or services, so as to eliminate errors related to patient misidentification. When the ICE system detects a a data entry error, it shall give specific 6.2.32 CR_0000000085 Error reporting information regarding the error: wrong format, wrong limits, incompatible values, etc. Battery status (% charge remaining, # of hours remaining, etc.) shall be conveyed to the clinician 6.2.33 CR_0000000086 Battery status if the charge is below a certain level, so the pump can operate safely when power fails or is unplugged. Making MRNs from The ICE system shall be various source available able to display the allows the clinician to patient's medical record check that the MRN from number (MRN) as it is devices and orders matches Patient medical stored in the ICE app, any the one encoded on the IV 6.2.34 CR_0000000087 record number connected medical bag or syringe prepared by devices, the patient's pharmacy to ensure that wristband, and any they have the RIGHT available medication label PATIENT before the such as a barcode. medication is administered. We need to ensure that ICE is able to capture The clinician shall verify ALL the device that all the devices Clinician information so the 6.2.35 CR_0000000167 connected to the patient Verification ICE System has the are displayed correctly on necessary data inputs the ICE System. and control outputs available. To avoid accidental The ICE system may Accidental require at least two steps tampering with settings 6.2.36 CR_0000000168 tampering to change some settings such as alarm limits, etc., or issue some commands. To provide clarity to While waiting for action the clinician and or input from the user, the 6.2.38 CR_0000000170 User prompting prevent waste of time ICE system shall clearly waiting for input from notify the user what action it is waiting for. user. Reference The ICE system shall 6.2.39 CR_0000000171 Databases and [need examples here] connect to other Clinical Reference Databases and
or services, so as to eliminate errors related to patient misidentification. 6.2.32 CR_0000000085 Error reporting When the ICE system detects a a data entry error, it shall give specific information regarding the error: wrong format, wrong limits, incompatible values, etc. 6.2.33 CR_0000000086 Battery status Battery status (% charge remaining, # of hours remaining, etc.) shall be conveyed to the clinician if the charge is below a certain level, so the pump can operate safely when power fails or is unplugged. 6.2.34 CR_0000000087 Patient medical record number The ICE system shall be able to display the patient's medical record number (MRN) as it is stored in the ICE app, any connected medical devices, the patient's wristband, and any available medication label such as a barcode. Making MRNs from various source available allows the clinician to check that the MRN from devices and orders matches the one encoded on the IV bag or syringe prepared by pharmacy to ensure that they have the RIGHT PATIENT before the medication is administered. 6.2.35 CR_0000000167 Clinician Verification The clinician shall verify that all the devices connected to the patient are displayed correctly on the ICE System. We need to ensure that ICE is able to capture ALL the device information so the ICE System has the necessary data inputs and control outputs available. 6.2.36 CR_0000000168 Accidental tampering The ICE system may require at least two steps to change some settings or issue some commands. To avoid accidental tampering with settings such as alarm limits, etc., 6.2.38 CR_0000000170 User prompting While waiting for action or input from the user, the ICE system shall clearly notify the user what action it is waiting for. To provide clarity to the clinician and prevent waste of time waiting for input from user. 6.2.39 CR_0000000171 Reference Databases and Clinical The ICE system shall connect to other Reference Databases and [need examples here]