ga en de ga fi en ga en ga ga ru ga ga gd ga ga ga ga ga oc pt en en de en en et es en ga es en ga ga ga en ga en ga ca sk en ga en cy en ga en ga en tr en sv en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en hu en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en en it es en en en hu en en en en ga en ga et ga en hu en de pt en en oc pt en en fa vi en id la ga en ga ga ga ga ga en ca ga ga en pt ga ga de als hu zh la de de en en nl ga de de en en en en ga ga ga ga ga ga ca ga ga nl ga ga ga ga en
35110, Desco 35110 Dáileacháin - Electronics-ic.com
Fáilte go dtí Electronics-ic.com    E-mail: info@electronics-ic.com
Féach ar an Leagan Béarla mar ár Official Version.
Brabhsáil de réir SoláthróirBrabhsáil de réir Táirge
Iontráil cuid uimhir iomlán nó páirteach monaróir le ar a laghad de 3 litreacha nó uimhreacha
Is íomhá le haghaidh tagartha amháin.
Request Price & Lead Time
Static Control, ESD, Clean Room Products
Figure 1. Soldering Tip Voltage Tester, Item No. 35110.
The Soldering Tip Voltage Tester consists of the circuit
board enclosed in a 2.4" (61mm) x 2.3" (58mm) x 1" (25mm)
high enclosure, 2 AAA batteries (included), an attached
sensor plate, a grounding wire with banana plug, and a pair
Attach the grounding wire banana plug to electrical ground,
such as a common ground point on the workbench or the
center grounding screw of an AC power outlet socket, etc.
Note: Keep in mind that with typical use the Soldering
Tip Voltage Tester will need to have the 2 AAA batteries
replaced every 1 to 2 years. This requires opening up
the enclosure by removing the 2 screws, separating
the enclosure halves, replacing the batteries, and then
Whenever the soldering iron tip contacts the cleaner or
sponge (which is in contact with the sensor plate) one of the
following conditions will occur:
Soldering Tip Voltage Tester Operation Instructions
The MENDA Soldering Tip Voltage Tester’s OK function
may be tested by grounding the sensor plate causing the
green LED to light if there is less than 1.25 Volts AC.
Place the sensor plate under the soldering iron cleaning
sponge or metallic type cleaner pad.
If using a sponge, dampen the sponge as you normally
would. Place the sponge in the holder, being sure it makes
good contact with the sensor plate.
Note: For sponges, best results are with a clean plastic
sponge holder. If the sponge holder is metal, it must not
ground the sensor plate. For metallic cleaner pads, if
the container is conductive it must be isolated from any
grounded surface by rubber feet or other insulators. Find
inappropriate location for the Soldering Tip Voltage Tester
where the LED indicators are easily viewed and the sensor
plate wire can be conveniently routed (be sure to avoid
locations with excessive heat or solder splashes).
Attach the Velcro pieces to hold the Soldering Tip Voltage
Use a Function Generator.
Make the following settings to the Function Generator:
• Set for a SINE wave. (~)
• Set Frequency to about 1 Hz.
• Set Amplitude to 1 volt.
1. Short the spade contact with the plate. The green (lower)
2. Connect the Soldering Tip Voltage Tester to the Function
Generator’s Main connector with the spade lug to the
ground and the plate to the center conductor.
3. The red FAIL LED should light and an audible beep
should sound if there is over 1.25 Volts AC. The alarm
will sound at a rate of about once per second at a
frequency of 1 Hz; the beep will increase in speed along
with the frequency. At about 10 Hz, it will sound as a
TB-4011 July 2011 Page 1 of 2
3651 Walnut Avenue, Chino, CA 91710
Phone: (909) 627-2453 Fax (951) 755-8848 Web Site: MendaTools.com
© 2011 DESCO INDUSTRIES INC.
Green Soldering Iron Tip grounded OK
Soldering Iron Tip not grounded Repair or
Sensor plate not installed or is not Correct the
properly contacting the cleaner problem
No Soldering Tip Voltage Tester Correct the
lights grounding wire not attached to a problem
Soldering Tip Voltage Tester Correct the
batteries need replacement (verify problem
by testing as described above)
Red Soldering Iron Tip has greater Caution - do
light than 1 Volt AC not use Iron on
Repair or replace Iron.
35110 Achoimre na n-ábhar
Figure 1. Soldering Tip Voltage Tester, Item No. 35110. Testing and Installation The Soldering Tip Voltage Tester consists of the circuit board enclosed in a 2.4" (61mm) x 2.3" (58mm) x 1" (25mm) high enclosure, 2 AAA batteries (included), an attached sensor plate, a grounding wire with banana plug, and a pair of Velcro pieces. Attach the grounding wire banana plug to electrical ground, such as a common ground point on the workbench or the center grounding screw of an AC power outlet socket, etc. Note: Keep in mind that with typical use the Soldering Tip Voltage Tester will nee
TB-4011 Page 2 of 2 3651 Walnut Avenue, Chino, CA 91710 Phone: (909) 627-2453 Fax (951) 755-8848 Web Site: MendaTools.com © 2011 DESCO INDUSTRIES INC. Employee Owned Limited Warranty Menda expressly warrants that for a period of one (1) year from the date of purchase, Soldering Tip Voltage Tester will be free of defects in material (parts) and workmanship (labor). Within the warranty period, the product will be tested, repaired, or replaced at Menda option, free of charge. Call our Customer Service Department at 909-627-2453 for a Return Material Authorization (RMA) and proper shipping instr
páirteanna gaolmhara do 35110
PUMP/BOTTLE PURE-TOUCH RND 4OZ
TESTER SOLDERING TIP VOLTAGE
TESTER SHOES/WRIST STRAP CERT
2017-03-24Cambridge IC Nochtann sonraí sensing, ar an 10 lá breithe
2017-03-21Déantúsaíocht Deiscirt: Abair hello Leictreonaic Seachta...
2017-03-21Saor in Aisce DO-178C / ED-12C lámhleabhar
2017-03-201Gsample TiePie s, tá 4 scóip Cainéal cumas difreálach
2017-03-20Analog Devices Múineann analógach agus RF leictreonaic
2017-03-17Video: Leabaithe Domhain 2017, an forbhreathnú
2017-03-17Ainmneacha comhlacht slándála IoT chathaoirleach nua
2017-03-17Harwin méaduithe fócas aeraspáis
2017-03-16EW: Agallamh Video - Technologies Keysight ar Ascalascóip ...
2017-03-16dul chun cinn James Webb teileascóp i dtreo sheoladh
2017-03-16EW: Agallamh Video - Córais QA ar Cantata aonaid agus uirl...
2017-03-15EW: feisteáin Dev ngleic éagsúlacht IoT
eochairfhocail a bhaineann le 35110
Cuardaigh Chuid Uimhir dhíth ort anseo:
Iontráil cuid uimhir iomlán nó páirteach monaróir le ar a laghad de 3 litreacha nó uimhreacha
Ceisteanna Coitianta / Cabhair
Is Déanaí Táirge | Nascóirí, Idirnaisc | Rialuithe Tionscail, Méadair | Comhthionóil cábla | Motors, Solenoids, Boird Tiománaí / Modúil | Lasca Keylock | Ciorcaid Comhtháite (ICS) | Rialuithe Tionscail, Méadair | isolators | Ionduchtóirí, Coils, Chokes | Cosaint Chuarda | trasraitheoirí | Accessories | Trasraitheoirí - Cuspóir Speisialta | Trasraitheoirí (BJT) - Aonair
Is Electronics-ic.com® seirbhís iomlán dáileoir mórdhíola neamhspleách chomhdhéanann í leictreonach. Cuirimid réiteach iomlán agus costas foinsiú éifeachtach Bunaidh Trealamh monaróirí, monaróirí Conartha, Service & Repair Eagraíochtaí, Dáileoirí, Taighde agus Grúpaí Forbartha agus comhlachtaí eile a bhfuil gá a chomhdhéanann í leictreonach.
© 2003-2016 Electronics-ic.com Gach ceart ar cosaint