Some negative Chl values. Chl records flagged 353 03-Jul-10 02-Oct-10 Double Cone as erroneous, sent to WetLabs for repair/service Failed during deployment. In water time since last calibration 36 weeks (failed on 2nd 1329 29-Jun-10 06-Oct-10 Dunk deployment). Sent to WetLabs for repair/service. Tropical Cyclone Yasi : star picket bent to 45° 828 06-Oct-10 12-Feb-11 Dunk angle. Logger still running but incorrectly positioned. Records look ok . Some negative Chl values. Chl records flagged 351 04-Oct-10 14-Feb-11 Geoffrey as erroneous, sent to WetLabs for repair/service Logger not logging - operator when set up. 846 30-Sep-10 17-Feb-11 Pelican Record lost, instrument checked and ok. No Chl & NTU readings during 01/01/2011 - 27/01/2011. In time since last calibration 1044 01-Oct-10 18-Feb-11 Pine 38 weeks. Sent to WET Labs: Replaced electronics with latest to eliminate instrument shutdown issues. Starpicket ripped out (anchor damage?). 2 Jan 2011 coral divers moved logger 25 m along to 819 02-Oct-10 19-Feb-11 Daydream picket 2 of same transect. Logger still running but incorrectly positioned. Records look ok. 16-Feb-11 16-Jun-11 1729 Barren Logging 'OFF' for deployment. Operator Negative Chl values with factory cal file, 17-Feb-11 16-Jun-11 817 Pelican records flagged as erroneous. Contacted WetLabs for solution.
353 03-Jul-10 02-Oct-10 Double Cone Some negative Chl values. Chl records flagged as erroneous, sent to WetLabs for repair/service 1329 29-Jun-10 06-Oct-10 Dunk Failed during deployment. In water time since last calibration 36 weeks (failed on 2nd deployment). Sent to WetLabs for repair/service. 828 06-Oct-10 12-Feb-11 Dunk Tropical Cyclone Yasi : star picket bent to 45° angle. Logger still running but incorrectly positioned. Records look ok . 351 04-Oct-10 14-Feb-11 Geoffrey Some negative Chl values. Chl records flagged as erroneous, sent to WetLabs for repair/service 846 30-Sep-10 17-Feb-11 Pelican Logger not logging - operator when set up. Record lost, instrument checked and ok. 1044 01-Oct-10 18-Feb-11 Pine No Chl & NTU readings during 01/01/2011 - 27/01/2011. In time since last calibration 38 weeks. Sent to WET Labs: Replaced electronics with latest to eliminate instrument shutdown issues. 819 02-Oct-10 19-Feb-11 Daydream Starpicket ripped out (anchor damage?). 2 Jan 2011 coral divers moved logger 25 m along to picket 2 of same transect. Logger still running but incorrectly positioned. Records look ok. 1729 16-Feb-11 16-Jun-11 Barren Logging 'OFF' for deployment. Operator 817 17-Feb-11 16-Jun-11 Pelican Negative Chl values with factory cal file, records flagged as erroneous. Contacted WetLabs for solution.
MMP_Inshore_Water_Quality_Monitoring_Report_2010_11.docx