text: With the advent of digitally based displays, some modern spectrum analyzers use analog-to-digital converters to sample spectrum amplitude after the VBW filter. Since displays have a discrete number of points, the frequency span measured is also digitised. Detectors are used in an attempt to adequately map the correct signal power to the appropriate frequency point on the display. There are in general three types of detectors: sample, peak, and average
  *Sample detection – sample detection simply uses the midpoint of a given interval as the display point value. While this method does represent random noise well, it does not always capture all sinusoidal signals.
  *Peak detection – peak detection uses the maximum measured point within a given interval as the display point value. This insures that the maximum sinusoid is measured within the interval; however, smaller sinusoids within the interval may not be measured. Also, peak detection does not give a good representation of random noise.
  *Average detection – average detection uses all of the data points within the interval to consider the display point value. This is done by power (rms) averaging, voltage averaging, or log-power averaging.