text: The interfacing mechanism is contained inside a common EI source, like that found in any GC-MS system. The liquid phase from a nano HPLC column is admitted from the capillary column port, where the connection tubing and the nebulizer are first introduced and sealed to prevent vacuum loss. The mechanism is based on the formation of an aerosol in high-vacuum conditions, followed by a quick droplet desolvation and final vaporization of the solute prior to the ionization. The completion of the process is quick and complete and reduces chances of thermal decomposition as reported in the Figure, where a scheme of the interface is shown. The core of the interface is represented by the micro-nebulizer. The nebulizer tip protrudes into the ion source so that the spray expansion is completely contained inside the ion volume. The eluate emerges as liquid phase at a flow rate of 300-500 nL/min, and any premature in-tube solvent evaporation is prevented by a convenient thermal insulation of the nebulizer and the connecting tubing from the surrounding source heat. The high temperature of the ion source, between 300 and 400°C, has a double function: to compensate for the latent heat of vaporization during the droplet desolvation, and to convert the solute into the gas phase. If all components of this simple interface are correctly placed and sized, then each substance separated by the nano-column is smoothly converted into the gas phase, the peak profile is nicely reproduced, and high quality mass spectra are generated. Major advantages of this technical solution are the following:  1) It delivers high-quality, fully library matchable mass spectra of most sub-1 kDa molecules amenable by HPLC, 2) It is a chemical ionization free interface (unless operated intentionally) with accurate reproduction of the expected isotope ion abundances, 3) Response is never influenced by matrix components in the sample or in the mobile phase, 4) It can be considered a universal detector for small molecules because response is not related to compound polarity.