Manual Brochure ClearPoint® (MRI Interventions, Inc., Irvine, California, USA, ) Flexible headcoil is a must (esp. for kids) from CD: load exams VA: CT, axial T1 with contrast, axial T1 without, sagittal CUBE FLAIR, hybrid exam (if using DTI fibers). VCU: CT, sagittal T1 with contrast, axial T2 if need to remove exams: END Remove exam shows exam numbers and preview → – select patient (if exams come under different names, use Series” to rename) will “Manage → automatically show available images (uncheck undesired ones). will need CT (as reference exam), axial T1 with contrast (axial T1 without is usually not needed), sagittal CUBE FLAIR, hybrid exam (if using DTI fibers). just click Orientation” (there is no way to really verify it). “Verify click Merge” (may use on all images at the same time) “Auto Merge” click on corresponding points on the scan and then sign. Once 4 “Point – “+” points are added it starts showing accuracy. Once happy, click “Verify” merge” (axial, coronal, sagittal planes look for sella) NEXT (we don’t care about 3D “Verify – → model) NEXT (make sure it says NEXFRAME as software automatically detects when frameless → no frame is on CT). How to achieve best levels on CT move both sliders to the left move (top slider) to the right until – → “L” brain shows & pepper” granularity move (bottom slider) to the right until see perfect “salt → “W” parenchymal view.
Manual Brochure ClearPoint® (MRI Interventions, Inc., Irvine, California, USA, ) Flexible headcoil is a must (esp. for kids) load exams from CD: VA: CT, axial T1 with contrast, axial T1 without, sagittal CUBE FLAIR, hybrid exam (if using DTI fibers). VCU: CT, sagittal T1 with contrast, axial T2 if need to remove exams: END → Remove exam – shows exam numbers and preview select patient (if exams come under different names, use “Manage Series” to rename) → will automatically show available images (uncheck undesired ones). will need CT (as reference exam), axial T1 with contrast (axial T1 without is usually not needed), sagittal CUBE FLAIR, hybrid exam (if using DTI fibers). just click “Verify Orientation” (there is no way to really verify it). click “Auto Merge” (may use on all images at the same time) “Point Merge” – click on corresponding points on the scan and then “+” sign. Once 4 points are added it starts showing accuracy. Once happy, click “Verify” “Verify merge” (axial, coronal, sagittal planes – look for sella) → NEXT (we don’t care about 3D model) → NEXT (make sure it says NEXFRAME as software automatically detects frameless when no frame is on CT). How to achieve best levels on CT – move both sliders to the left → move “L” (top slider) to the right until brain shows “salt & pepper” granularity → move “W” (bottom slider) to the right until see perfect parenchymal view.
Op360.%20DBS.docx