text: In gravimetric method, the weight change of the adsorbent sample in the gravity field due to adsorption from the gas phase is recorded. Various types of sensitive microbalance have been developed for this purpose. A continuous-flow gravimetric technique coupled with wavelet rectification allows for higher precision, especially in the near-critical region.
  Major advantages of gravimetric method include sensitivity, accuracy, and the possibility of checking the state of activation of an adsorbent sample. However, consideration must be given to buoyancy correction in gravimetric measurement. A counterpart is used for this purpose. The solid sample is placed in a sample holder on one arm of the microbalance while the counterpart is loaded on the other arm. Care must be taken to keep the volume of the sample and the counterpart as close as possible to reduce the buoyancy effect. The system is vacuumed and the balance is zeroed before starting experiments. Buoyancy is measured by introducing helium and pressurizing up to the highest pressure of the experiment. It is assumed that helium does not adsorb and any weight change (ΔW) is due to buoyancy. Knowing the density of helium (), one can determine the difference in volume (ΔV) between the sample and the counterpart:
  The measured weight can be corrected for the buoyancy effect at a specified temperature and pressure:
   is the weight reading before correction.