text: The simplest way of measuring the contact angle of a sessile drop is with a contact angle goniometer, which allows the user to measure the contact angle visually. A droplet is deposited by a syringe which is positioned above the sample surface, and a high resolution camera captures the image from the profile or side view. The image can then be analyzed either by eye (with a protractor) or more often is measured using image analysis software. This type of measurement is referred to as a static contact angle measurement.
  The contact angle is affected not only by the surface chemistry but also by the surface roughness. The Young equation, which is the basis for the contact angle, assumes a homogeneous surface with no surface roughness. In case surface roughness is present, the droplet can be in Wenzel state (homogeneous wetting), Cassie-Baxter state (heterogeneous wetting) or in an intermediary state. The surface roughness amplifies the wetting behavior caused by the surface chemistry.
  In order to measure the contact angle hysteresis, the sessile droplet can be increased gradually in volume. The maximum possible contact angle is referred to as the advancing contact angle. The receding contact angle can be measured by removing volume from the drop until dewetting occurs. The minimum possible contact angle is referred to as the receding contact angle. The contact angle hysteresis is the difference between the advancing and receding contact angle.