text: An ISFET electrode sensitive to H concentration can be used as a conventional glass electrode to measure the pH of a solution. However, it also requires a reference electrode to operate. If the reference electrode used in contact with the solution is of the AgCl or HgCl classical type, it will suffer the same limitations as conventional pH electrodes (junction potential, KCl leak, and glycerol leak in case of gel electrode). A conventional reference electrode can also be bulky and fragile. A too large volume constrained by a classical reference electrode also precludes the miniaturization of the ISFET electrode, a mandatory feature for some biological or in vivo clinical analyses (disposable mini-catheter pH probe). The breakdown of a conventional reference electrode could also make problem in on-line measurements in the pharmaceutical or food industry if highly valuable products are contaminated by electrode debris or toxic chemical compounds at a late production stage and must be discarded for the sake of safety.
  For this reason, since more than 20 years many research efforts have been dedicated to on-chip embedded tiny reference field effect transistors (REFET). Their functioning principle, or operating mode, can vary, depending on the electrode producers and are often proprietary and protected by patents. Semi-conductor modified surfaces required for REFET are also not always in thermodynamical equilibrium with the test solution and can be sensitive to aggressive or interfering dissolved species or not well characterized aging phenomena. This is not a real problem if the electrode can be frequently re-calibrated at regular time interval and is easily maintained during its service life. However, this may be an issue if the electrode has to remain immersed on-line for prolonged period of time, or is inaccessible for particular constrains related to the nature of the measurements itself (geochemical measurements under elevated water pressure in harsh environments or under anoxic or reducing conditions easily disturbed by atmospheric oxygen ingress or pressure changes).
  A crucial factor for ISFET electrodes, as for conventional glass electrodes, remains thus the reference electrode. When troubleshooting electrode malfunctions, often, most of the problems have to be searched for from the side of the reference electrode.