text: The parallel configuration or long optical path-length arrangement only provides information about the spectral changes that occur in the solution adjacent to the working electrode surface, improving the sensitivity to soluble compounds because the length of the optical pathway can be as longer as the length of the electrode.
  The light beam travels parallel to the working electrode surface, sampling the first micrometers of the solution adjacent to the working electrode surface, and collecting the information on the spectrometer.
  <br />
  Usually, aligning light beams has been a difficult task. However, simple alternatives have been developed to perform measurements in parallel configuration. There are several advantages in this configuration respect to the normal one: better sensitivity, lower detection limits; optically transparent electrodes are not required; and the spectral changes are related only to the diffusion layer.