Mass flow meter
The mass flow meter does not measure the volume per unit time (e.g., cubic meters per second) passing through the device; it measures the mass per unit time (e.g., kilograms per second) flowing through the device. Volumetric flow rate is the mass flow rate divided by the fluid density. If the density is constant, then the relationship is simple. If the fluid has varying density, then the relationship is not simple. The density of the fluid may change with temperature, pressure, or composition, for example. The fluid may also be a combination of phases such as a fluid with entrained bubbles.
Actual density can be determined due to dependency of sound velocity on the controlled liquid concentration.
DM3 density meter
The DM3 weighs media (slurry) as it travels through a metered section of pipe using a patented, high resolution load cell. Using linear, direct sensing of mass per unit volume, measurement system within a reinforced rubber tube. This in-line section of pipe calculates % dry solids, wet density and wet or dry mass flow units. When coupled with compatible volumetric flow meters and processing units, this system gives a true value for density and mass flow. The DM3 is a bi-directional unit which allows media to flow continuously through either ends. "C" Clamps located within the DM3 allows the flow tube to withhold high levels of pressure encountered. The flow tube was designed to focus on the highest amount of accuracy, resolution, and least amount of weight without compromising mechanical stability and reliability. Because of the sensitivity of measurement provided by DM3, each unit must be programmed and calibrated to accommodate the variation in Earth's gravitational force.