text: Usually the time-of-flight tube used in mass spectrometry is praised for simplicity, but for precision measurements of charged low energy particles the electric and the magnetic field in the tube has to be controlled within 10 mV and 1 nT respectively.
  The work function homogeneity of the tube can be controlled by a Kelvin probe. The magnetic field can be measured by a fluxgate compass. High frequencies are passively shielded and damped by radar absorbent material. To generate arbitrary low frequencies field the screen is parted into plates (overlapping and connected by capacitors) with bias voltage on each plate and a bias current on coil behind plate whose flux is closed by an outer core. In this way the tube can be configured to act as a weak achromatic quadrupole lens with an aperture with a grid and a delay line detector in the diffraction plane to do angle resolved measurements. Changing the field the angle of the field of view can be changed and a deflecting bias can be superimposed to scan through all angles.
  When no delay line detector is used focusing the ions onto a detector can be accomplished through the use of two or three einzel lenses placed in the vacuum tube located between the ion source and the detector.
  The sample should be immersed into the tube with holes and apertures for and against stray light to do magnetic experiments and to control the electrons from their start.