text: The apparatus consists of a gas chromatograph equipped with an odour port (ODP), in place of or in addition to conventional detectors, from with human assessors sniff the eluates. The odour port is characterised by its nose-cone design connected to the GC instrument by a transfer line. The odour port is commonly glass or polytetrafluoroethylene. It is generally placed 30–60 cm away from the instrument, extending from the side such that it is not affected by the hot GC oven. The deactivated silica transfer line is generally heated to prevent the condensation of less-volatile compounds. It is flexible so that the assessor can adjust it according to their comfortable sitting position. As traditional warm and dry carrier gases may dehydrate the mucous membrane of the nose, volatiles are delivered via auxiliary gas or humidified carrier gas, with relative humidity (RH) of 50–75%, to ease the dehydration. 
  The olfactometric detector may be coupled with, or connected in parallel to, a flame ionization detector (FID) or mass spectrometer (MS). Moreover, multiple odour ports may be set-up. In these cases, the eluate is generally split evenly between the detectors to allow it to reach the detectors simultaneously.