8 Lead Length 1000 mm Mating connectors for 4-wire PRT Instruction Manual 9 Accessories: (standard/optional) Traceable Calibration Certificate (ISO 17025) 4. PORTABLE METROLOGY WELL/DRY BLOCK °C to 140 °C 1 Temperature range at 23 °C –95 ± 0.05 °C full range 2 Accuracy ± 0.015 °C full range 3 Stability ± 0.05 °C full range 4 Axial uniformity ±0.01 °C full range 5 Radial gradient 160 mm or more 6 Immersion (well) 200 V to 230 V (± 10 %) 50Hz 7 Power requirements PROCESS ELECTRONICS OPTIONAL 8 Accessories: Instruction Manual (standard/optional) 9 Traceable Calibration Certificate (ISO 5. PORTABLE IR THERMOMETER CALIBRATOR/IR SOURCE -LOW TEMPERATURE 1 Temperature range °C to 150 °C at 23 °C ambient –30 2 Accuracy ± 0.4 °C 3 Stability ± 0.1 °C 4 Target Emissivity 0.95 (± 0.02 from 8 to 14 µm) ≥ 5 Resolution 0.1 °C 6 Power 230 V ac (± 10%), 5 A,50Hz Instruction Manual Accessories: 7 Traceable Calibration Certificate (ISO (standard/optional) 17025)
8 Lead Length 1000 mm 9 Accessories: (standard/optional) Mating connectors for 4-wire PRT Instruction Manual Traceable Calibration Certificate (ISO 17025) 4. PORTABLE METROLOGY WELL/DRY BLOCK 1 Temperature range at 23 °C –95 °C to 140 °C 2 Accuracy ± 0.05 °C full range 3 Stability ± 0.015 °C full range 4 Axial uniformity ± 0.05 °C full range 5 Radial gradient ±0.01 °C full range 6 Immersion (well) 160 mm or more 7 Power requirements 200 V to 230 V (± 10 %) 50Hz 8 PROCESS ELECTRONICS OPTIONAL 9 Accessories: (standard/optional) Instruction Manual Traceable Calibration Certificate (ISO 5. PORTABLE IR THERMOMETER CALIBRATOR/IR SOURCE -LOW TEMPERATURE 1 Temperature range –30 °C to 150 °C at 23 °C ambient 2 Accuracy ± 0.4 °C 3 Stability ± 0.1 °C 4 Target Emissivity ≥ 0.95 (± 0.02 from 8 to 14 µm) 5 Resolution 0.1 °C 6 Power 230 V ac (± 10%), 5 A,50Hz 7 Accessories: (standard/optional) Instruction Manual Traceable Calibration Certificate (ISO 17025)
17175-RTD-151217.docx