text: Combinations of chromatographic methods can be used to purify a target molecule. The purpose of purifying proteins with FPLC is to deliver quantities of the target at sufficient purity in a biologically active state to suit its further use. The quality of the end product varies depending the type and amount of starting material, efficiency of separation, and selectivity of the purification resin. The ultimate goal of a given purification protocol is to deliver the required yield and purity of the target molecule in the quickest, cheapest, and safest way for acceptable results. The range of purity required can be from that required for basic analysis (SDS-PAGE or ELISA, for example), with only bulk impurities removed, to pure enough for structural analysis (NMR or X-ray crystallography), approaching >99% target molecule. Purity required can also mean pure enough that the biological activity of the target is retained. These demands can be used to determine the amount of starting material required to reach the experimental goal. If the starting material is limited and full optimization of purification protocol cannot be performed, then a safe standard protocol that requires a minimum adjustment and optimization steps are expected. This may not be optimal with respect to experimental time, yield, and economy but it will achieve the experimental goal. On the other hand, if the starting material is enough to develop more complete protocol, the amount of work to reach the separation goal depends on the available sample information and target molecule properties. Limits to development of purification protocols many times depends on the source of the substance to be purified, whether from natural sources (harvested tissues or organisms, for example), recombinant sources (such as using prokaryotic or eukaryotic vectors in their respective expression systems), or totally synthetic sources.
  No chromatographic techniques provide 100% yield of active material and overall yields depend on the number of steps in the purification protocol. By optimizing each step for the intended purpose and arranging them that minimizes inter step treatments, the number of steps will be minimized.
  A typical multistep purification protocol starts with a preliminary capture step which often utilizes ion exchange chromatography (IEC). The media (stationary phase) resin consists of beads, which range in size from being large (good for fast flow rates and little to no sample clarification at the expense of resolution) to small (for best possible resolution with all other factors being equal). Short and wide column geometries are amenable to high flow rates also at the expense of resolution, typically because of lateral diffusion of sample on the column. For techniques such as size exclusion chromatography to be useful, very long, thin columns and minimal sample volumes (maximum 5% of column volume) are required. Hydrophobic interaction chromatography (HIC) can also be used for first and/ or intermediate steps. Selectivity in HIC is independent of running pH and descending salt gradients are used. For HIC, conditioning involves adding ammonium sulfate to the sample to match the buffer A concentration. If HIC is used before IEC, the ionic strength would have to be lowered to match that of buffer A for IEC step by dilution, dialysis or buffer exchange by gel filtration. This is why IEC is usually performed prior to HIC as the high salt elution conditions for IEC are ideal for binding to HIC resins in the next purification step. Polishing is used to achieve the final level of purification required and is commonly performed on a gel filtration column. An extra intermediate purification step can be added or optimization of the different steps is performed for improving purity. This extra step usually involves another round of IEC under completely different conditions.
  Although this is an example of a common purification protocol for proteins, the buffer conditions, flow rates, and resins used to achieve final goals can be chosen to cover a broad range of target proteins. This flexibility is imperative for a functional purification system as all proteins behave differently and often deviate from predictions.