Troubleshooting Problem: Possible cause: Action: The pump will not start Fit a battery No battery present Re-align battery terminals Battery inserted Fit a new battery Service incorrectly Battery is depleted/very required low Pump is faulty Infusion ended Check displayed rate Incorrect rate set early/going too quickly against Wrong syringe brand prescription change if selected during set necessary up Pump faulty or Ensure correct incorrectly understanding of calibrated user Service/calibration required Exhausted battery Fit new battery, turn The pump has Faulty pump pump on, stopped before emptying syringe confirm syringe size and brand select to resume infusion Return for service CME T34 Pump Alarms When the pump detects a problem four things occur The infusion stops An audible alarm is activated A message appears on the display screen indicating the cause of the alarm The LED indicator turns red
Troubleshooting Problem: Possible cause: Action: The pump will not start No battery present Battery inserted incorrectly Battery is depleted/very low Pump is faulty Fit a battery Re-align battery terminals Fit a new battery Service required Infusion ended early/going too quickly Incorrect rate set Wrong syringe brand selected during set up Pump faulty or incorrectly calibrated Check displayed rate against prescription change if necessary Ensure correct understanding of user Service/calibration required The pump has stopped before emptying syringe Exhausted battery Faulty pump Fit new battery, turn pump on, confirm syringe size and brand select to resume infusion Return for service CME T34 Pump Alarms When the pump detects a problem four things occur The infusion stops An audible alarm is activated A message appears on the display screen indicating the cause of the alarm The LED indicator turns red
ExRes-West-Midlands_7-Chapter-2-Planning-for-Palliative-Care.docx