As part of Trial 2, twelve instruments successfully recorded the release. For this as well as subsequent trials, the internal instrument clocks were synchronized to within one second of the official DPG (JR2 control) test clock. Instruments 006 and 012 did not register a significant increase throughout the trial. Pre-trial and post-trial QC checks indicate both instruments were properly collecting data. The highest recorded chlorine concentration for any of the instruments was 95,509 ppm; therefore all measurements for this trial fell within the linear portion of the standard curve and can be considered quantitative. Four instruments (001, 007, 010 and 015) did not collect. Instruments 001, 007 and 015 failed due to a power loss after the pre trial QC check and before the release. As for Trial 1, crimping/recrimping the power leads fixed this problem, Instrument 010 had a light source failure. The light source was replaced and the instrument recalibrated prior to Trial 3. Table 3: Data for Trial 2 Instruments Instruments Instruments Clock Trial Date Tonnage/Lbs. Functioning Disabled Saturated Synchronization 8 Ton/ 8/28/2015 12 4 0 Within 1s 17969 Lbs Baseline Highest Measurement Evidence of Quantitative Instrument (ppm) (ppm) Saturation? Value? JAZ-001 Disabled JAZ-002 473 20,634 N Y JAZ-003 943 35,224 N Y JAZ-004 271 34,761 N Y JAZ-005 706 47,575 N Y JAZ-006 554 601 N Y JAZ-007 Disabled JAZ-008 165 27,387 N Y JAZ-009 155 95,509 N Y JAZ-010 Disabled JAZ-011 280 52,742 N Y JAZ-012 347 379 N Y JAZ-013 360 34,124 N Y JAZ-014 217 26,826 N Y JAZ-015 Disabled JAZ-016 469 20,408 N Y 3.3 Trial 3
As part of Trial 2, twelve instruments successfully recorded the release. For this as well as subsequent trials, the internal instrument clocks were synchronized to within one second of the official DPG (JR2 control) test clock. Instruments 006 and 012 did not register a significant increase throughout the trial. Pre-trial and post-trial QC checks indicate both instruments were properly collecting data. The highest recorded chlorine concentration for any of the instruments was 95,509 ppm; therefore all measurements for this trial fell within the linear portion of the standard curve and can be considered quantitative. Four instruments (001, 007, 010 and 015) did not collect. Instruments 001, 007 and 015 failed due to a power loss after the pre trial QC check and before the release. As for Trial 1, crimping/recrimping the power leads fixed this problem, Instrument 010 had a light source failure. The light source was replaced and the instrument recalibrated prior to Trial 3. Table 3: Data for Trial 2 Trial Date Tonnage/Lbs. Instruments Functioning Instruments Disabled Instruments Saturated Clock Synchronization 8/28/2015 8 Ton/ 17969 Lbs 12 4 0 Within 1s Instrument Baseline (ppm) Highest Measurement (ppm) Evidence of Saturation? Quantitative Value? JAZ-001 Disabled JAZ-002 473 20,634 N Y JAZ-003 943 35,224 N Y JAZ-004 271 34,761 N Y JAZ-005 706 47,575 N Y JAZ-006 554 601 N Y JAZ-007 Disabled JAZ-008 165 27,387 N Y JAZ-009 155 95,509 N Y JAZ-010 Disabled JAZ-011 280 52,742 N Y JAZ-012 347 379 N Y JAZ-013 360 34,124 N Y JAZ-014 217 26,826 N Y JAZ-015 Disabled JAZ-016 469 20,408 N Y 3.3 Trial 3