Point Numbering: Instrument Points use single digit numbers 1-9 – Benchmarks use 90-99 – Turning Points use 70-79 – Side Shots start at 100. If you make a turn to another instrument set up, change side – shot numbers to differentiate Instrument Point 1 from Instrument Point 2 (eg, 100-199 for IP 1; 200-299 for IP 2; if you have more than 99 shots for IP1, use 300-399 for IP2, etc.) DEFAULT SETTINGS for the data collector: Survey Pro Main Menu JOB ->-> SETTINGS ->->-> – INSTRUMENT: [TOPCON] “Brand”: [GPT SERIES] “Model”: Port”: [COM 1] “Serial Rate”: [1200] “Baud [Even] “Parity”: UNITS for distances”: [US SURVEY FEET] “Units for angles”: [DEGREES] “Units directions as”: [AZIMUTH] “Display type”: AZIMUTH] “Azimuth FORMAT This relates to the precision you want for northing/easting, elevation, etc. FILES The only files currently loaded are the File” which is the description codes for the “Description survey shots. This is an ASCI file created on the PC and transferred to the data collector. Please refer to the users manual for descriptions of the Control File & Feature Code File. SURVEYING for Description” and for Height of Rod” should be checked on. All others “Prompt “Prompt are checked off. (Make sure with True Azimuths” is checked off. If this is checked “Survey on and you do a traverse ( or turn) the points are not in the right place after the first move.) STAKEOUT start stakeout with coarse mode” is the only thing checked on. “Always REPETITION tolerance”: [30] sec. “Horizontal Tolerance”: [30] sec. “Zenith Tolerance”: [0.4999999 usft] “Distance DATE/TIME Set current date & time GENERAL - User preference Charge Times: Data Collector 1 to 3 hours; Total Station 1 hour 48 minutes – –
Point Numbering: Instrument Points – use single digit numbers 1-9 Benchmarks – use 90-99 Turning Points – use 70-79 Side Shots – start at 100. If you make a turn to another instrument set up, change side shot numbers to differentiate Instrument Point 1 from Instrument Point 2 (eg, 100-199 for IP 1; 200-299 for IP 2; if you have more than 99 shots for IP1, use 300-399 for IP2, etc.) DEFAULT SETTINGS for the data collector: Survey Pro Main Menu – JOB ->-> SETTINGS ->->-> INSTRUMENT: “Brand”: [TOPCON] “Model”: [GPT SERIES] “Serial Port”: [COM 1] “Baud Rate”: [1200] “Parity”: [Even] UNITS “Units for distances”: [US SURVEY FEET] “Units for angles”: [DEGREES] “Display directions as”: [AZIMUTH] “Azimuth type”: AZIMUTH] FORMAT This relates to the precision you want for northing/easting, elevation, etc. FILES The only files currently loaded are the “Description File” which is the description codes for the survey shots. This is an ASCI file created on the PC and transferred to the data collector. Please refer to the users manual for descriptions of the Control File & Feature Code File. SURVEYING “Prompt for Description” and “Prompt for Height of Rod” should be checked on. All others are checked off. (Make sure “Survey with True Azimuths” is checked off. If this is checked on and you do a traverse ( or turn) the points are not in the right place after the first move.) STAKEOUT “Always start stakeout with coarse mode” is the only thing checked on. REPETITION “Horizontal tolerance”: [30] sec. “Zenith Tolerance”: [30] sec. “Distance Tolerance”: [0.4999999 usft] DATE/TIME Set current date & time GENERAL - User preference Charge Times: Data Collector – 1 to 3 hours; Total Station – 1 hour 48 minutes
nrcs144p2_014206.docx