text: In addition to the above-mentioned classification based on energy, muon beams are also divided according to the time structure of the particle accelerator, i.e. continuous or pulsed.
  For continuous muon sources no dominating time structure is present. By selecting an appropriate incoming muon rate, muons are implanted into the sample one-by-one. The main advantage is that the time resolution is solely determined by the detector construction and the read-out electronics. There are two main limitations for this type of source, however: (i) unrejected charged particles accidentally hitting the detectors produce non-negligible random background counts; this compromises measurements after a few muon lifetimes, when the random background exceeds the true decay events; and (ii) the requirement to detect muons one at a time sets a maximum event rate. The background problem can be reduced by the use of electrostatic deflectors to ensure that no muons enter the sample before the decay of the previous muon. PSI and TRIUMF host the two continuous muon sources available for µSR experiments.
  At pulsed muon sources protons hitting the production target are bunched into short, intense, and widely separated pulses that provide a similar time structure in the secondary muon beam. An advantage of pulsed muon sources is that the event rate is only limited by detector construction. Furthermore, detectors are active only after the incoming muon pulse, strongly reducing the accidental background counts. The virtual absence of background allows the extension of the time window for measurements up to about ten times the muon mean lifetime. The principal downside is that the width of the muon pulse limits the time resolution. ISIS Neutron and Muon Source and J-PARC are the two pulsed muon sources available for µSR experiments.