Mass flow meter
The mass flow meter does not measure the volume per unit time (e.g., cubic meters per second) passing through the device; it measures the mass per unit time (e.g., kilograms per second) flowing through the device. Volumetric flow rate is the mass flow rate divided by the fluid density. If the density is constant, then the relationship is simple. If the fluid has varying density, then the relationship is not simple. The density of the fluid may change with temperature, pressure, or composition, for example. The fluid may also be a combination of phases such as a fluid with entrained bubbles.
Actual density can be determined due to dependency of sound velocity on the controlled liquid concentration.
Beer measurement
The most common method measuring the density of a liquid is with a hydrometer; hydrometers can be calibrated with a number of scales. A common scale is that of specific gravity (SG); that is to say the density of a liquid relative to the density of pure water (at a standard temperature). Specific gravity can also be measured by a pycnometer or oscillating U-tube electronic meter. Water has a SG of 1.000, absolute alcohol has a SG of 0.789. Other density scales are discussed below.