d-PIEZO Resolution approximately 1mm of water over 100m  range YieldPoint’s d-Piezo digital piezometer integrates a -35C to +85 C Temperature  MEMS transducer with a micro-controller to provide a 0.1C Temp resolution.  robust digital solution for water pressure and depth monitoring. The output signal is directly in kPa(0.01kPa RS485 output signal with 9600baud rate.  resolution) and degC (0.1C resolution). The digital output string is tramsmitted as an RS485 signal, with 9600 baud ASCII encoded output string with Unique ID.  rate(9600,9,n,1), and is ASCII encoded. Each instrument has a unique ID. RS485 signals tolerate wet conditions and Operates with data-loggers + wireless telemetry  the signals can be transmitted up to 1000m without solutions amplification. On board signal processing  Data from an of d-Piezos can be remotely monitored using YieldPoint’s Plug & Play data-loggers and telemetry RS485 output signal tional on-board digital temperature  solutions. Low cost wireless networking is available using sensor for additional information d-Mesh 900MHz radios which have a standard LOS range of 500m. All solutions are eminently scalable. Diameter 25mm  High survivability electronics  Easy to install and maintain  Arrives on site fully assembled  Features: digital technology MEMS (unvented). 0-1000kPa 0.01kPa resolution.  d-SLOUGH Operation Specifications
d-PIEZO YieldPoint’s d-Piezo digital piezometer integrates a MEMS transducer with a micro-controller to provide a robust digital solution for water pressure and depth monitoring. The output signal is directly in kPa(0.01kPa resolution) and degC (0.1C resolution). The digital output string is tramsmitted as an RS485 signal, with 9600 baud rate(9600,9,n,1), and is ASCII encoded. Each instrument has a unique ID. RS485 signals tolerate wet conditions and the signals can be transmitted up to 1000m without amplification. Data from an of d-Piezos can be remotely monitored using YieldPoint’s Plug & Play data-loggers and telemetry solutions. Low cost wireless networking is available using d-Mesh 900MHz radios which have a standard LOS range of 500m. All solutions are eminently scalable. Features: MEMS digital technology 0-1000kPa (unvented).  0.01kPa resolution.  Resolution approximately 1mm of water over 100m range  -35C to +85 C Temperature  0.1C Temp resolution.  RS485 output signal with 9600baud rate.  ASCII encoded output string with Unique ID.  Operates with data-loggers + wireless telemetry solutions  On board signal processing  RS485 output signal tional on-board digital temperature sensor for additional information  Diameter 25mm  High survivability electronics  Easy to install and maintain  Arrives on site fully assembled d-SLOUGH Operation Specifications