text: The earliest measurements combining AFM with infrared spectroscopy were performed in 1999 by Hammiche et al. at the University of Lancaster in the United Kingdom, in an EPSRC-funded project led by M Reading and H M Pollock. Separately, Anderson at the Jet Propulsion Laboratory in the United States made a related measurement in 2000. Both groups used a conventional Fourier transform infrared spectrometer (FTIR) equipped with a broadband thermal source, the radiation was focused near the tip of a probe that was in contact with a sample. The Lancaster group obtained spectra by detecting the absorption of infrared radiation using a temperature sensitive thermal probe. Anderson took the different approach of using a conventional AFM probe to detect the thermal expansion. He reported an interferogram but not a spectrum; the first infrared spectrum obtained in this way was reported by Hammiche et al. in 2004: this represented the first proof that spectral information about a sample could be obtained using this approach.
  Both of these early experiments used a broadband source in conjunction with an interferometer; these techniques could, therefore, be referred to as AFM-FTIR although Hammiche et al. coined the more general term photothermal microspectroscopy or PTMS in their first paper. PTMS has various subgroups; including techniques that measure temperature measure thermal expansion use broadband sources. use lasers excite the sample using evanescent waves, illuminate the sample directly from above etc. and different combinations of these. Fundamentally, they all exploit the photothermal effect. Different combinations of sources, methods, methods of detection and methods of illumination have benefits for different applications. Care should be taken to ensure that it is clear which form of PTMS is being used in each case. Currently there is no universally accepted nomenclature. The original technique dubbed AFM-IR that induced resonant motion in the probe using a Free Electron Laser has developed by exploiting the foregoing permutations so that it has evolved into various forms.
  The pioneering experiments of Hammiche et al and Anderson had limited spatial resolution due to thermal diffusion - the spreading of heat away from the region where the infrared light was absorbed. The thermal diffusion length (the distance the heat spreads) is inversely proportional to the root of the modulation frequency. Consequently, the spatial resolution achieved by the early AFM-IR approaches was around one micron or more, due to the low modulation frequencies of the incident radiation created by the movement of the mirror in the interferometer. Also, the first thermal probes were Wollaston wire devices that were developed originally for Microthermal analysis (in fact PTMS was originally considered to be one of a family of microthermal techniques). The comparatively large size of these probes also limited spatial resolution. Bozec et al. and Reading et al. used thermal probes with nanoscale dimensions and demonstrated higher spatial resolution. Ye et al  described a MEM-type thermal probe giving sub-100 nm spatial resolution, which they used for nanothermal analysis. The process of exploring laser sources began in 2001 by Hammiche et al when they acquired the first spectrum using a tuneable laser (see Resolution improvement with pulsed laser source).
  A significant development was the creation by Reading et al. in 2001 of a custom interface that allowed measurements to be made while illuminating the sample from above; this interface focused the infrared beam to a spot of circa 500μm diameter, close to the theoretical maximum. The use of top-down or top-side illumination has the important benefit that samples of arbitrary thickness can be studied on arbitrary substrates. In many cases this can be done without any sample preparation. All subsequent experiments by Hammiche, Pollock, Reading and their co-workers were made using this type of interface including the instrument constructed by Hill et al. for nanoscale imaging using a pulsed laser. The work of the University of Lancaster group in collaboration with workers from the University of East Anglia led to the formation of a company, Anasys Instruments, to exploit this and related technologies (see Commercialization).