MQ-210 Calibration ± 5 % (see calibration Traceability below) Uncertainty Measurement Less than 1 % Repeatability Long-term Drift Less than 2 % per year (Non-stability) Non-linearity Less than 1 % (up to 3000 µmol m-2 s-1) Response Time Less than 1 ms of View 180° 410 to 655 nm (wavelengths where response is greater than 50 % of maximum; Spectral Range see Spectral Response below) Directional (Cosine) ± 5 % at 75° zenith angle (see Cosine Response below) Response Temperature 0.06 ± 0.06 % per C (see Temperature Response Response 0 to 50 C; less than 90 % non-condensing relative humidity up to 30 C; less than 70 % Operating non-condensing relative humidity from 30 to 50 C; separate sensors can be Environment submerged in up to depths of 30 m Meter Dimensions 126 mm length; 70 mm width; 24 mm height Sensor Dimensions 24 mm diameter; 33 mm height Mass 180 g 2 m of two conductor, shielded, twisted-pair wire; TPR jacket (high water resistance, Cable high UV stability, flexibility in cold conditions) Calibration Traceability
MQ-210 Calibration Uncertainty ± 5 % (see calibration Traceability below) Measurement Repeatability Less than 1 % Long-term Drift (Non-stability) Less than 2 % per year Non-linearity Less than 1 % (up to 3000 µmol m-2 s-1) Response Time Less than 1 ms of View 180° Spectral Range 410 to 655 nm (wavelengths where response is greater than 50 % of maximum; see Spectral Response below) Directional (Cosine) Response ± 5 % at 75° zenith angle (see Cosine Response below) Temperature Response 0.06 ± 0.06 % per C (see Temperature Response Operating Environment 0 to 50 C; less than 90 % non-condensing relative humidity up to 30 C; less than 70 % non-condensing relative humidity from 30 to 50 C; separate sensors can be submerged in up to depths of 30 m Meter Dimensions 126 mm length; 70 mm width; 24 mm height Sensor Dimensions 24 mm diameter; 33 mm height Mass 180 g Cable 2 m of two conductor, shielded, twisted-pair wire; TPR jacket (high water resistance, high UV stability, flexibility in cold conditions) Calibration Traceability
MQ-210-manual.docx