In this study, the level of oxygen within the blood will be reduced in a controlled manner by reducing the concentration of oxygen the participant breathes. The accuracy of a noninvasive pulse oximeter sensor will be assessed by comparison to the oxygen saturation measurements from a laboratory blood gas analyzer.~It is required that the Accuracy Root Mean Square (ARMS) performance of the Stryker pulse oximetry sensors will meet a specification of +/-3% or better in non-motion conditions for the range of 70 - 100% SaO2 (typically, saturation is determined once with air breathing and then at one of 6 levels, e.g. 94%, 90%, 85%, 80%, 75% and 70% saturation for about 30-60 seconds at each level), thereby demonstrating an acceptable SpO2 accuracy performance specification.
The purpose of this clinical study is to validate the SpO2 accuracy of the Stryker Sustainability Solutions pulse oximetry sensors during non-motion conditions over the range of 70-100% SaO2 as compared to arterial blood samples assessed by CO-Oximetry. The end goal is to provide supporting documentation for the SpO2 accuracy validation of the reprocessed sensors.