text: The electrical field of terahertz pulses is measured in a detector simultaneously illuminated with an ultrashort laser pulse. Two common detection schemes are used in THz-TDS: photoconductive sampling and electro-optical sampling. The power of THz pulses can be detected by bolometers (heat detectors cooled to liquid-helium temperatures), but since bolometers can only measure the total energy of a terahertz pulse rather than its electric field over time, they are unsuitable for THz-TDS.
  Because the measurement technique is coherent, it naturally rejects incoherent radiation.  Additionally, because the time slice of the measurement is extremely narrow, the noise contribution to the measurement is extremely low.
  The signal-to-noise ratio (S/N) of the resulting time-domain waveform depends on experimental conditions (e.g., averaging time). However due to the coherent sampling techniques described, high S/N values (>70 dB) are routinely observed with 1 minute averaging times.