1. Hardware Specifications 1-1. Overall View 1-2. Main Body 1-3. Head Unit 1-4. Utility Body 1-5. Pressure Bottle 2. Daily Operation for Sample Dispensing Window Layout * The Command Window has 4 groups Work, Daily Operation, Program, and Equipment Setup. – 2-1. Turn on the System Open the air cylinder and maintain the pressure of compressed air at 100 psi. Turn on the vacuum pump, the humidifier, and the chiller. Make sure to close all valves connected to the pressure bottles. Turn on the external switch and then the 110V internal switch in the utility body. Turn on the controller switch in the interior of the utility body. Turn on the computer and the monitor. Run the software. The software will initialize the system automatically when running. ‘ezAOI’ Open the outlet valves connected to the pressure bottles. 2-2. Fill the Pressure Bottles with Water or Alcohol Select the Operation’ window. ‘Daily Close the outlet valves connected to the pressure bottles. You may want to clean the pressure bottles containing water once a week to avoid microbial growth. Select in the Operation’ box. ‘Initialization’ ‘Daily Click in the box. ‘Release’ ‘Pressure’ Open the inlet valves connected to the pressure bottles. Fill the pressure bottles with water or alcohol. Close the inlet valves connected to the pressure bottles. Click of the menu in the box. ‘Start’ ‘Pneumatic’ ‘Initialization’ Open the outlet valves connected to the pressure bottles.
1. Hardware Specifications 1-1. Overall View 1-2. Main Body 1-3. Head Unit 1-4. Utility Body 1-5. Pressure Bottle 2. Daily Operation for Sample Dispensing Window Layout * The Command Window has 4 groups – Work, Daily Operation, Program, and Equipment Setup. 2-1. Turn on the System Open the air cylinder and maintain the pressure of compressed air at 100 psi. Turn on the vacuum pump, the humidifier, and the chiller. Make sure to close all valves connected to the pressure bottles. Turn on the external switch and then the 110V internal switch in the utility body. Turn on the controller switch in the interior of the utility body. Turn on the computer and the monitor. Run the ‘ezAOI’ software. The software will initialize the system automatically when running. Open the outlet valves connected to the pressure bottles. 2-2. Fill the Pressure Bottles with Water or Alcohol Select the ‘Daily Operation’ window. Close the outlet valves connected to the pressure bottles. You may want to clean the pressure bottles containing water once a week to avoid microbial growth. Select ‘Initialization’ in the ‘Daily Operation’ box. Click ‘Release’ in the ‘Pressure’ box. Open the inlet valves connected to the pressure bottles. Fill the pressure bottles with water or alcohol. Close the inlet valves connected to the pressure bottles. Click ‘Start’ of the ‘Pneumatic’ menu in the ‘Initialization’ box. Open the outlet valves connected to the pressure bottles.
MicroArray_Manual.docx