text: Also known as heated persulfate, the method utilizes the same free radical formation as UV persulfate oxidation except uses heat to magnify the oxidizing power of persulfate. Chemical oxidation of carbon with a strong oxidizer, such as persulfate, is highly efficient, and unlike UV, is not susceptible to lower recoveries caused by turbidity in samples. The analysis of system blanks, necessary in all chemical procedures, is especially necessary with heated persulfate TOC methods because the method is so sensitive that reagents cannot be prepared with carbon contents low enough to not be detected. Persulfate methods are used in the analysis of wastewater, drinking water, and pharmaceutical waters. When used in conjunction with sensitive NDIR detectors heated persulfate TOC instruments readily measure TOC at single digit parts per billion (ppb) up to hundreds of parts per million (ppm) depending on sample volumes.