In the Beam Attenuator tab, set the Type to “None” In the Filter tab, set the to Type “None” In the Soller Slit tab, use the Type drop-down menu to indicate what size V.2.D Soller slit you inserted into the instrument. Click OK to close the dialog window Configure the Diffracted Beam Optics tab in the Instrument Window Double-click on the item Diffracted Beam Path to open the Incident Beam window Optics In the PreFIX Module tab, set the Type to “X’Celerator” if the monochromator, set the Type to & “X’Celerator Monochromator Cu” In the Anti-scatter Slit tab, set to match the incident-beam divergence slit Select AS Slit” from the drop-down menu for Type “Prog With the Aperature (°) should be identical to that for the “Fixed” incident-beam Programmable Divergence Slit With Usage the Observed Length (mm) will be the same “Automatic” as the Irradiated length (mm) for the incident-beam Divergence Slit In the Receiving Slit tab, set the Type to “None” In the Filter tab, set the Type to “Nickel” In the Mask tab, set the Type to “None” In the Soller Slit tab, use the Type drop-down menu to indicate what size Soller slit you inserted into the instrument. In the Monochromator tab, set the Type to “None” The Monochromator will be set to Type Beam Flat 1x graphite “Diffr. for Cu (X’Cel)” if using the optional monochromator. In the Collimator tab, set the Type to “None” In the Detector tab Set Type to “X’Celerator[2]” Set Usage to “Scanning” Set Active angle (°2Theta) to “2.122”
In the Beam Attenuator tab, set the Type to “None” In the Filter tab, set the Type to “None” In the Soller Slit tab, use the Type drop-down menu to indicate what size Soller slit you inserted into the instrument. Click OK to close the dialog window Configure the Diffracted Beam Optics tab in the Instrument Window Double-click on the item Diffracted Beam Path to open the Incident Beam Optics window In the PreFIX Module tab, set the Type to “X’Celerator” if the monochromator, set the Type to “X’Celerator & Monochromator Cu” In the Anti-scatter Slit tab, set to match the incident-beam divergence slit Select “Prog AS Slit” from the drop-down menu for Type With “Fixed” the Aperature (°) should be identical to that for the incident-beam Programmable Divergence Slit With Usage “Automatic” the Observed Length (mm) will be the same as the Irradiated length (mm) for the incident-beam Divergence Slit In the Receiving Slit tab, set the Type to “None” In the Filter tab, set the Type to “Nickel” In the Mask tab, set the Type to “None” In the Soller Slit tab, use the Type drop-down menu to indicate what size Soller slit you inserted into the instrument. In the Monochromator tab, set the Type to “None” The Monochromator Type will be set to “Diffr. Beam Flat 1x graphite for Cu (X’Cel)” if using the optional monochromator. In the Collimator tab, set the Type to “None” In the Detector tab Set Type to “X’Celerator[2]” Set Usage to “Scanning” Set Active angle (°2Theta) to “2.122” V.2.D