text: The TAS Measuring Station installed at the GSI had a tape transport system that allowed the collection of the ions coming out of the separator (they were implanted in the tape), and the transportation of those ions from the collection position to the center of the TAS for the measurement (by means of the movement of the tape). The TAS at this facility was made of a cylindrical NaI crystal of Φ = h = 35.6 cm, with a concentric cylindrical hole in the direction of the symmetry axis. This hole was filled by a plug detector (4.7x15.0 cm) with a holder that allowed the placement of ancillary detectors and two rollers for a tape.