Photoacoustic imaging
Photoacoustic imaging or optoacoustic imaging is a biomedical imaging modality based on the photoacoustic effect. Non-ionizing laser pulses are delivered into biological tissues and part of the energy will be absorbed and converted into heat, leading to transient thermoelastic expansion and thus wideband (i.e. MHz) ultrasonic emission. The generated ultrasonic waves are detected by ultrasonic transducers and then analyzed to produce images. It is known that optical absorption is closely associated with physiological properties, such as hemoglobin concentration and oxygen saturation. As a result, the magnitude of the ultrasonic emission (i.e. photoacoustic signal), which is proportional to the local energy deposition, reveals physiologically specific optical absorption contrast. 2D or 3D images of the targeted areas can then be formed. Biomedical imaging 

The optical absorption in biological tissues can be due to endogenous molecules such as hemoglobin or melanin, or exogenously delivered contrast agents. As an example, Fig. 2 shows the optical absorption spectra of oxygenated hemoglobin (HbO2) and deoxygenated hemoglobin (Hb) in the visible and near infrared region. Since blood usually has orders of magnitude higher absorption than surrounding tissues, there is sufficient endogenous contrast for photoacoustic imaging to visualize blood vessels. Recent studies have shown that photoacoustic imaging can be used in vivo for tumor angiogenesis monitoring, blood oxygenation mapping, functional brain imaging, skin melanoma detection, methemoglobin measuring, etc. Two types of photoacoustic imaging systems, photoacoustic/thermoacoustic computed tomography (also known as photoacoustic/thermoacoustic tomography, i.e., PAT/TAT) and photoacoustic microscopy (PAM), have been developed. A typical PAT system uses an unfocused ultrasound detector to acquire the photoacoustic signals, and the image is reconstructed by inversely solving the photoacoustic equations. A PAM system, on the other hand, uses a spherically focused ultrasound detector with 2D point-by-point scanning, and requires no reconstruction algorithm. Photoacoustic computed tomography

General equation
Given the heating function , the generation and propagation of photoacoustic wave pressure  in an acoustically homogeneous inviscid medium is governed by

where  is the speed of sound in medium,  is the thermal expansion coefficient, and  is the specific heat capacity at constant pressure. Eq. (1) holds under thermal confinement to ensure that heat conduction is negligible during the laser pulse excitation. The thermal confinement occurs when the laser pulsewidth is much shorter than the thermal relaxation time. The forward solution of Eq. (1) is given by

In stress confinement, which occurs when the laser pulsewidth is much shorter than the stress relaxation time, Eq. (2) can be further derived as

where  is the initial photoacoustic pressure. Universal reconstruction algorithm
In a PAT system, the acoustic pressure is detected by scanning an ultrasonic transducer over a surface that encloses the photoacoustic source. To reconstruct the internal source distribution, we need to solve the inverse problem of equation (3) (i.e. to obtain ). A representative method applied for PAT reconstruction is known as the universal backprojection algorithm. This method is suitable for three imaging geometries: planar, spherical, and cylindrical surfaces. The universal back projection formula is

where  is the solid angle subtended by the entire surface  with respect to the reconstruction point  inside , and

Simple system
A simple PAT/TAT/OAT system is shown in the left part of Fig. 3. The laser beam is expanded and diffused to cover the whole region of interest. Photoacoustic waves are generated proportional to the distribution of optical absorption in the target, and are detected by a single scanned ultrasonic transducer. A TAT/OAT system is the same as PAT except that it uses a microwave excitation source instead of a laser. Although single-element transducers have been employed in these two systems, the detection scheme can be extended to use ultrasound arrays as well. Biomedical applications
Intrinsic optical or microwave absorption contrast and diffraction-limited high spatial resolution of ultrasound make PAT and TAT promising imaging modalities for wide biomedical applications:

Brain lesion detection
Soft tissues with different optical absorption properties in the brain can be clearly identified by PAT. Hemodynamics monitoring
Since HbO2 and Hb are the dominant absorbing compounds in biological tissues in the visible spectral range, multiple wavelength photoacoustic measurements can be used to reveal the relative concentration of these two chromophores. Thus, the relative total concentration of hemoglobin (HbT) and the hemoglobin oxygen saturation (sO2) can be derived. Therefore, cerebral hemodynamic changes associated with brain function can be successfully detected with PAT. Breast cancer diagnosis
By utilizing low scattered microwave for excitation, TAT is capable of penetrating thick (several cm) biological tissues with less than mm spatial resolution. Since cancerous tissue and normal tissue have about the same responses to radio frequency radiation, TAT has limited potential in early breast cancer diagnosis. Photoacoustic microscopy

The imaging depth of photoacoustic microscopy is mainly limited by the ultrasonic attenuation. The spatial (i.e. axial and lateral) resolutions depend on the ultrasonic transducer used. An ultrasonic transducer with high central frequency and broader bandwidth are chosen to obtain high axial resolution. The lateral resolution is determined by the focal diameter of the transducer. For instance, a 50 MHz ultrasonic transducer provides 15 micrometre axial and 45 micrometre lateral resolution with ~3 mm imaging depth. Photoacoustic microscopy has multiple important applications in functional imaging: it can detect changes in oxygenated/deoxygenated hemoglobin in small vessels. Other applications 
Photoacoustic imaging was introduced recently in the context of artwork diagnostics with emphasis on the uncovering of hidden features such as underdrawings or original sketch lines in paintings. Photoacoustic images, collected from miniature oil paintings on canvas, illuminated with a pulsed laser on their reverse side, revealed clearly the presence of pencil sketch lines coated over by several paint layers. Advances in photoacoustic imaging 
Photoacoustic imaging has seen recent advances through the integration of deep learning principles and compressed sensing. For more information on the deep learning applications in photoacoustic imaging, see Deep learning in photoacoustic imaging. See also
Multispectral optoacoustic tomography
Photoacoustic microscopy
Deep learning in photoacoustic imaging
Photoacoustic effect

References

External links
Recent advances in application of acoustic, acousto-optic and photoacoustic methods in biology and medicine

Medical imaging