text: A Kingdon trap consists of a thin central wire, an outer cylindrical electrode and isolated end cap electrodes at both ends. A static applied voltage results in a radial logarithmic potential between the electrodes. In a Kingdon trap there is no potential minimum to store the ions; however, they are stored with a finite angular momentum about the central wire and the applied electric field in the device allows for the stability of the ion trajectories. In 1981, Knight introduced a modified outer electrode that included an axial quadrupole term that confines the ions on the trap axis. The dynamic Kingdon trap has an additional AC voltage that uses strong defocusing to permanently store charged particles. The dynamic Kingdon trap does not require the trapped ions to have angular momentum with respect to the filament.  An Orbitrap is a modified Kingdon trap that is used for mass spectrometry. Though the idea has been suggested and computer simulations performed neither the Kingdon nor the Knight configurations were reported to produce mass spectra, as the simulations indicated mass resolving power would be problematic.