en en de ga fi en ga en ga ga ru ga ga gd ga ga ga ga ga vi en en en en en en et es en ga es en ga ga ga en ga en ga el tr en sv fr de en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en fa en en en en en en en en en en ga en ga et ga en hu en vi en en en vi en en en en en en en en en ja en en en en en ar en en en en en en en vi en en en ro en en en vi en en en ro en en id la ga en ga ga ga ga ga en ca ga ga en en ga ga en en ast en la en de en en en ga en nb en en en en ga ga ga ga ga ga ca ga ga nl ga ga ga ga en
IMI04169, Red Lion Controls IMI04169 Dáileacháin - Electronics-ic.com
Fáilte go dtí Electronics-ic.com    E-mail: info@electronics-ic.com
Féach ar an Leagan Béarla mar ár Official Version.
Brabhsáil de réir SoláthróirBrabhsáil de réir Táirge
Iontráil cuid uimhir iomlán nó páirteach monaróir le ar a laghad de 3 litreacha nó uimhreacha
Is íomhá le haghaidh tagartha amháin.
Request Price & Lead Time
6-DIGIT, 0.56" (14.2 mm) HIGH RED LED DISPLAY
ACCEPTS INPUT RATES UP TO 50 KHz
SECONDARY DISPLAY INDICATES TOTALIZATION or
DUAL ALARM RELAY OUTPUTS (Optional)
4 to 20 mA or 0 to 10 VDC ANALOG OUTPUT (Optional)
The Apollo Intelligent Meter for Digital Rate Inputs (IMI) accepts
frequencies up to 50 KHz which can be scaled to any desired engineering units.
The sensor input is user configurable and allows for a wide selection of
compatible sensors. The adjustable minimum and maximum update times
provide optimal display response at any input frequency.
The IMI provides two display functions in a single package. The display can
indicate the input rate and then be toggled to display either totalization or
efficiency. The max. display for all functions is 999,999. The totalizer features
independent scaling and a low signal cut-out to suit a variety of applications.
The efficiency feature can display the input rate in percent of target (process
efficiency). Additionally, nine segments can easily be programmed to linearize
transducers with non-linear outputs, such as square law devices.
The indicator features a choice of two different scaling procedures for the rate
display, which greatly simplifies initial set-up. English-style display prompts
and front panel buttons aid the operator in set-up and operation. A front panel
programmable lock-out menu used with the program disable input protects set-
up data and guarantees no unwanted changes occur during operation.
Programmable digital filtering enhances the stability of the reading.
Programmable remote inputs “E1-CON” and “E2-CON” can be used to control
a variety of functions, such as totalizing, alarm control, display hold or triggered
input. All set-up data is stored in nonvolatile memory.
Peak/valley (max/min) indication is included and is easily recalled and
controlled by either the front panel buttons or a remote input. The Peak/Valley
buffers can be assigned to either the rate or efficiency display and all readings
are retained at power-down.
Optional dual relays that operate in parallel with solid state outputs are fully
programmable to operate in a wide variety of modes to suit many control or
The optional 20 mA current loop, half-duplex serial communications,
provides computer and printer interfacing to extend the capabilities of the
indicator. More than one unit can be connected in the loop with other RLC
products which have serial communications capabilities.
An optional 4 to 20 mA or 0 to 10 VDC analog output can be scaled by the
user to interface with a host of recorders, indicators and controllers. The
indicator has several built-in diagnostic functions to alert operators of most
malfunctions. Extensive testing of noise interference mechanisms and full burn-
in make the indicator extremely reliable in industrial environments. The die-cast
front bezel meets NEMA 4/IP65 requirements for washdown applications, when
properly installed. Plug-in style terminal blocks simplify installation and wiring
All safety related regulations, local codes and instructions that appear in the
manual or on equipment must be observed to ensure personal safety and to
prevent damage to either the instrument or equipment connected to it. If
equipment is used in a manner not specified by the manufacturer, the protection
provided by the equipment may be impaired.
Do not use this unit to directly command motors, valves, or other actuators
not equipped with safeguards. To do so, can be potentially harmful to persons
or equipment in the event of a fault to the unit.
1. DISPLAY: 6-digit, 0.56" (14.2 mm) High Red LED, programmable decimal
points, maximum display 999,999. Flashing display during totalizer
overflow. “———” displayed during rate/efficiency display out of range.
A.C. Power: Switch selectable 115/230 VAC, ±10%, 50/60 Hz, 14 VA
3. CONTROLS: Three front panel push buttons for modifying alarm values
and indicator set-up. Three external inputs; one for disabling the front panel,
two for programmable function inputs.
4. SIGNAL INPUT: Switch selectable to accept signals from a wide variety of
Max. Input Frequency: 50 KHz, 50% Duty Cycle.
Max. Input Voltage & Current:
With NPN O.C. switch OFF; ±50 V peak, ±5 mA.
Input Impedance: With MAG. PKUP. & NPN O.C. switches OFF; 1 MΩ.
MAG PKUP: Connects a 0.1 µF input damping capacitor.
Sensitivity: 125 mV peak (typical @ 12 VDC)
Input Impedance: 26.5 KΩ@ 60 Hz
Max. Input Voltage: ±50 V peak
MODEL IMI - APOLLO 6 DIGIT INTELLIGENT METER FOR RATE INPUTS
DIMENSIONS In inches (mm) Note: Recommended minimum clearance (behind the panel) for mounting
clip installation is 2.1" (53.3)H x 5.5" (140)W. PANEL CUT-OUT
Tel +1 (717) 767-6511
Fax +1 (717) 764-0839
IMI04169 Achoimre na n-ábhar
1  6-DIGIT , 0.56" (14.2 mm) HIGH RED LED DISPLA Y  ACCEPTS INPUT RA TES UP T O 50 KHz  SECONDAR Y DISPLA Y INDICA TES TOT ALIZA TION or EFFICIENCY  0.01% ACCURACY  PEAK/V ALLEY INDICA TION & MEMORY  PROGRAMMABLE FRONT P ANEL LOCK-OUT  SENSOR EXCIT A TION SUPPL Y  DUAL ALARM RELA Y OUTPUTS (Optional)  BI-DIRECTIONAL SERIAL COMMUNICA TIONS (Optional)  4 to 20 mA or 0 to 10 VDC ANALOG OUTPUT (Optional)  NEMA 4/IP65 SEALED MET AL FRONT BEZEL GENERAL DESCRIPTION The Apollo Intelligent Meter for Digital Rate Inputs (IMI) accepts frequencies up to 50 KHz wh
2 SPECIFICA TIONS (Cont’d) 4. SIGNAL INPUT : (Cont’d) LOGIC : Sets input trigger level. T rigger Level : +2.5V with 30 mV hysteresis. NPN O.C. : Provides internal 3.9 K Ω pull-up resistor to +12 VDC. I SNK = 3 mA MAX @ 12 VDC V MAX IN = 28 VDC 5. DISPLA Y COMBINA TIONS: Rate/T otal or Rate/Efficiency . 6. RA TE/EFFICIENCY : Accuracy : 0.01% ±1 digit Min. Input Frequency : 0.01 Hz Programmable Update Time Range : 0.2 secs to 100.0 secs 7. TOT ALIZER: Programmable time-base, scale factor (0.001 to 100.000) and low-end cutout. Max. response time 0.2 secs. 8. PROGRAM DISABLE (PGM.DIS.): Us
3 PROGRAMMABLE FUNCTIONS Programming of the indicator is divided into modular steps. Each module is a short sequence of data entries. The front panel “UP” and “DOWN” buttons (shown as “arrows” on the front panel) are used to change the data and set-ups, while the “P” button is used to save or enter the data. After pressing “P”, the user gains entry into the programming mode. The programming modules are identified by the message “Pro” and a module number in the display . “UP” and “DOWN” are used to select the desired programming module and “P” is used to ent
PROGRAMMING EXAMPLE An IMI is used to monitor web speed of a textile machine. A display of the web speed in FPM and efficiency is also desired. Underspeed and overspeed protection is required in addition to an analog output to a chart recorder . The optimum speed of the web is 1 100 FPM. The absolute maximum speed equals 120% of the optimum speed and the absolute minimum speed equals 70% of the optimum speed. If the web speed exceeds these limits, the appropriate alarm will latch and shut-down the machine. The “on” delay feature is used for both setpoints to allow the textile machine five
páirteanna gaolmhara do IMI04169
W/EXC ALARMS SERIAL & V ANLG
Murata Power Solutions Inc
AMMETER LED 8-36VDC 50A/500A RED
LCD DPM +9V 200MV 4.5 DIGIT
Murata Power Solutions Inc
DPM LED 20VDC 4.5DIGIT LP RED
Omron Electronics Inc-IA Div
ROTARY PULSE INDICATR NPN/V INPT
RATE METER 6DIGIT 115/230VAC
W/ALARMS SERIAL & ANALOG
2017-03-24Cambridge IC Nochtann sonraí sensing, ar an 10 lá breithe
2017-03-21Déantúsaíocht Deiscirt: Abair hello Leictreonaic Seachta...
2017-03-21Saor in Aisce DO-178C / ED-12C lámhleabhar
2017-03-201Gsample TiePie s, tá 4 scóip Cainéal cumas difreálach
2017-03-20Analog Devices Múineann analógach agus RF leictreonaic
2017-03-17Video: Leabaithe Domhain 2017, an forbhreathnú
2017-03-17Ainmneacha comhlacht slándála IoT chathaoirleach nua
2017-03-17Harwin méaduithe fócas aeraspáis
2017-03-16EW: Agallamh Video - Technologies Keysight ar Ascalascóip ...
2017-03-16dul chun cinn James Webb teileascóp i dtreo sheoladh
2017-03-16EW: Agallamh Video - Córais QA ar Cantata aonaid agus uirl...
2017-03-15EW: feisteáin Dev ngleic éagsúlacht IoT
eochairfhocail a bhaineann le IMI04169
Red Lion Controls IMI04169
Cuardaigh Chuid Uimhir dhíth ort anseo:
Iontráil cuid uimhir iomlán nó páirteach monaróir le ar a laghad de 3 litreacha nó uimhreacha
Ceisteanna Coitianta / Cabhair
Is Déanaí Táirge | Nascóirí, Idirnaisc | Rialuithe Tionscail, Méadair | Comhthionóil cábla | Motors, Solenoids, Boird Tiománaí / Modúil | Lasca Keylock | Ciorcaid Comhtháite (ICS) | Rialuithe Tionscail, Méadair | isolators | Ionduchtóirí, Coils, Chokes | Cosaint Chuarda | trasraitheoirí | Accessories | Trasraitheoirí - Cuspóir Speisialta | Trasraitheoirí (BJT) - Aonair
Is Electronics-ic.com® seirbhís iomlán dáileoir mórdhíola neamhspleách chomhdhéanann í leictreonach. Cuirimid réiteach iomlán agus costas foinsiú éifeachtach Bunaidh Trealamh monaróirí, monaróirí Conartha, Service & Repair Eagraíochtaí, Dáileoirí, Taighde agus Grúpaí Forbartha agus comhlachtaí eile a bhfuil gá a chomhdhéanann í leictreonach.
© 2003-2016 Electronics-ic.com Gach ceart ar cosaint