text: In the first experiments with laser heating, temperature came from a calibration of laser power made with known melting points of various materials. When using the pulsed ruby laser this was unreliable due to the short pulse. YAG lasers quickly become the standard, heating for relatively long duration, and allowing observation of the sample throughout the heating process. It was with the first use of YAG lasers that Bassett used an optical pyrometer to measure temperatures in the range of 1000 °C to 1600 °C. The first temperature measurements had a standard deviation of 30 °C from the brightness temperature, but due to the small sample size was estimated to be 50 °C with the possibility that the true temperature of the sample being was 200 °C higher than that of the brightness measurement. Spectrometry of the incandescent light became the next method of temperature measurement used in Bassett's group. The energy of the emitted radiation could be compared to known black-body radiation spectra to derive a temperature. Calibration of these systems is done with published melting points or melting points as measured by resistive heating.