Test Results Parameter Target Test Plan results Reference Min Flow Rate 158 g/min Check flow rate at lowest Yes Pump pump settings testing, 11/8/2013 Max Flow Rate 1500 g/min Check flow rate at highest Yes, highest is 1504 Pump pump settings g/min testing, 11/8/2013 System Pressure 5 to 80 psi Check Pressure during Yes, pressure can Pump operation span 0.5-30psi with testing, supply 11/8/2013 Pump Response < 1 second Check time to new steady Yes, pump response Operating Time state after change is made time can be set via manual the PowerFlex drive pg. 3-12 Flow Accuracy < 1% error Compare flow meter value Yes, dependant on Pump against another flow flow rate, ranged testing, meter or total volume per between 0.1-1% 11/8/2013 time innacuracy Control Valve < 5 seconds Check time to new steady < 1 second Response Time state after change is made Leakage No Leaks Check for leaks under Yes normal operating procedures Operator 2 Minimum, 3 Attempt to operate the Yes Maximum system with 2 people Automatic Functioning PID Implement new set-points Yes Control control with ability and observe system to alter constants response Manual Control Can adjust flow Adjust flow Yes manually Automatic Data Labview records Ensure it exists and check Yes Collection data that can be for accuracy graphed or exported to excel Manual Data Real time data is Ensure it exists and check Yes Collection displayed for accuracy Modularity Can alter system to Reconfigure system to put Yes put different different components in comonents in alternate order series or parallel Portability Easy to move, Connent and disconnect Yes but wheels need connect to and system from utilities. Push to be replaced disconnect from cart around. utilities Noise 2 Sources Test if our sources effect Yes the data
Test Results Parameter Target Test Plan results Reference Min Flow Rate 158 g/min Check flow rate at lowest pump settings Yes Pump testing, 11/8/2013 Max Flow Rate 1500 g/min Check flow rate at highest pump settings Yes, highest is 1504 g/min Pump testing, 11/8/2013 System Pressure 5 to 80 psi Check Pressure during operation Yes, pressure can span 0.5-30psi with supply Pump testing, 11/8/2013 Pump Response Time < 1 second Check time to new steady state after change is made Yes, pump response time can be set via the PowerFlex drive Operating manual pg. 3-12 Flow Accuracy < 1% error Compare flow meter value against another flow meter or total volume per time Yes, dependant on flow rate, ranged between 0.1-1% innacuracy Pump testing, 11/8/2013 Control Valve Response Time < 5 seconds Check time to new steady state after change is made < 1 second Leakage No Leaks Check for leaks under normal operating procedures Yes Operator 2 Minimum, 3 Maximum Attempt to operate the system with 2 people Yes Automatic Control Functioning PID control with ability to alter constants Implement new set-points and observe system response Yes Manual Control Can adjust flow manually Adjust flow Yes Automatic Data Collection Labview records data that can be graphed or exported to excel Ensure it exists and check for accuracy Yes Manual Data Collection Real time data is displayed Ensure it exists and check for accuracy Yes Modularity Can alter system to put different comonents in series or parallel Reconfigure system to put different components in alternate order Yes Portability Easy to move, connect to and disconnect from utilities Connent and disconnect system from utilities. Push cart around. Yes but wheels need to be replaced Noise 2 Sources Test if our sources effect the data Yes
Final%20Design%20Review%20P13630.docx