There were two versions of the Oudin coil:    Although it doesn't include a capacitor, the secondary winding is also a resonant circuit (electrical resonator); the parasitic capacitance between the ends of the secondary coil resonates with the large inductance of the secondary at a particular resonant frequency. When it is excited at this frequency by the primary, large oscillating voltages are induced in the secondary. The number of turns in the primary winding, and thus the resonant frequency of the primary, could be adjusted with a tap on the coil. When the two tuned circuits are adjusted to resonate at the same frequency, the large turns ratio of the coil, aided by the high Q of the tuned circuits, steps up the primary voltage to hundreds of thousands to millions of volts at the secondary. The secondary is directly connected to the primary circuit, which carries lethal low frequency 50/60 Hz currents at thousands of volts from the power transformer. Since the Oudin coil was a medical device, with the secondary current applied directly to a person's body, for safety the Oudin circuit has "two" capacitors "(C)", one in each leg of the primary, to completely isolate the coil and output electrode from the supply transformer at the mains frequency. Because two identical capacitors in series have half the capacitance of a single capacitor, the resonant frequency of the Oudin circuit is    The high voltage terminal of the coil was attached through a wire to various types of handheld electrode which the physician used to apply the high voltage to the patient's body. The treatment was not painful for the patient, because alternating current in the radio frequency (RF) range, above 10 kHz in frequency, does not generally cause the sensation of electric shock. The Oudin coil was a "unipolar" generator, with the lower end of the coil grounded, so sometimes only one electrode was applied to the patient and the return path for the currents was through the ground. However usually a ground wire from the bottom of the coil was used; attached to a ground electrode which the patient held. A drawback of the Oudin coil was that movement of the electrode and wire during use changed the capacitance of the top terminal of the secondary coil, and thus its resonant frequency. This threw the secondary coil out of resonance with the primary, causing a reduction in voltage. So the tap point on the primary coil had to be constantly adjusted during use to keep the primary and secondary in "tune" "(seen in left image above)".