text: Thermal mass flowmeters generally use combinations of heated elements and temperature sensors to measure the difference between static and flowing heat transfer to a fluid and infer its flow with a knowledge of the fluid's specific heat and density. The fluid temperature is also measured and compensated for. If the density and specific heat characteristics of the fluid are constant, the meter can provide a direct mass flow readout, and does not need any additional pressure temperature compensation over their specified range.
  Technological progress has allowed the manufacture of thermal mass flowmeters on a microscopic scale as MEMS sensors; these flow devices can be used to measure flow rates in the range of nanoliters or microliters per minute.
  Thermal mass flowmeter (also called thermal dispersion or thermal displacement flowmeter) technology is used for compressed air, nitrogen, helium, argon, oxygen, and natural gas. In fact, most gases can be measured as long as they are fairly clean and non-corrosive. For more aggressive gases, the meter may be made out of special alloys (e.g. Hastelloy), and pre-drying the gas also helps to minimize corrosion.
  Today, thermal mass flowmeters are used to measure the flow of gases in a growing range of applications, such as chemical reactions or thermal transfer applications that are difficult for other flowmetering technologies. Some other typical applications of flow sensors can be found in the medical field like, for example, CPAP devices, anesthesia equipment or respiratory devices. This is because thermal mass flowmeters monitor variations in one or more of the thermal characteristics (temperature, thermal conductivity, and/or specific heat) of gaseous media to define the mass flow rate.