text: A glow-discharge device (GD) serves as a versatile source, as it can simultaneously introduce and atomize the sample. The glow discharge occurs in a low-pressure argon gas atmosphere between 1 and 10 torr. In this atmosphere lies a pair of electrodes applying a DC voltage of 250 to 1000 V to break down the argon gas into positively charged ions and electrons. These ions, under the influence of the electric field, are accelerated into the cathode surface containing the sample, bombarding the sample and causing neutral sample atom ejection through the process known as sputtering. The atomic vapor produced by this discharge is composed of ions, ground state atoms, and fraction of excited atoms. When the excited atoms relax back into their ground state, a low-intensity glow is emitted, giving the technique its name.
  The requirement for samples of glow discharge atomizers is that they are electrical conductors. Consequently, atomizers are most commonly used in the analysis of metals and other conducting samples. However, with proper modifications, it can be utilized to analyze liquid samples as well as nonconducting materials by mixing them with a conductor (e.g. graphite).