text: The liquid is placed within the annulus of one cylinder inside another.  One of the cylinders is rotated at a set speed. This determines the shear rate inside the annulus. The liquid tends to drag the other cylinder round, and the force it exerts on that cylinder (torque) is measured, which can be converted to a shear stress.
  One version of this is the Fann V-G Viscometer, which runs at two speeds, (300 and 600 rpm) and therefore only gives two points on the flow curve.  This is sufficient to define a Bingham plastic model which was once widely used in the oil industry for determining the flow character of drilling fluids. In recent years rheometers that spin at 600, 300, 200, 100, 6 & 3 RPM have become more commonplace. This allows for more complex fluids models such as Herschel–Bulkley to be used. Some models allow the speed to be continuously increased and decreased in a programmed fashion, which allows the measurement of time-dependent properties.