text: The cross-sectional area of the flow is calculated from a depth measurement and the average velocity of the flow is measured directly (Doppler and propeller methods are common). Velocity times the cross-sectional area yields a flow rate which can be integrated into volumetric flow. There are two types of area velocity flowmeter: (1) wetted; and (2) non-contact. Wetted area velocity sensors have to be typically mounted on the bottom of a channel or river and use Doppler to measure the velocity of the entrained particles. With depth and a programmed cross-section this can then provide discharge flow measurement. Non-contact devices that use laser or radar are mounted above the channel and measure the velocity from above and then use ultrasound to measure the depth of the water from above. Radar devices can only measure surface velocities, whereas laser-based devices can measure velocities sub-surface.