Acceptable. 6 QAP states that pH and temperature 40 CFR N/A Advanced TENORM measurements for 136.3 Services does not compliance samples must be Table II collect samples. obtained within hold time (temperature immediately, WWCM1 pH within 15 minutes of Chapter collection). Reported II results (i.e., DMR) must Section be from analyses performed 10 within hold time by personnel of a lab certified to perform the analyses. 7 pH SOP does not Record pH meter slope WWCM1 Acceptable. N/A include recording monthly, after Chapter of the slope or calibration. Slope shall II an acceptance be between 95% and 105%. Section range for the 8.1.5 slope. 8 pH SOP states ± A pH calibration check WWCM1 Acceptable. N/A 0.2 pH units as solution shall be analyzed Chapter the acceptance and read within ± 0.1 S.U. II range for of the true value. Section calibration 8.1.6 verification. SM 4500H+ B-2000 Section 2a
6 QAP states that Advanced TENORM Services does not collect samples. pH and temperature measurements for compliance samples must be obtained within hold time (temperature immediately, pH within 15 minutes of collection). Reported results (i.e., DMR) must be from analyses performed within hold time by personnel of a lab certified to perform the analyses. 40 CFR 136.3 Table II WWCM1 Chapter II Section 10 Acceptable. N/A 7 pH SOP does not include recording of the slope or an acceptance range for the slope. Record pH meter slope monthly, after calibration. Slope shall be between 95% and 105%. WWCM1 Chapter II Section 8.1.5 Acceptable. N/A 8 pH SOP states ± 0.2 pH units as the acceptance range for calibration verification. A pH calibration check solution shall be analyzed and read within ± 0.1 S.U. of the true value. WWCM1 Chapter II Section 8.1.6 SM 4500H+ B-2000 Section 2a Acceptable. N/A