the CU to draw the features in the map function. Also, identifying the point as a control point will make it appear differently on the map function in the CU. After that is stored the CU will prompt for the Back sight information. This includes name, pcode, height, azimuth (can be set to zero or other desired azimuth), and method of measure. Method of measure will allow the back sight to be recorded as an angle only or angle and distance. Once all of this information is entered the station setup can be completed by pressing the soft key “Store.” (SLIDE #11) e. Initialize for Topography: To start a topography survey, tap the Survey icon and select Measure topo under the menu. At this point, the CU will prompt for the starting point name (usually set to 101 for first part of a survey), pcode, method (must be set to angles and distance for topography), and target height. Before pressing the Measure soft key the S6 will need to be sighted on the target. In Standard measurement mode the soft keys will need to be pressed repeatedly in order to measure and store the points. However, engaging the auto lock will allow for faster data collection. The auto lock feature can be engage by selecting it under the instrument icon on the bottom right of the main menu. The default display will show the slope distance, horizontal and vertical angles. To turn the page for more information such as horizontal distance and coordinates simply press the soft key to the left of the display. (SLIDE #12) f. Measure Rounds: Measure rounds is the program for running a traverse from inside survey controller. Most of the information entered in Survey Controller will be the same as for topography. The most unique feature to the CU is that is can perform an almost fully automated traverse. After the station setup is complete access the traversing module by pressing the Survey icon and selecting Measure Rounds. The next screen will require back sight information and it is at this point you can select the options soft key. On the options screen you can choose automated rounds by checking the box. Other options include Face order, observation
the CU to draw the features in the map function. Also, identifying the point as a control point will make it appear differently on the map function in the CU. After that is stored the CU will prompt for the Back sight information. This includes name, pcode, height, azimuth (can be set to zero or other desired azimuth), and method of measure. Method of measure will allow the back sight to be recorded as an angle only or angle and distance. Once all of this information is entered the station setup can be completed by pressing the soft key “Store.” (SLIDE #11) e. Initialize for Topography: To start a topography survey, tap the Survey icon and select Measure topo under the menu. At this point, the CU will prompt for the starting point name (usually set to 101 for first part of a survey), pcode, method (must be set to angles and distance for topography), and target height. Before pressing the Measure soft key the S6 will need to be sighted on the target. In Standard measurement mode the soft keys will need to be pressed repeatedly in order to measure and store the points. However, engaging the auto lock will allow for faster data collection. The auto lock feature can be engage by selecting it under the instrument icon on the bottom right of the main menu. The default display will show the slope distance, horizontal and vertical angles. To turn the page for more information such as horizontal distance and coordinates simply press the soft key to the left of the display. (SLIDE #12) f. Measure Rounds: Measure rounds is the program for running a traverse from inside survey controller. Most of the information entered in Survey Controller will be the same as for topography. The most unique feature to the CU is that is can perform an almost fully automated traverse. After the station setup is complete access the traversing module by pressing the Survey icon and selecting Measure Rounds. The next screen will require back sight information and it is at this point you can select the options soft key. On the options screen you can choose automated rounds by checking the box. Other options include Face order, observation