Verify that the Live Rule Set matches Appendix C: Instrument Side Rules and includes changes made to SetReceivingFacility and SetInstrumentID (Figure 20). Figure : Example of Live Rules Set View If problems are encountered, file a service request to coordinate assistance with installation. Do not proceed until the issue is resolved. Close the Configuration Options and Mappings window. Import VBECS (HL7) Side Configuration Only one HL7 configuration is needed even if you use multiple Vision instruments. All Visions will share the same HL7 configuration. Modifying rules or test code mappings in the Instrument Manager configuration outside of this Setup Guide is not allowed and may lead to malfunction of the Automated Instrument to VBECS interface. Navigate to Configuration -> Configuration Editor (Figure 21). Click on the Import button. Figure : Example of Configuration Editor Window Once the Restore Driver Configuration window opens, click Browse the button and select C:\Temp folder (Figure 22). Figure : Example of Configuration Import Window Select Vision HL7 Interface Configuration Template for VBECS 09122018.gsb file from – the list. Enter that contains 3 letter location code of the instrument (e.g., Configuration Name for Hines VAMC), word and sequence number (1 for the first configuration, for HIN VBECS 2 the second etc.). Example Configuration Name for VBECS side configuration located at Hines would be: HIN_VBECS_1 This configuration name will be further referred in this document as <HL7 Side Configuration>. Enter Configuration Description and click Import button. Verify that the confirmation window in Figure 23 displays. Figure : Example of Successful Configuration Import If you are using newer version of the driver than the one mentioned in section 3, the Instrument Manager will warn you about the discrepancy in driver versions. Acknowledge this warning and continue. Click OK and close the Restore Driver Configuration window. Verify that Configuration Editor shows the new configuration on the list (Figure 24).
Verify that the Live Rule Set matches Appendix C: Instrument Side Rules and includes changes made to SetReceivingFacility and SetInstrumentID (Figure 20). Figure : Example of Live Rules Set View If problems are encountered, file a service request to coordinate assistance with installation. Do not proceed until the issue is resolved. Close the Configuration Options and Mappings window. Import VBECS (HL7) Side Configuration Only one HL7 configuration is needed even if you use multiple Vision instruments. All Visions will share the same HL7 configuration. Modifying rules or test code mappings in the Instrument Manager configuration outside of this Setup Guide is not allowed and may lead to malfunction of the Automated Instrument to VBECS interface. Navigate to Configuration -> Configuration Editor (Figure 21). Click on the Import button. Figure : Example of Configuration Editor Window Once the Restore Driver Configuration window opens, click Browse the button and select C:\Temp folder (Figure 22). Figure : Example of Configuration Import Window Select Vision – HL7 Interface Configuration Template for VBECS 09122018.gsb file from the list. Enter Configuration Name that contains 3 letter location code of the instrument (e.g., HIN for Hines VAMC), word VBECS and sequence number (1 for the first configuration, 2 for the second etc.). Example Configuration Name for VBECS side configuration located at Hines would be: HIN_VBECS_1 This configuration name will be further referred in this document as <HL7 Side Configuration>. Enter Configuration Description and click Import button. Verify that the confirmation window in Figure 23 displays. Figure : Example of Successful Configuration Import If you are using newer version of the driver than the one mentioned in section 3, the Instrument Manager will warn you about the discrepancy in driver versions. Acknowledge this warning and continue. Click OK and close the Restore Driver Configuration window. Verify that Configuration Editor shows the new configuration on the list (Figure 24).