1. Amp. Gain is too small Check standard weight’s weight with set value. 2. Sig+ and Sig- wire If is difference between set value and real Err 07 connection error weight, please re-input the value 3. Test weight is not loaded (set value is too big) Under model, “F-function” Err 08 the correct value and re-input set value is “N.A” When Y.Y has the between 3.9 ~ 9.9 at Y.YXXXX as Span value, Change the Max.capacity/digit value Err 09 If standard weight value is (Ex: digit 01 05) less than 10% of Max. Capacity When there is continuous - Find vibration cause and remove vibration on the weighing - Load cell check Err A part,, indicator can not - Load cell cable and connecting condition process calibration any check more. WARRANTEE CETIFICATION This product is passed strict quality test. “CAS”s If there is defect of manufacturing or abnormal detection within warrantee period, please contact our Agent or Distributor with this Warrantee certificate. Then, we will repair or replace free of charge. WARRANTEE CLAUSE
Err 07 1. Amp. Gain is too small 2. Sig+ and Sig- wire connection error 3. Test weight is not loaded Check standard weight’s weight with set value. If is difference between set value and real weight, please re-input the value (set value is too big) Err 08 Under “F-function” model, set value is “N.A” the correct value and re-input Err 09 When Y.Y has the between 3.9 ~ 9.9 at Y.YXXXX as Span value, If standard weight value is less than 10% of Max. Capacity Change the Max.capacity/digit value (Ex: digit 01 05) Err A When there is continuous vibration on the weighing part,, indicator can not process calibration any more. - Find vibration cause and remove - Load cell check - Load cell cable and connecting condition check WARRANTEE CETIFICATION This product is passed “CAS”s strict quality test. If there is defect of manufacturing or abnormal detection within warrantee period, please contact our Agent or Distributor with this Warrantee certificate. Then, we will repair or replace free of charge. WARRANTEE CLAUSE