text: Flow rates through an orifice plate can be calculated without specifically calibrating the individual flowmeter so long as the construction and installation of the device complies with the stipulations of the relevant standard or handbook. The calculation takes account of the fluid and fluid conditions, the pipe size, the orifice size and the measured differential pressure; it also takes account of the coefficient of discharge of the orifice plate, which depends upon the orifice type and the positions of the pressure tappings. With local pressure tappings (corner, flange and D+D/2), sharp-edged orifices have coefficients around 0.6 to 0.63, while the coefficients for conical entrance plates are in the range 0.73 to 0.734 and for quarter-circle plates 0.77 to 0.85. The coefficients of sharp-edged orifices vary more with fluids and flow rates than the coefficients of conical-entrance and quarter-circle plates, especially at low flows and high viscosities.
  For compressible flows such as flows of gases or steam, an expansibility factor or expansion factor is also calculated. This factor is primarily a function of the ratio of the measured differential pressure to the fluid pressure and so can vary significantly as the flow rate varies, especially at high differential pressures and low static pressures.
  The equations provided in American and European national and industry standards and the various coefficients used to differ from each other even to the extent of using different combinations of correction factors, but many are now closely aligned and give identical results; in particular, they use the same Reader-Harris/Gallagher (1998) equation for the coefficient of discharge for sharp-edged orifice plates. The equations below largely follow the notation of the international standard ISO 5167 and use SI units.
  Volume flow rate:<br />
  Mass flow rate:<br />