text: These depend upon analysis of the "halo" of diffracted light produced when a laser beam passes through a dispersion of particles in air or in a liquid. The angle of diffraction increases as particle size decreases, so that this method is particularly good for measuring sizes between 0.1 and 3,000 μm. Advances in sophisticated data processing and automation have allowed this to become the dominant method used in industrial PSD determination. This technique is relatively fast and can be performed on very small samples. A particular advantage is that the technique can generate a continuous measurement for analyzing process streams.
  Laser diffraction measures particle size distributions by measuring the angular variation in intensity of light scattered as a laser beam passes through a dispersed particulate sample. Large particles scatter light at small angles relative to the laser beam and small particles scatter light at large angles. The angular scattering intensity data is then analyzed to calculate the size of the particles responsible for creating the scattering pattern, using the Mie theory or Fraunhofer approximation of light scattering. The particle size is reported as a volume equivalent sphere diameter.