text: Material may be separated by means of air elutriation, which employs an apparatus with a vertical tube through which fluid is passed at a controlled velocity. When the particles are introduced, often through a side tube, the smaller particles are carried over in the fluid stream while the large particles settle against the upward current. If we start with low flow rates small less dense particle attain terminal velocities, and flow with the stream, the particle from the stream is collected in overflow and hence will be separated from the feed. Flow rates can be increased to separate higher size ranges. Further size fractions may be collected if the overflow from the first tube is passed vertically upwards through a second tube of greater cross-section, and any number of such tubes can be arranged in series.
  Advantages: a bulk sample is analyzed using centrifugal classification and the technique is non-destructive. Each cut-point can be recovered for future size-respective chemical analyses. This technique has been used for decades in the air pollution control industry (data used for design of control devices). This technique determines particle size as a function of settling velocity in an air stream (as opposed to water, or some other liquid).
  Disadvantages: a bulk sample (about ten grams) must be obtained. It is a fairly time-consuming analytical technique. The actual test method has been withdrawn by ASME due to obsolescence. Instrument calibration materials are therefore no longer available.