text: The columns used in FPLC are large [mm id] tubes that contain small [µ] particles or gel beads that are known as stationary phase. The chromatographic bed is composed by the gel beads inside the column and the sample is introduced into the injector and carried into the column by the flowing solvent.  As a result of different components adhering to or diffusing through the gel, the sample mixture gets separated.
  Columns used with an FPLC can separate macromolecules based on size (size-exclusion chromatography), charge distribution (ion exchange), hydrophobicity, reverse-phase or biorecognition (as with affinity chromatography). For easy use, a wide range of pre-packed columns for techniques such as ion exchange, gel filtration (size exclusion), hydrophobic interaction, and affinity chromatography are available. FPLC differs from HPLC in that the columns used for FPLC can only be used up to maximum pressure of 3-4 MPa (435-580 psi). Thus, if the pressure of HPLC can be limited, each FPLC column may also be used in an HPLC machine.