text: The indexing results are used to generate a map of the crystallographic orientation at each point on the surface being studied. Thus, scanning the electron beam in a prescribed fashion (typically in a square or hexagonal grid, correcting for the image foreshortening due to the sample tilt) results in many rich microstructural maps. These maps can spatially describe the crystal orientation of the material being interrogated and can be used to examine microtexture and sample morphology. Some maps describe grain orientation, boundary, and diffraction pattern (image) quality. Various statistical tools can measure the average misorientation, grain size, and crystallographic texture. From this dataset, numerous maps, charts and plots can be generated. The orientation data can be visualised using a variety of techniques, including colour-coding, contour lines, and pole figures.
  Microscope misalignment, image shift, scan distortion that increases with decreasing magnification, roughness and contamination of the specimen surface, boundary indexing failure and detector quality can lead to uncertainties in determining the crystal orientation. The EBSD signal-to-noise ratio depends on the material and decreases at excessive acquisition speed and beam current, thereby affecting the angular resolution of the measurement.