_:nd6f50490180b47f4943ca6b0f2a13cb5xb0
<http://schema.org/description>
"Nickel- and beryllium-free Co-Cr-W-Mo ceramic bonding and casting alloy for the crown and bridge technique.      \tCombibond&reg; BST Triumph exhibits outstanding biocompatibility and corrosion resistance. Combibond&reg; BST Triumph&nbsp;therefore represents an excellent alternative to all precious metal bonding and casting alloys for removable dentures. Combibond&reg; BST Triumph has a fine flow capacity and can be easily cast in the flame, centrifugal and vacuum pressure casting methods. Combibond&reg; BST Triumph can be finished and polished easily and has a bright, silvery surface. Combibond&reg; BST Triumph is suitable for laser welding.  \t  \tComposition %:  \t  \t  \t\t  \t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tCo ~ 60  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tCr 25  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tW 9,0  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tNb 2,0  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tV 1,0  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tMo 1,0  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tSi 1,0  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tFe, Mn &lt; 1  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\tC &lt; 0,1  \t\t\t\t  \t\t\t\t  \t\t\t  \t\t  \t    \t  \t\tTechnical data:  \t\t  \t\t  \t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tEN ISO 9693/EN ISO 22674  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tType 5  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tRP 0,2 [N/mm 2 ] *  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t510  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tTensile strength [N/mm 2 ] *  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t790  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tElongation [A5 %] *  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t&gt; 5  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tVickers hardness HV 10  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t&gt; 280  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tDensity [g/cm 3 ]  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t8,4  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tCTE (20-600 &deg;C) [&micro;/K]  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t14,2  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tMelting range [&deg;C]  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t1285-1380  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tOxidation temperature. [&deg;C]  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\t960/vac. (2 min.)  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tSupply form  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t\t\tCylinder  \t\t\t\t\t  \t\t\t\t\t  \t\t\t\t  \t\t\t  \t\t  \t\t* data are valid for room temperature  \t  "
<https://leickert-dental.de/en/nonprecious-metal.html>