Field Measurement (1.1.5.2.) Materials Multi-Probe w/ Black Protective Sleeve Handheld Connecting Cord Pen/Pencil Paper Extra Batteries for Handheld and Multi-Probe Procedure Connect the multi-probe to the handheld device using the long cord. Make sure to attach the safety clips and securely tighten the twist on protective cover. – Turn on the handheld device by pressing and holding the on/off button (it will take about 2 minutes to turn on) Connect handheld to multi-probe sonde by selecting the rotating arrows, then selecting when “rescan” your probe is found, select “connect”. Turn on the GPS by selecting the button ‘gears’ Select “Handheld” Arrow down to then push either the left or right arrow button. This will turn the GPS to “GPS” the “On” position. Select then “Apply”, “Okay” Press the button to return to the home screen “ESC” Under the running man tab, click on “Create Site” Under change the name to and insert date. For example, “Site name” “site_” today’s “site_07.07.2015” Press then “Create”, “Ok” Tab over to the running man. Click on Wait until GPS is no longer set to latitude “Dashboard” “-370” and longitude before beginning analysis. Once GPS has found signal, select “Capture Data” Under this screen press number six which corresponds with “change active site” Find the site you created and select it. This should change the active site (located in red at the bottom of the to the previously created site. “capture” screen) Insert probe into desired sample location. Press the 3 button on the handheld. At the top a green bar will appear that says When this bar goes away, you can take the probe out of the “Capture Data” sample location. Write down the time that corresponds with your sample. Continue to further sample locations and repeat as desired and wait until green bar appears (press “3” then disappears). Make sure to record the corresponding time with each sample.
Field Measurement (1.1.5.2.) Materials Multi-Probe w/ Black Protective Sleeve Handheld Connecting Cord Pen/Pencil Paper Extra Batteries for Handheld and Multi-Probe Procedure Connect the multi-probe to the handheld device using the long cord. Make sure to attach the safety clips and securely tighten the twist – on protective cover. Turn on the handheld device by pressing and holding the on/off button (it will take about 2 minutes to turn on) Connect handheld to multi-probe sonde by selecting the rotating arrows, then selecting “rescan” when your probe is found, select “connect”. Turn on the GPS by selecting the ‘gears’ button Select “Handheld” Arrow down to “GPS” then push either the left or right arrow button. This will turn the GPS to the “On” position. Select “Apply”, then “Okay” Press the “ESC” button to return to the home screen Under the running man tab, click on “Create Site” Under “Site name” change the name to “site_” and insert today’s date. For example, “site_07.07.2015” Press “Create”, then “Ok” Tab over to the running man. Click on “Dashboard” Wait until GPS is no longer set to “-370” latitude and longitude before beginning analysis. Once GPS has found signal, select “Capture Data” Under this screen press number six which corresponds with “change active site” Find the site you created and select it. This should change the active site (located in red at the bottom of the “capture” screen) to the previously created site. Insert probe into desired sample location. Press the 3 button on the handheld. At the top a green bar will appear that says “Capture Data” When this bar goes away, you can take the probe out of the sample location. Write down the time that corresponds with your sample. Continue to further sample locations and repeat as desired (press “3” and wait until green bar appears then disappears). Make sure to record the corresponding time with each sample.