text: Put simply, ion suppression describes the adverse effect on detector response due to reduced ionisation efficiency for analyte(s) of interest, resulting from the presence of species in the sample matrix which compete for ionisation, or inhibit efficient ionisation in other ways. Use of MS/MS as a means of detection may give the impression that there are no interfering species present, since no chromatographic impurities are detected. However, species which are not isobaric may still have an adverse effect on the sensitivity, accuracy and precision of the assay owing to suppression of the ionisation of the analyte of interest.
  Although the precise chemical and physical factors involved in ion suppression are not fully understood, it has been proposed that basicity, high concentration, mass and more intuitively, co-elution with the analyte of interest are factors which should not be ignored.
  The most common atmospheric pressure ionisation techniques used in LC-MS/MS are electrospray ionization (ESI) and atmospheric pressure chemical ionization (APCI). APCI is less prone to pronounced ion suppression than ESI, an inherent property of the respective ionisation mechanisms.
  In APCI, the sole source of ion suppression can be attributed to the change of colligative properties in the solute during evaporization (King et al, J. Am. Soc. Mass Spectrom 2000, 11, 942-950).
  ESI has a more complex ionisation mechanism, relying heavily on droplet charge excess and as such there are many more factors to consider when exploring the cause of ion suppression. It has been widely observed that for many analytes, at high concentrations, ESI exhibits a loss of detector response linearity, perhaps due to reduced charge excess caused by analyte saturation at the droplet surface, inhibiting subsequent ejection of gas phase ions from further inside the droplet. Thus competition for space and/or charge may be considered as a source of ion suppression in ESI. Both physical and chemical properties of analytes (e.g. basicity and surface activity) determine their inherent ionisation efficiency. Biological sample matrices naturally tend to contain many endogenous species with high basicity and surface activity, hence the total concentration of these species in the sample will quickly reach levels at which ion suppression should be expected.
  Another explanation of ion suppression in ESI considers the physical properties of the droplet itself rather than the species present. High concentrations of interfering components give rise to an increased surface tension and viscosity, giving a reduction in desolvation (solvent evaporation), which is known to have a marked effect of ionisation efficiency.
  The third proposed theory for ion suppression in ESI relates to the presence of non-volatile species which can either cause co-precipitation of analyte in the droplet (thus preventing ionisation) or prevent the contraction of droplet size to the critical radius required for the ion evaporation and/or charge residue mechanisms to form gas phase ions efficiently.
  It is worthwhile to consider that the degree of ion suppression may be dependent on the concentration of the analyte being monitored. A higher analyte/matrix ratio can give a reduced effect of ion suppression.