text: Primary, calibration, and reference materials are only available in small quantities and purchase is often limited to once every few years. Depending on the specific isotope systems and instrumentation, a shortage of available reference materials can be problematic for daily instrument calibrations or for researchers attempting to measure isotope ratios in a large number of natural samples. Rather than using primary materials or reference materials, a laboratory measuring stable isotope ratios will typically purchase a small quantity of the relevant reference materials and measure the isotope ratio of an in-house material against the reference, making that material into a working standard specific to that analytical facility. Once this lab-specific working standard has been calibrated to the international scale the standard is used to measure the isotopic composition of unknown samples. After measurement of both sample and working standard against a third material (commonly called the working gas or the transfer gas) the recorded isotopic distributions are mathematically corrected back to the international scale. It is thus critical to measure the isotopic composition of the working standard with high precision and accuracy (as well as possible given the precision of the instrument and the accuracy of the purchased reference material) because the working standard forms the ultimate basis for accuracy of most mass spectrometric observations. Unlike reference materials, working standards are typically not calibrated across multiple analytical facilities and the accepted δ value measured in a given laboratory could reflect bias specific to a single instrument. However, within a single analytical facility this bias can be removed during data reduction. Because each laboratory defines unique working standards the primary, calibration, and reference materials are long-lived while still ensuring that the isotopic composition of unknown samples can be compared across laboratories.