text: Measuring isotopic ratios by mass spectrometry includes multiple steps in which samples can undergo cross-contamination, including during sample preparation, leakage of gas through instrument valves, the generic category of phenomena called memory effects, and the introduction of blanks (foreign analyte measured as part of the sample). As a result of these instrument-specific effects the range in measured δ values can be lower than the true range in the original samples. To correct for such scale compression researchers calculate a "stretching factor" by measuring two isotopic reference materials (Coplen, 1988). For the hydrogen system the two reference materials are commonly VSMOW2 and SLAP2, where δH = 0 and δH = -427.5 vs. VSMOW. If the measured difference between the two references is less than 427.5‰, all measured  H/H ratios are multiplied by the stretching factor required to bring the difference between the two reference materials in line with expectations. After this scaling, a factor is added to all measured isotopic ratios so that the reference materials attain their defined isotopic values. The carbon system also uses two anchoring reference materials (Coplen et al., 2006a; 2006b).