text: Chemical oxidation analysers inject the sample into a chamber with phosphoric acid followed by persulfate. The analysis is separated into two steps. One removes inorganic carbon by acidification and purging. After removal of inorganic carbon persulfate is added and the sample is either heated or bombarded with UV light from a mercury vapor lamp. Free radicals form persulfate and react with any carbon available to form carbon dioxide. The carbon from both determination (steps) is either run through membranes which measure the conductivity changes that result from the presence of varying amounts of carbon dioxide, or purged into and detected by a sensitive NDIR detector. Same as the combustion analyser, the total carbon formed minus the inorganic carbon gives a good estimate of the total organic carbon in the sample.
  This method is often used in online applications because of its low maintenance requirements.