text: The above technique can obviously not correct for a background with fine structure, as in this case the absorbance will be different at each of the correction pixels. In this case HR-CS AAS is offering the possibility to measure correction spectra of the molecule(s) that is (are) responsible for the background and store them in the computer. These spectra are then multiplied with a factor to match the intensity of the sample spectrum and subtracted pixel by pixel and spectrum by spectrum from the sample spectrum using a least-squares algorithm. This might sound complex, but first of all the number of di-atomic molecules that can exist at the temperatures of the atomizers used in AAS is relatively small, and second, the correction is performed by the computer within a few seconds. The same algorithm can actually also be used to correct for direct line overlap of two atomic absorption lines, making HR-CS AAS the only AAS technique that can correct for this kind of spectral interference.