text: Acoustic rheometers employ a piezo-electric crystal that can easily launch a successive wave of extensions and contractions into the fluid. This non-contact method applies an oscillating extensional stress. Acoustic rheometers measure the sound speed and attenuation of ultrasound for a set of frequencies in the megahertz range. Sound speed is a measure of system elasticity. It can be converted into fluid compressibility. Attenuation is a measure of viscous properties. It can be converted into viscous longitudinal modulus. In the case of a Newtonian liquid, attenuation yields information on the volume viscosity. This type of rheometer works at much higher frequencies than others. It is suitable for studying effects with much shorter relaxation times than any other rheometer.