text: A microwave spectrometer can be most simply constructed using a source of microwave radiation, an absorption cell into which sample gas can be introduced and a detector such as a superheterodyne receiver. A spectrum can be obtained by sweeping the frequency of the source while detecting the intensity of transmitted radiation. A simple section of waveguide can serve as an absorption cell. An important variation of the technique in which an alternating current is applied across electrodes within the absorption cell results in a modulation of the frequencies of rotational transitions. This is referred to as Stark modulation and allows the use of phase-sensitive detection methods offering improved sensitivity. Absorption spectroscopy allows the study of samples that are thermodynamically stable at room temperature.  The first study of the microwave spectrum of a molecule () was performed by Cleeton & Williams in 1934. Subsequent experiments exploited powerful sources of microwaves such as the klystron, many of which were developed for radar during the Second World War. The number of experiments in microwave spectroscopy surged immediately after the war. By 1948, Walter Gordy was able to prepare a review of the results contained in approximately 100 research papers. Commercial versions  of microwave absorption spectrometer were developed by Hewlett-Packard in the 1970s and were once widely used for fundamental research. Most research laboratories now exploit either Balle-Flygare or chirped-pulse Fourier transform microwave (FTMW) spectrometers.