_:nd4c51f8209ff4828a77ae5bca68f76f3xb0
<http://schema.org/description>
"Description\tSaleae Logic Pro 8 Logic Analyzer 8 Channels &amp; 100MHz (Red)\tColor: Red\tOffers an 8-channel logic analyzer\tGreat for debugging embedded applications\tPowerful, easy-to-use software\tFour Data Export Formats: CSV, Binary, VCD, and MATLAB\tCross Platform: Windows, Linux, and OSXThe Saleae Logic Pro 8 Logic Analyzer 8 Channels &amp; 100MHz (Red) is an 8-channel logic analyzer with each input dual-purpose for analog data recording. The device connects to a PC over USB and uses the Saleae Logic Software to record and view digital and analog signals.A logic analyzer is a debugging tool used to record and view digital signals. It operates by sampling a digital input connected to a device under test (DUT) at a high sample rate. These samples are recorded in a sample buffer, and at the end of the capture, the buffer is displayed in the software for review.Logic analyzers are great for debugging embedded applications. In the most common case, a developer working on firmware for a microcontroller will write code to communicate with another component, possibly using protocols like serial, I2C, or SPI.To verify the functionality or to diagnose errors in the firmware, a logic analyzer is connected to the digital IO used for communication and records the activity during testing. The recording is then shown on the display so the user can view the actual behavior of the firmware, and compare that with the expected behavior to narrow down and identify the source of the issue - or verify that the operation is correct.Applications\tFirmware Debugging\tFPGA Debugging\tFunctional Verification\tPerformance Profiling\tReverse Engineering\tProtocol Decoding\tData LoggingView moreWhat&#39;s Included\t1x Saleae Logic Pro 8 USB Logic Analyzer\t2x 4-Channel Wire Harness\t16x Micro-Gripper Hooks\t1x Saleae Carrying Case\t1x USB 3.0 cable\t1x Getting Started GuideDimensionsSize\t53 x 53 x 12 mm / 2.1 x 2.1 x 0.46 inchLinksPDF File\tSaleae Logic Pro 8 Logic Analyzer 8 Channels &amp;amp; 100MHz (Red) - DatasheetSpecifications\tColor: Red\tUSB Type: USB 3.0Inputs\tMulti-Use (A/D/Both): 8\tInput Resistance: 2 MΩ\tInput Capacitance: 10 pF\tInput Protection: &amp;plusmn; 25 VDigital\tSample Rate (max): 500 MS/s\tFastest Digital Signal: 100 MHz\tSample Depth (typical): 10+ Billion\tSupported Logic Levels: 1.2 V - 5.5 V\tWorks with RS-232, 422/3, 485: Yes, directly\tThreshold Voltages: Selectable 6 V / 0.9 V / 1.65 VAnalog\tSample Rate (max): 50 MS/s\tBandwidth (-3 dB): 5 MHz\tNumber of bits: 12 bits\tInput Voltage Range: -10 V to 10 V\tAttenuation at Nyquist: -50 dB or better\tChannel-Channel Crosstalk: -50 dB or betterPhysical\tWeight: 60 g"
<https://jp.robotshop.com/en/products/saleae-logic-pro-8-logic-analyzer-8-channels-100mhz-red>