text: The measurement time required for a single impedance measurement point strongly depends on the frequency of interest. While frequencies above about 1 Hz can be measured within seconds, the measurement time increases significantly in the lower frequency range.
  Although the exact duration for measuring a complete impedance spectrum depends on the measauring device as well as on internal settings, the following measurement times can be considered as rules of thumb when measuring the frequency measurement points sequentially, with the upper frequency assumed as 100 kHz or 1 MHz:
  * Down to approx. 1 Hz, the measuring time is approx. 1 minute
  * Down to 0.1 Hz approx. 5 minutes
  * Down to 0.05 Hz approx. 10 minutes
  * Down to 0.02 Hz approx. 15 minutes
  * Down to 0.01 Hz approx. 30 minutes
  Measurements down to or below 0.01 Hz are typically associated with measurement times in the range of several hours. Therefore, a spectrum can be roughly divided into three sub-ranges with regard to the occurrence of artifacts: in the high-frequency domain (approx. > 100 to 1000 Hz), induction or mutual induction can dominate. In the low frequency region (< 1 Hz), drift can occur due to noticeable change in the system. The range between about 1 Hz and 1000 Hz is usually not affected by high- or low-frequency artifacts. However, the mains frequency (50/60 Hz) may come into play as distorting artifact in this region.