text: Atmospheric pressure (AP) matrix-assisted laser desorption/ionization (MALDI) is an ionization technique (ion source) that in contrast to vacuum MALDI operates at normal atmospheric environment. The main difference between vacuum MALDI and AP-MALDI is the pressure in which the ions are created. In vacuum MALDI, ions are typically produced at 10 mTorr or less while in AP-MALDI ions are formed in atmospheric pressure. In the past, the main disadvantage of the AP-MALDI technique compared to the conventional vacuum MALDI has been its limited sensitivity; however, ions can be transferred into the mass spectrometer with high efficiency and attomole detection limits have been reported. AP-MALDI is used in mass spectrometry (MS) in a variety of applications ranging from proteomics to drug discovery. Popular topics that are addressed by AP-MALDI mass spectrometry include: proteomics; mass analysis of DNA, RNA, PNA, lipids, oligosaccharides, phosphopeptides, bacteria, small molecules and synthetic polymers, similar applications as available also for vacuum MALDI instruments. The AP-MALDI ion source is easily coupled to an ion trap mass spectrometer or any other MS system equipped with electrospray ionization (ESI) or nanoESI source.
  MALDI with ionization at reduced pressure is known to produce mainly singly-charged ions (see "Ionization mechanism" below). In contrast, ionization at atmopsheric pressure can generate highly-charged analytes as was first shown for infrared  and later also for nitrogen lasers. Multiple charging of analytes is of great importance, because it allows to measure high-molecular-weight compounds like proteins in instruments, which provide only smaller m/z detection ranges such as quadrupoles. Besides the pressure, the composition of the matrix is important to achieve this effect.