_worksheets.docx