review, these instruments did not appear to reach true saturation (with a single exception) although these measurements were well beyond the linear range for quantitation as described above. Baseline or background measurements for each instrument were calculated by averaging the readings taken during the last full minute preceding the trial. Baseline values typically returned to within one standard deviation of this average within 17 minutes after the start of the trial. Plots of the chlorine concentration (ppm) over these 17 minutes for each instrument are shown in Appendix B. For Appendix B, plots labeled with red on the y-axis (concentration) denote a different scale (0 to 250,000 ppm) than those labeled in black (0 to 100,000 ppm). JazTM UV-Vis measurements in excess of 100,000 ppm were relatively infrequent across the five trials. Table 1: Overall Test Data Jaz Instruments Instruments Instruments Clock Trial Date Tonnage/Lbs. Instruments Functioning Disabled Saturated Synchronization 5 Ton/ 8/24/2015 16 13 3 0 Within 60s 9940 Lbs 8 Ton/ 8/28/2015 16 12 4 0 Within 1s 17969 Lbs 5 Ton/ 8/29/2015 16 15 1 0 Within 1s 9947 Lbs 7.7 Ton/ 9/1/2015 16 16 0 1 Within 1s 15366 Lbs 9 Ton/ 9/3/2015 16 15 1 0 Within 1s 18306 Lbs 3.1 Trial 1 Trial 1 was carried out on August 24, 2015. Instruments were placed within the mock urban layout as illustrated by Figure 1. More specific instrument locations including proximity to survey points are located in Appendix C. Figure 1. Instrument Layout for Trial 1 Quality control checks were performed on each instrument before and after the trial using a 10,000 ppm chlorine standard. Figure 2 shows high concordance between the pre-trial (blue
review, these instruments did not appear to reach true saturation (with a single exception) although these measurements were well beyond the linear range for quantitation as described above. Baseline or background measurements for each instrument were calculated by averaging the readings taken during the last full minute preceding the trial. Baseline values typically returned to within one standard deviation of this average within 17 minutes after the start of the trial. Plots of the chlorine concentration (ppm) over these 17 minutes for each instrument are shown in Appendix B. For Appendix B, plots labeled with red on the y-axis (concentration) denote a different scale (0 to 250,000 ppm) than those labeled in black (0 to 100,000 ppm). JazTM UV-Vis measurements in excess of 100,000 ppm were relatively infrequent across the five trials. Table 1: Overall Test Data Trial Date Tonnage/Lbs. Jaz Instruments Instruments Functioning Instruments Disabled Instruments Saturated Clock Synchronization 8/24/2015 5 Ton/ 9940 Lbs 16 13 3 0 Within 60s 8/28/2015 8 Ton/ 17969 Lbs 16 12 4 0 Within 1s 8/29/2015 5 Ton/ 9947 Lbs 16 15 1 0 Within 1s 9/1/2015 7.7 Ton/ 15366 Lbs 16 16 0 1 Within 1s 9/3/2015 9 Ton/ 18306 Lbs 16 15 1 0 Within 1s 3.1 Trial 1 Trial 1 was carried out on August 24, 2015. Instruments were placed within the mock urban layout as illustrated by Figure 1. More specific instrument locations including proximity to survey points are located in Appendix C. Figure 1. Instrument Layout for Trial 1 Quality control checks were performed on each instrument before and after the trial using a 10,000 ppm chlorine standard. Figure 2 shows high concordance between the pre-trial (blue