text: Any real measuring instrument has a limited range over which it can accurately measure absorbance. An instrument must be calibrated and checked against known standards if the readings are to be trusted. Many instruments will become non-linear (fail to follow the Beer–Lambert law) starting at approximately 2 AU (~1% transmission). It is also difficult to accurately measure very small absorbance values (below ) with commercially available instruments for chemical analysis. In such cases, laser-based absorption techniques can be used, since they have demonstrated detection limits that supersede those obtained by conventional non-laser-based instruments by many orders of magnitude (detection has been demonstrated all the way down to ). The theoretical best accuracy for most commercially available non-laser-based instruments is attained in the range near 1 AU. The path length or concentration should then, when possible, be adjusted to achieve readings near this range.