text: Libby's first detector was a Geiger counter of his own design. He converted the carbon in his sample to lamp black (soot) and coated the inner surface of a cylinder with it. This cylinder was inserted into the counter in such a way that the counting wire was inside the sample cylinder, in order that there should be no material between the sample and the wire. Any interposing material would have interfered with the detection of radioactivity, since the beta particles emitted by decaying  are so weak that half are stopped by a  thickness of aluminium.
  Libby's method was soon superseded by gas proportional counters, which were less affected by bomb carbon (the additional  created by nuclear weapons testing). These counters record bursts of ionization caused by the beta particles emitted by the decaying  atoms; the bursts are proportional to the energy of the particle, so other sources of ionization, such as background radiation, can be identified and ignored. The counters are surrounded by lead or steel shielding, to eliminate background radiation and to reduce the incidence of cosmic rays. In addition, anticoincidence detectors are used; these record events outside the counter and any event recorded simultaneously both inside and outside the counter is regarded as an extraneous event and ignored.
  The other common technology used for measuring  activity is liquid scintillation counting, which was invented in 1950, but which had to wait until the early 1960s, when efficient methods of benzene synthesis were developed, to become competitive with gas counting; after 1970 liquid counters became the more common technology choice for newly constructed dating laboratories. The counters work by detecting flashes of light caused by the beta particles emitted by  as they interact with a fluorescing agent added to the benzene. Like gas counters, liquid scintillation counters require shielding and anticoincidence counters.
  For both the gas proportional counter and liquid scintillation counter, what is measured is the number of beta particles detected in a given time period. Since the mass of the sample is known, this can be converted to a standard measure of activity in units of either counts per minute per gram of carbon (cpm/g C), or becquerels per kg (Bq/kg C, in SI units). Each measuring device is also used to measure the activity of a blank sample – a sample prepared from carbon old enough to have no activity. This provides a value for the background radiation, which must be subtracted from the measured activity of the sample being dated to get the activity attributable solely to that sample's . In addition, a sample with a standard activity is measured, to provide a baseline for comparison.