text: In conventional DLTS the capacitance transients are investigated by using a lock-in amplifier or double box-car averaging technique when the sample temperature is slowly varied (usually in a range from liquid nitrogen temperature to room temperature 300 K or above). The equipment reference frequency is the voltage pulse repetition rate. In the conventional DLTS method this frequency multiplied by some constant (depending on the hardware used) is called the "rate window". During the temperature scan, peaks appear when the emission rate of carriers from some defect equals the rate window. By setting up different rate windows in subsequent DLTS spectra measurements one obtains different temperatures at which some particular peak appears. Having a set of the emission rate and corresponding temperature pairs one can make an Arrhenius plot, which allows for the deduction of defect activation energy for the thermal emission process. Usually this energy (sometimes called the defect energy level) together with the plot intercept value are defect parameters used for its identification or analysis. On samples with low free carrier density conductance transients have also been used for a DLTS analysis.
  In addition to the conventional temperature scan DLTS, in which the temperature is swept while pulsing the device at a constant frequency, the temperature can be kept constant and sweep the pulsing frequency. This technique is called the frequency scan DLTS. In theory the frequency and temperature scan DLTS should yield same results. Frequency scan DLTS is specifically useful when an aggressive change in temperature might damage the device. An example when frequency scan is shown to be useful is for studying modern MOS devices with thin and sensitive gate oxides.
  DLTS has been used to study quantum dots and perovskite solar cells.