Modifying Cable Length Although it is possible to splice additional cable to the separate sensor of the MU-200, note that the cable wires are soldered directly into the circuit board of the meter. Care should be taken to remove the back panel of the meter in order to access the board and splice on the additional cable, otherwise two splices would need to be made between the meter and sensor head. See Apogee webpage for further details on how to extend sensor cable length: (http://www.apogeeinstruments.com/how-to-make-a-weatherproof-cable-splice/). RETURN WARRANTY POLICY Apogee Instruments will accept returns within 30 days of purchase as long as the product is in new condition (to be determined by Apogee). Returns are subject to a 10 % restocking fee. POLICY What is Covered All products manufactured by Apogee Instruments are warranted to be free from defects in materials and craftsmanship for a period of four (4) years from the date of shipment from our factory. To be considered for warranty coverage an item must be evaluated either at our factory or by an authorized distributor. Products not manufactured by Apogee (spectroradiometers, chlorophyll content meters) are covered for a period of one (1) year. What is Not Covered The customer is responsible for all associated with the removal, reinstallation, and shipping of suspected warranty items to our factory. The warranty does not cover equipment that has been damaged due to the following conditions: 1. Improper installation or abuse. 2. Operation of the instrument outside of its specified operating range. 3. Natural occurrences such as lightning, fire, etc. 4. Unauthorized modification. 5. Improper or unauthorized repair. Please note that nominal accuracy is normal over time. Routine recalibration of sensors/meters is considered part of proper maintenance and is not covered under warranty.
Modifying Cable Length Although it is possible to splice additional cable to the separate sensor of the MU-200, note that the cable wires are soldered directly into the circuit board of the meter. Care should be taken to remove the back panel of the meter in order to access the board and splice on the additional cable, otherwise two splices would need to be made between the meter and sensor head. See Apogee webpage for further details on how to extend sensor cable length: (http://www.apogeeinstruments.com/how-to-make-a-weatherproof-cable-splice/). RETURN POLICY Apogee Instruments will accept returns within 30 days of purchase as long as the product is in new condition (to be determined by Apogee). Returns are subject to a 10 % restocking fee. WARRANTY POLICY What is Covered All products manufactured by Apogee Instruments are warranted to be free from defects in materials and craftsmanship for a period of four (4) years from the date of shipment from our factory. To be considered for warranty coverage an item must be evaluated either at our factory or by an authorized distributor. Products not manufactured by Apogee (spectroradiometers, chlorophyll content meters) are covered for a period of one (1) year. What is Not Covered The customer is responsible for all associated with the removal, reinstallation, and shipping of suspected warranty items to our factory. The warranty does not cover equipment that has been damaged due to the following conditions: 1. Improper installation or abuse. 2. Operation of the instrument outside of its specified operating range. 3. Natural occurrences such as lightning, fire, etc. 4. Unauthorized modification. 5. Improper or unauthorized repair. Please note that nominal accuracy is normal over time. Routine recalibration of sensors/meters is considered part of proper maintenance and is not covered under warranty.