Signal Processing / Electronic Module: either head-mount on flow-tube or remote 16 transmitter (in which case 20 metre cable from flow-tube to remote transmitter shall be part of supply). Remote transmitter shall be provided with 17 accessory for Panel-mount, DIN or Cable Entry ½” NPT threaded / similar for 18 glanding (on the remote transmitter and flow-tube) 19 Housing: IP-67 Supplier agrees to submit Certificate of Calibration 20 for both flowmeters from NABL / ISO-17025 accredited Laboratory; Flow Calibration Range: 1 to 50 m3/h 20a (Minimum calibration in water), with measured
16 Signal Processing / Electronic Module: either head-mount on flow-tube or remote transmitter (in which case 20 metre cable from flow-tube to remote transmitter shall be part of supply). 17 Remote transmitter shall be provided with accessory for Panel-mount, DIN or 18 Cable Entry ½” NPT threaded / similar for glanding (on the remote transmitter and flow-tube) 19 Housing: IP-67 20 Supplier agrees to submit Certificate of Calibration for both flowmeters from NABL / ISO-17025 accredited Laboratory; 20a Flow Calibration Range: 1 to 50 m3/h (Minimum calibration in water), with measured
17213-CMFM.docx