information. The ICE system shall allow user clinicians to be able to leave customized messages for other clinicians caring for the Caregiver 7.2.7 CR_0000000047 patient - "medication notes requires a filter, monitor patient closely for signs of reaction, follow chemotherapy precautions, etc." Upon encountering an error condition, the ICE Error system shall ensure that 7.2.8 CR_0000000048 conditions - the remainder of any bolus active bolus shall be cancelled and further doses shall be locked out. The pump shall indicate Pump status 7.2.9 CR_0000000092 when it is under the notification control of the ICE system. Important for point of care troubleshooting; "control" is limited The pump shall information - also Pump indicate when it is need to know if 7.2.10 CR_0000000093 commincation communicating with communicating and the ICE system. [jmg] not able to assume control, or communicating and monitoring status, etc. The ICE system shall provide a means to capture/display and utilize the patient's active Patient medication list and This info would affect 7.2.11 CR_0000000105 Medical medication history and dosage rate History other medical history, especially of comorbidities such as sleep apnea, allergies, or sensitivity to opioids. The pump stop time The PCA infusion pump must stop delivering needs to be medication within a determined and 7.2.12 CR_0000000107 STOP time defined time period after regulated in order to receiving the STOP be able to intervene on command from the ICE a timely basis. system. When the infusion pump 7.2.13 CR_0000000108 Remote versus Clinician needs to be is under control by the
information. 7.2.7 CR_0000000047 Caregiver notes The ICE system shall allow user clinicians to be able to leave customized messages for other clinicians caring for the patient - "medication requires a filter, monitor patient closely for signs of reaction, follow chemotherapy precautions, etc." 7.2.8 CR_0000000048 Error conditions - bolus Upon encountering an error condition, the ICE system shall ensure that the remainder of any active bolus shall be cancelled and further doses shall be locked out. 7.2.9 CR_0000000092 Pump status notification The pump shall indicate when it is under the control of the ICE system. 7.2.10 CR_0000000093 Pump commincation The pump shall indicate when it is communicating with the ICE system. [jmg] Important for point of care troubleshooting; "control" is limited information - also need to know if communicating and not able to assume control, or communicating and monitoring status, etc. 7.2.11 CR_0000000105 Patient Medical History The ICE system shall provide a means to capture/display and utilize the patient's active medication list and medication history and other medical history, especially of comorbidities such as sleep apnea, allergies, or sensitivity to opioids. This info would affect dosage rate 7.2.12 CR_0000000107 STOP time The PCA infusion pump must stop delivering medication within a defined time period after receiving the STOP command from the ICE system. The pump stop time needs to be determined and regulated in order to be able to intervene on a timely basis. 7.2.13 CR_0000000108 Remote versus When the infusion pump is under control by the Clinician needs to be