Cable Entry ½” NPT threaded / similar for 18 glanding (on the remote transmitter and flow-tube) 19 Housing: IP-67 Supplier agrees to submit Certificate of Calibration 20 for flow from NABL / accredited Laboratory Flow Calibration Range: 1 to 80 m3/h (Minimum 20a calibration in gas), with measured pressure loss across meter for each flow; Density Calibration for the nominal 20b measurement range 1 to 5 kg/m3. Supplier agrees that FCRI reserves the right to witness the 20c Calibration of the Flowmeters prior to despatch / acceptance
18 Cable Entry ½” NPT threaded / similar for glanding (on the remote transmitter and flow-tube) 19 Housing: IP-67 20 Supplier agrees to submit Certificate of Calibration for flow from NABL / accredited Laboratory 20a Flow Calibration Range: 1 to 80 m3/h (Minimum calibration in gas), with measured pressure loss across meter for each flow; 20b Density Calibration for the nominal measurement range 1 to 5 kg/m3. 20c Supplier agrees that FCRI reserves the right to witness the Calibration of the Flowmeters prior to despatch / acceptance
17213-CMFM.docx