_:n67d3f1e83d334962a3233fad4a2bd9d6xb0
<http://schema.org/description>
"<p style=\"text-align: center;\"><span style=\"font-size: 18px;\"><strong>R</strong><strong>eal</strong><strong>-time PCR Detection Kit for Classic Swine Fever Virus, Porcine Reproductive and Respiratory Syndrome </strong><strong>and</strong><strong> Pseudorabies Virus</strong></span></p><p><br></p><p><strong>[Product Name] </strong>Real-time PCR Detection Kit for Classic Swine Fever Virus, Porcine Reproductive and Respiratory Syndrome and Pseudorabies Virus.</p><p><strong>[Package] </strong>50 Tests</p><p><strong>[Indication] </strong>The Real-time PCR Detection Kit for Classic Swine Fever Virus, Porcine Reproductive and Respiratory Syndrome and Pseudorabies Virus is applicable to detect the nucleic acid of CSFV/PRRSV/PRS in tonsil, lymph, saliva, blood and semem samples. The test results are for research purpose only and not for clinical diagnosis.</p><p>[<strong>Main components and content</strong>]</p><table><tbody><tr class=\"firstRow\"><td><p>Name</p></td><td><p>Specification</p></td><td><p>Quantity</p></td></tr><tr><td><p>CSFV/PRRSV/PRS Reaction Solution</p></td><td><p>1000µl/Tube</p></td><td><p>1</p></td></tr><tr><td><p>CSFV/PRRSV/PRS Positive Control</p></td><td><p>250µl/Tube</p></td><td><p>1</p></td></tr><tr><td><p>Negative Control</p></td><td><p>250µl/Tube</p></td><td><p>1</p></td></tr></tbody></table><p>[<strong>Storage and Shelf Life</strong>]</p><p>Stored at -20±5℃, Repeated freezing and thawing ≤ 3 times, the shelf life is 12 months.</p><p>[<strong>Test Method</strong>]</p><p>1. <strong>Nucleic Acid Extraction</strong></p><p>A commercialized DNA/RNA extraction kit can be carried out for nucleic acid extraction, please follow the instruction of the kit.</p><p>2. <strong>PCR amplification</strong></p><p>2.1 Calculate the number of test samples, take n+2 PCR reaction tubes, and add <strong>2</strong><strong>0</strong><strong>µl</strong> to of reaction solution to each tube.</p><p>2.2 Add <strong>5µl</strong> nucleic acid of negative control, positive control and samples into the above PRC reaction tubes respectively, centrifuge at 8000rpm for several seconds, and put them into the thermal cycler.</p><p>2.3 The reaction conditions are set as follows:</p><table><tbody><tr class=\"firstRow\"><td colspan=\"4\"><p>Relevant parameters of the amplifier</p></td></tr><tr><td><p>System</p></td><td colspan=\"3\"><p>Total volume: 30µl</p></td></tr><tr><td rowspan=\"3\"><p>Signal collection</p></td><td colspan=\"3\"><p>Classic Swine Fever Virus (CSFV) - FAM channel collects fluorescence signal</p></td></tr><tr><td colspan=\"3\"><p>Porcine Reproductive and Respiratory Syndrome (PRRS) – HEX/VIC channel collects fluorescence signal</p></td></tr><tr><td colspan=\"3\"><p>Pseudorabies Virus (PRV) - ROX channel collects fluorescence signal</p></td></tr><tr><td rowspan=\"5\"><p><br></p><p><br></p><p>PCR reaction conditions</p></td><td><p>Stage</p></td><td><p>Condition</p></td><td><p>Cycle number</p></td></tr><tr><td><p>Reverse Transcription</p></td><td><p>50℃: 5 minutes</p></td><td><p>1</p></td></tr><tr><td><p>Predegeneration</p></td><td><p>95℃: 30 seconds</p></td><td><p>1</p></td></tr><tr><td rowspan=\"2\"><p><br></p><p>PCR</p></td><td><p>95℃: 10 seconds</p></td><td rowspan=\"2\"><p><br></p><p>40</p></td></tr><tr><td><p>60℃: 30 seconds</p><p>(Set to collect fluorescent signal at the end of this stage)</p></td></tr></tbody></table><p><br></p><p>[<strong>Interpretation of the results</strong>]</p><p>1. Determination of test kit effectiveness:</p><p>(1) Positive control: Ct value of FAM, HEX/VIC and ROX channels ≤ 32, amplification curve with obvious exponential phase.</p><p>(2) Negative control: FAM, HEX/VIC and ROX channels have no amplification curve, or the amplification curve is straight or slight oblique.</p><p>2. Determination of the results:</p><table><tbody><tr class=\"firstRow\"><td><p>Result Judgement</p></td><td><p>FAM channel</p></td><td><p>HEX/VIC channel</p></td><td><p>ROX channel</p></td></tr><tr><td><p>CSFV acid positive</p></td><td><p><strong>+</strong></p></td><td><p><strong>-</strong></p></td><td><p><strong>-</strong></p></td></tr><tr><td><p>PRRSV nucleic acid positive</p></td><td><p><strong>-</strong></p></td><td><p><strong>+</strong></p></td><td><p><strong>-</strong></p></td></tr><tr><td><p>PRV nucleic acid positive</p></td><td><p><strong>-</strong></p></td><td><p><strong>-</strong></p></td><td><p><strong>+</strong></p></td></tr><tr><td><p>CSFV and PRRSV nucleic acid positive</p></td><td><p><strong>+</strong></p></td><td><p><strong>+</strong></p></td><td><p><strong>-</strong></p></td></tr><tr><td><p>CSFV and PRV nucleic acid positive</p></td><td><p><strong>+</strong></p></td><td><p><strong>-</strong></p></td><td><p><strong>+</strong></p></td></tr><tr><td><p>PRRSV and PRV nucleic acid positive</p></td><td><p><strong>-</strong></p></td><td><p><strong>+</strong></p></td><td><p><strong>+</strong></p></td></tr><tr><td><p>CSFV, PRRSV and PRV nucleic acid positive</p></td><td><p><strong>+</strong></p></td><td><p><strong>+</strong></p></td><td><p><strong>+</strong></p></td></tr><tr><td><p>CSFV, PRRSV and PRV nucleic acid negative</p></td><td><p><strong>-</strong></p></td><td><p><strong>-</strong></p></td><td><p><strong>-</strong></p></td></tr></tbody></table><p>*Note:</p><p>If there is an Exponential growth amplification curve and Ct value ≤ 36, it is determined as “<strong>+</strong>”.</p><p>If there is no amplification curve, it is determined as “<strong>-</strong>”.</p><p>If 36 &lt; Ct Value &lt; 40, it is a <strong>doubtful sample</strong>, and the analysis should be repeated for confirmation.</p><p>[<strong>Precautions</strong>]</p><p>1. The laboratory management shall strictly follow the management specification of PCR gene amplification laboratory. The laboratory personnel must be trained professionally. The experiment process shall be conducted strictly in different areas (Reagent preparation area, specimen preparation area, amplification and product analysis area). All consumables shall be disposable after sterilization. Special apparatus, equipment and supplies at each stage of the experiment operation shall not be cross used.</p><p>2. Please prepare the biological safety cabinet for the reagent and specimen preparation stage. The Lab coat, disposable gloves and pipette shall be carried out during the experiment.</p><p>3. Repeated freezing&amp;thawing of reagents shall be avoided as far as possible. Before use, the reagents shall be completely thawed and centrifuged at 8000rpm for several seconds.</p><p>4. Please put the pipette used in the specimen preparation area into the container containing disinfectant and discard with the waste after sterilization.</p><p>5. After the experiment, the worktable and pipette shall be treated with 10% hypochlorite or 75% alcohol or ultraviolet lamp.</p><p><br></p><p>[<strong>Manufacture</strong>]</p><p><strong>Name: </strong>Shandong Xinda Gene Technology Co., Ltd</p><p><strong>A subsidiary of the Shandong Sinder Technology Co., Ltd </strong></p><p><strong>Address</strong>: Building B2, Bandaohuigu Industrial Park, Shungeng Road, Zhucheng City, Shandong Province</p><p><strong>Post Code: 262233</strong></p><p><strong>Phone: +86 - 0532 5882 0810</strong></p><p><span style=\"font-size: 18px;\"> </span></p><p><span style=\"font-size: 18px;\"> </span></p><p><br></p>"
<https://www.sinderanimalhealth.com/CSFV-PRRSV-PRV-PCR-pd785738198.html>