Participants The 15 participants and the time schedule are shown in Table 1. The comparison measurements started in February 2010 and finished in May 2011. The CCM.FF-K6.2011 using the same transfer standard started in August 2010 and finished in December 2012. Each laboratory had several weeks for providing the measurements and for sending the transfer standard to the next laboratory. Due to some problems with customs documents the transfer standard shipment was delayed several times. The transfer standard was calibrated 7 times by the pilot laboratory before, during and after the comparison to assess its calibration stability. Tab.1 - Participants and the time schedule Country NMI Contact Date of calibration SMU Slovakia 15.2. to 28.2.2010 Slovak Institute of Stefan Makovnik (PILOT LAB) Metrology PTB 1.3.to 20.3.2010 Germany Physikalisch-Technische Bodo Mickan Bundesanstalt CEM (Centro Español de 20.3. to 30.4.2010 Spain Metrología) Nieves Medina Enagas S.A. LNE-LADG 5.5. to 12.5.2010 France Laboratoire Associé de Christophe Windenberg Débitmétrie Gazeuse VSL 18.5. to 28.5.2010 Netherlands National Metrology Mijndert P. van der Beek Institute of the Netherlands CMI Czech Republic Czech Metrology Institute Tomas 31.5. to 9.6.2010 Regional Inspectorate Valenta Pardubice SMU Slovakia 9.6. to 14.6.2010 Slovak Institute of Stefan Makovnik (PILOT LAB) Metrology BEV 14.6. to 20.6.2010 Austria Bundesamt für Eich- und Manfred Macek Vermessungswesen GUM Monika Kusyk 24.6. to 30.6.2010 Poland Główny Urząd Miar (Central Office of Measures) MKEH Hungarian Trade Licensing Hungary Csaba Czibulka 2.7. to 6.7.2010 Office. Section of Flow Measurement SP Technical Research 10.7. to 23.7.2010 Sweden Per Jacobsson Institute of Sweden
Participants The 15 participants and the time schedule are shown in Table 1. The comparison measurements started in February 2010 and finished in May 2011. The CCM.FF-K6.2011 using the same transfer standard started in August 2010 and finished in December 2012. Each laboratory had several weeks for providing the measurements and for sending the transfer standard to the next laboratory. Due to some problems with customs documents the transfer standard shipment was delayed several times. The transfer standard was calibrated 7 times by the pilot laboratory before, during and after the comparison to assess its calibration stability. Tab.1 - Participants and the time schedule Country NMI Contact Date of calibration Slovakia (PILOT LAB) SMU Slovak Institute of Metrology Stefan Makovnik 15.2. to 28.2.2010 Germany PTB Physikalisch-Technische Bundesanstalt Bodo Mickan 1.3.to 20.3.2010 Spain CEM (Centro Español de Metrología) Enagas S.A. Nieves Medina 20.3. to 30.4.2010 France LNE-LADG Laboratoire Associé de Débitmétrie Gazeuse Christophe Windenberg 5.5. to 12.5.2010 Netherlands VSL National Metrology Institute of the Netherlands Mijndert P. van der Beek 18.5. to 28.5.2010 Czech Republic CMI Czech Metrology Institute Regional Inspectorate Pardubice Tomas Valenta 31.5. to 9.6.2010 Slovakia (PILOT LAB) SMU Slovak Institute of Metrology Stefan Makovnik 9.6. to 14.6.2010 Austria BEV Bundesamt für Eich- und Vermessungswesen Manfred Macek 14.6. to 20.6.2010 Poland GUM Główny Urząd Miar (Central Office of Measures) Monika Kusyk 24.6. to 30.6.2010 Hungary MKEH Hungarian Trade Licensing Office. Section of Flow Measurement Csaba Czibulka 2.7. to 6.7.2010 Sweden SP Technical Research Institute of Sweden Per Jacobsson 10.7. to 23.7.2010
EURAMET-P1180_FLOW_Final_Report.docx