France/LNE-LADG Expanded Relative Expanded uncertainty of Flow of error of uncertainty of measurement the the measurement extended by transfer di/% Eni transfer declared by stability UTS standard standard laboratory and linking U(D) Q/(m3/h) x/(%) Uxi /(%) / (%) 0.267 0.046 0.17 13.07 0.15 0.25 0.267 0.035 0.13 16.09 0.18 0.25 0.267 0.021 0.08 24.35 0.19 0.25 0.267 0.034 0.12 32.60 0.21 0.25 0.267 0.042 0.16 40.78 0.23 0.25 0.267 0.029 0.11 51.18 0.24 0.25 0.267 0.016 0.06 60.82 0.24 0.25 0.267 0.004 0.01 70.87 0.25 0.25 0.267 -0.010 0.04 80.89 0.25 0.25 0.274 0.003 0.01 90.92 0.26 0.25 0.274 0.003 0.01 100.81 0.26 0.25 Netherlands/VSL Expanded Relative Expanded uncertainty of Flow of error of uncertainty of measurement the the measurement extended by transfer di/% Eni transfer declared by stability UTS standard standard laboratory and linking U(D) Q/(m3/h) x/(%) Uxi /(%) / (%) 0.094 0.134 0.006 0.04 2.00 -0.09 0.094 0.134 0.105 0.66 4.50 0.07 0.094 0.134 0.084 0.53 6.60 0.11
France/LNE-LADG Flow of the transfer standard Q/(m3/h) Relative error of the transfer standard x/(%) Expanded uncertainty of measurement declared by laboratory Uxi /(%) Expanded uncertainty of measurement extended by stability UTS and linking U(D) / (%) di/% Eni 13.07 0.15 0.25 0.267 0.046 0.17 16.09 0.18 0.25 0.267 0.035 0.13 24.35 0.19 0.25 0.267 0.021 0.08 32.60 0.21 0.25 0.267 0.034 0.12 40.78 0.23 0.25 0.267 0.042 0.16 51.18 0.24 0.25 0.267 0.029 0.11 60.82 0.24 0.25 0.267 0.016 0.06 70.87 0.25 0.25 0.267 0.004 0.01 80.89 0.25 0.25 0.267 -0.010 0.04 90.92 0.26 0.25 0.274 0.003 0.01 100.81 0.26 0.25 0.274 0.003 0.01 Netherlands/VSL Flow of the transfer standard Q/(m3/h) Relative error of the transfer standard x/(%) Expanded uncertainty of measurement declared by laboratory Uxi /(%) Expanded uncertainty of measurement extended by stability UTS and linking U(D) / (%) di/% Eni 2.00 -0.09 0.094 0.134 0.006 0.04 4.50 0.07 0.094 0.134 0.105 0.66 6.60 0.11 0.094 0.134 0.084 0.53
EURAMET-P1180_FLOW_Final_Report.docx