text: Original ATM techniques involve rotating the sample at the focal point of a linearly polarized THz beam using a mechanically rotated sample mount. For this reason, the configuration is typically a far-field instrument in which a balanced detector (sensitive to infrared light) is placed a considerable distance from the sample. In the terahertz time-domain spectroscopy configuration, both the infrared and THz beams are transmitted through an electro-optic (EO) crystal like ZnTe or GaP. Here, the infrared beam detects the change in birefringence of the EO crystal due to the THz beam. When a sample is placed in the THz beam, the polarized THz beam is perturbed and the resulting degree of birefringence in the EO crystal is changed. The resulting perturbation of the infrared beam is sensed at the balanced detector.
  Rotated sample ATM is very useful for large samples (0.1 to 1 cm). However, when measuring samples such as protein crystals that must be isolated inside a hydration chamber, for example, the sample cannot be easily rotated. Additionally, it is challenging to maintain the same location of a rotated sample at the precise focal point of a THz beam.