text: The operating principle of CCC equipment requires a column consisting of a tube coiled around a bobbin. The bobbin is rotated in a double-axis gyratory motion (a cardioid), which causes a variable g-force to act on the column during each rotation. This motion causes the column to see one partitioning step per revolution and components of the sample separate in the column due to their partitioning coefficient between the two immiscible liquid phases. "High-performance" countercurrent chromatography (HPCCC) works in much the same way as HSCCC. A seven-year research and development process produced HPCCC instruments that generated 240 gs, compared to the 80 gs of the HSCCC machines. This increase in g-force and larger bore of the column has enabled a ten-fold increase in throughput, due to improved mobile phase flow rates and a higher stationary phase retention. Countercurrent chromatography is a preparative liquid chromatography technique, however with the advent of the higher-g HPCCC instruments it is now possible to operate instruments with sample loadings as low as a few milligrams, whereas in the past hundreds of milligrams had been necessary. Major application areas for this technique include natural product purification and drug development.