text: An MC-ICP-MS instrument is a multiple collector mass spectrometer with a plasma source. MC-ICP-MS was developed to improve the precision achievable by ICP-MS during isotope-ratio measurements.  Conventional ICP-MS analysis uses a quadrupole analyser, which only allows single-collector analysis. Due to the inherent instability of the plasma, this limits the precision of ICP-MS with a quadrupole analyzer to around 1%, which is insufficient for most radiogenic isotope systems.
  Isotope-ratio analysis for radiometric dating has normally been determined by TIMS. However, some systems (e.g. Hf-W and Lu-Hf) are difficult or impossible to analyse by TIMS, due to the high ionization potential of the elements involved. Therefore, these methods can now be analysed using MC-ICP-MS.
  The Ar-ICP produces an ion-beam with a large inherent kinetic energy distribution, which makes the design of the mass-spectrometer somewhat more complex than it is the case for conventional TIMS instruments. First, different from Quadrupole ICP-MS systems, magnetic sector instruments have to operate with a higher acceleration potential (several 1000 V) in order to minimize the energy distribution of the ion beam. Modern instruments operate at 6-10kV. 
  The radius of deflection of an ion within a magnetic field depends on the kinetic energy and the mass/charge ratio of the ion (strictly, the magnet is a momentum analyzer not just a mass analyzer). Because of the large energy distribution, ions with similar mass/charge ratio can have very different kinetic energies and will thus experience different deflection for the same magnetic field. In practical terms one would see that ions with the same mass/charge ratio focus at different points in space. However, in a mass-spectrometer one wants ions with the same mass/charge ratio to focus at the same point, e.g. where the detector is located. In order to overcome these limitations, commercial MC-ICP-MS are double-focusing instruments. In a double-focusing mass-spectrometer ions are focused due to kinetic energy by the ESA (electro-static-analyzer) and kinetic energy + mass/charge (momentum) in the magnetic field. Magnet and ESA are carefully chosen to match the energy focusing properties of one another and are arranged so that the direction of energy focusing is in opposite directions. To simplify, two components have an energy focus term, when arranged properly, the energy term cancels out and ions with the same mass/charge ratio focus at the same point in space. It is important to note, double-focusing does not reduce the kinetic energy distribution and different kinetic energies are not filtered or homogenized. Double-focusing works for single as well as multi-collector instruments. In single collector instruments ESA and magnet can be arranged in either forward geometry (first ESA then magnet) or reversed geometry (magnet first then ESA), as only point-to-point focusing is required. In multi-collector instruments, only forward geometry (ESA then magnet) is possible due to the array of detectors and the requirements of a focal plane rather than a focal point.