The following table indicates the appropriate actions to be taken Alarm: Possible cause: Action: Occlusion or syringe Remove occlusion and Patient access device empty restart or re-load syringe blocked, kinked, clamped or occluded Flush or replace access Actuator has reached device minimum travel position Release clamp End of program, turn pump OFF Syringe displaced Syringe has been Check & confirm syringe removed or displaced seated correctly and resume Pump paused too long Pump left or no key Start infusion, presses detected for 2 continue programming minutes or switch off Near end 15 minutes from end of Prepare to change infusion syringe or switch off End program Infusion Complete Pump will either default to KVO (keep vein open) or it will alarm in which case switch pump off and await nurse to replenish/remove Low battery Prepare to Battery is almost change battery depleted (30 minutes left) End battery Battery is depleted Change battery
The following table indicates the appropriate actions to be taken Alarm: Possible cause: Action: Occlusion or syringe empty Patient access device blocked, kinked, clamped or occluded Actuator has reached minimum travel position Remove occlusion and restart or re-load syringe Flush or replace access device Release clamp End of program, turn pump OFF Syringe displaced Syringe has been removed or displaced Check & confirm syringe seated correctly and resume Pump paused too long Pump left or no key presses detected for 2 minutes Start infusion, continue programming or switch off Near end 15 minutes from end of infusion Prepare to change syringe or switch off End program Infusion Complete Pump will either default to KVO (keep vein open) or it will alarm in which case switch pump off and await nurse to replenish/remove Low battery Battery is almost depleted (30 minutes left) Prepare to change battery End battery Battery is depleted Change battery