Thunder scientific 2500: Humidity 10% to 95 % 0.5 % RH Humidity Generator Typical accuracy, Internal Reference: 25 Temperature: SPRT (0 °C, 100 Ω Ω Resistance 0 to 500 kΩ Super -Thermometer: Hart Probes & refr, 1 mA) 0.002°C, 10 Thermometry Resistance range Scientific 1575 thermistors kΩ thermistor (25°C, 10 kΩ refr, 0.01 mA) 0.0005 °C Temp MUA includes only drift and -200-420 °C 0.013°C (max) quartz Fluke 5683 measurement cal uncertainty of probe Temp MUA includes only and 420-990 °C 0.009°C (max) quartz Flue 5684 measurement cal uncer of probe Temp MUA includes only drift and 990-1200 °C 1.5°C (for max temp) S type TC measurement cal uncer of probe Temperature/ Precision Thermodynamic Thermocouples _10 mV to 100 mV 0.05°C Fluke 2565 with Hart 1560 Thermocouple Module Temperature/ Precision Reference ±0.02°C Kaye Ice point K150 Thermocouple Temperature 0°C Module Temperature/ Precision Thermometer/all 0.03% of reading + Isotech F150 Precision Thermodynamic Thermometers kinds of TC & _210 0C to 1200 0C 0.005% scale Thermometer PRT100: for instance, J type Thermocouple Thermodynamic Sources Hart 7080SD: stirred bath Temperature/ Does not include uncertainty Thermodynamic (temperature _80 0C to 110 0C 0.007°C (max) (must be monitored with source of measuring standard baths and external std) drywells)
Humidity 10% to 95 % 0.5 % RH Thunder scientific 2500: Humidity Generator Resistance Thermometry Temperature: Probes & thermistors 0 to 500 kΩ Resistance range Typical accuracy, Internal Reference: 25 Ω SPRT (0 °C, 100 Ω refr, 1 mA) 0.002°C, 10 kΩ thermistor (25°C, 10 kΩ refr, 0.01 mA) 0.0005 °C Super -Thermometer: Hart Scientific 1575 Temp measurement -200-420 °C 0.013°C (max) quartz Fluke 5683 MUA includes only drift and cal uncertainty of probe Temp measurement 420-990 °C 0.009°C (max) quartz Flue 5684 MUA includes only and cal uncer of probe Temp measurement 990-1200 °C 1.5°C (for max temp) S type TC MUA includes only drift and cal uncer of probe Thermodynamic Thermocouples Temperature/ Precision Thermocouple Module _10 mV to 100 mV 0.05°C Fluke 2565 with Hart 1560 Temperature/ Precision Thermocouple Module Reference Temperature 0°C ±0.02°C Kaye Ice point K150 Thermodynamic Thermometers Temperature/ Precision Thermometer/all kinds of TC & PRT100: for instance, J type Thermocouple _210 0C to 1200 0C 0.03% of reading + 0.005% scale Isotech F150 Precision Thermometer Thermodynamic Thermodynamic Sources (temperature baths and drywells) Temperature/ source _80 0C to 110 0C 0.007°C (max) Hart 7080SD: stirred bath (must be monitored with external std) Does not include uncertainty of measuring standard
161300-SOL-001-010.docx