text: The measured full SPR curves (x-axis: angle, y-axis: reflected light intensity) can be transcribed into sensograms (x-axis: time, y-axis: selected parameter such as peak minimum, light intensity, peak width). The sensograms can be fitted using binding models to obtain kinetic parameters including on- and off-rates and affinity. The full SPR curves are used to fit Fresnel equations to obtain thickness and refractive index of the layers. Also due to the ability of scanning the whole SPR curve, MP-SPR is able to separate bulk effect and analyte binding from each other using parameters of the curve. 
  While QCM-D measures wet mass, MP-SPR and other optical methods measure dry mass, which enables analysis of water content of nanocellulose films.