_:n2fda4de5a23d4723b1700006542a1886xb0
<http://schema.org/description>
"\n\n\n\nParameter\n\n\nSpecifications\n\n\n\n\nModel\n\n\nMultitrip Green - Multi Use Temperature Data Logger\n\n\n\n\nMemory\n\n\n8K Non-volatile\n\n\n\n\nSample Capacity\n\n\nApprox. 7,935 samples minimum\n\n\n\n\nSample Period\n\n\nProgrammable between 2 second and 36 hours in 2 second increments\n\n\n\n\nStart Delay\n\n\nProgrammable between 10 seconds and 236 hours\n\n\n\n\nStart and Stop\n\n\nLogger Button (if enabled), or TRW Software and a Temprecord Reader Interface\n\n\n\n\nRecording Duration\n\n\n4.4 hrs up to 2 years\n\n\n\n\nOperating temperature range of case\n\n\n-25 ˚C to +60 ˚C or -13 ˚F to +140 ˚F\n\n\n\nOperating temperature range with standard probe\n\n-50 ˚C to +110 ˚C or -58 ˚F to +230 ˚F\n\n\n\n\n*Temperature Accuracy\n\n\n0.2 ˚C / 0.35 ˚F over the range -20 ˚C to +50 ˚C or -4 ˚F +122 ˚F, else 0.5 ˚C / 0.9 ˚F\n\n\n\n\nDisplay Resolution\n\n\n0.01 ˚C or 0.01 ˚F\n\n\n\n\nWithin Limits Indicator\n\n\nOne flash from green LED indicator every 10 seconds\n\n\n\n\nOut of Limits Indicator\n\n\nOne flash from red LED indicator every 10 seconds\n\n\n\n\nBattery Type\n\n\nManganese Dioxide Lithium (3.0V)\n\n\n\n\nLogger Dimensions\n\n\nL 129mm x W 73.5mm x 14mm or 3 x 2 7/8 x 5/8 inches\n\n\n\n\nWeight\n\n\n81g or 2.85oz\n\n\n\n\nCase Colour\n\n\nGreen case\n\n\n\n\nCase Protection\n\n\nIP67 or NEMA 6. Fully welded FDA approved food grade casing\n\n\n\n\nWarranty Period\n\n\n12 months on data logger and 2 years on battery from date of purchase. Warranty void if traces of water found within logger case\n\n\n\n\n*Accuracy is calibrated with a traceable reference thermometer to 0.2 ˚C (0.35 ˚F) at temperatures -15 ˚C, 0 ˚C and +40 ˚C"
<https://shop.temprecord.com/collections/calibrated-non-display-data-loggers/products/multitrip-data-logger>