Discussion Chain ID: 19151
================================================================================

User asked: How is relative humidity determined from a wet and dry bulb readings?

The question [What is the second thermometer in the image from the Esperanza Antarctic temperature record?](https://earthscience.stackexchange.com/q/19145/6031) shows what might be a wet/dry bulb setup. I've given one to a friend to measure humidity for fun (it's me who thinks it's fun). I'd learned about them in Earth Science class a zillion years ago, so when I happened to see one recently I grabbed it. I understand the very basics of how it can indicate humidity; the drier the air, the faster water will evaporate into it and so the colder the wet bulb will be with respect to the dry bulb. To get a numerical value for humidity with this set up you use the coarse table on the front (there's a finer table on the back of the box) and look up the dry bulb temperature and the dry minus wet difference to find an approximate humidity. For example If I look up 19 and 2 °C for those respectively (roughly what's shown in the image) I get about 81% relative humidity. But suppose I hooked up a camera to a Raspberry Pi and imaged the two thermometers, processed the image and determined the two temperatures and wanted to calculate the relative humidity without interpolating the table. **Question:** How is relative humidity determined from a wet and dry bulb readings? What are the steps and where's a good source for the equations and parameters involved? _Click for full size_ [](https://i.sstatic.net/Yu941.jpg)

Discussion:
Comment: different but related: [How is the specific heat of a wet bulb determined?](https://earthscience.stackexchange.com/q/15349/6031) (which is currently unanswered) and also [How would I use data to find specific humidity and mixing ratio?](https://earthscience.stackexchange.com/q/13511/6031)

Answer 1 (7 votes):
_Edit: 1 May 2021_ The following procedure uses the less accurate method from page 455 onward from the scanned sections of the book pictured below, from the original answer. The procedure is a multistage process ideally suited to either a spreadsheet or programming code. The equations use SI units. _**1\. Calculate the saturated vapor pressure at the dry bulb temperature**_ This is done using the equation: $P''_{ws} \ = \ 0.61078\ e\ ^{17.27T_{dry}/(273.3+T_{dry})} \ \ kPa$ Where $T_{dry}$ is the dry bulb temperature. _**2\. Calculate the saturated vapor pressure at the wet bulb temperature**_ This is done using the equation: $P'_{ws} \ = \ 0.61078\ e\ ^{17.27T_{wet}/(273.3+T_{wet})} \ \ kPa$ Where $T_{wet}$ is the wet bulb temperature. _**3\. Calculate the moisture content of saturated air at the wet bulb temperature**_ Use the equation: $r_o \ = \ \frac{18.016 \ (1.0048)P'_{ws}} {28.9664 \ (P_{atm}-1.0048)P'_{ws}} \ \ kg/kg$ Where $P_{atm}$ is atmospheric pressure _**4\. Calculate the enthalpy of water vapor pressure at the wet bulb temperature**_ $H'_{wo} \ = \ -0.00000662 \ t_{wet}^3-0.000194 \ t_{wet}^2+1.8375 \ t_{wet}+2500.83 \ \ kJ/kg$ _**5\. Calculate the enthalpy of liquid water at the wet bulb temperature**_ $H'_{wl} \ = \ 0.0000063 \ t_{wet}^3-0.000727 \ t_{wet}^2+4.2058 \ t_{wet}+0.03 \ \ kJ/kg$ _**6\. Calculate the enthalpy of water vapor pressure at the dry bulb temperature**_ $H'_{wi} \ = \ -0.00000662 \ t_{dry}^3-0.000194 \ t_{dry}^2+1.8375 \ t_{dry}+2500.83 \ \ kJ/kg$ _**7\. Calculate the moisture content of saturated air at the wet bulb temperature**_ $r \ = \ \frac{r_o*(H'_{wo}-H'_{wl})-(1.005 \ t_{dry}-1.005 \ t_{wet})}{(H'_{wi}-H'_{wl})} \ \ kg/kg$ _**8\. Calculate the vapor pressure**_ $P_w \ = \ \frac{r \ P_{atm}} {1.0048 \ (18.016/28.9664+r)} \ \ kPa$ _**9\. Calculate the relative humidity**_ $\phi \ = \ P_w/P''_{ws}$ * * * _**Previous Answer - scans from relevant pages from reference book**_ I'm going to go old school with this one. The following pictures are the relevant pages from: _Environmental Engineering in South African Mines_ , The Mine Ventilation Society of South Africa, 1989, pp 451-455. They give the psychometric equations that are required to calculate relative humidity from wet & dry bulb temperatures. [](https://i.sstatic.net/DfrY7.jpg) [](https://i.sstatic.net/sKMCm.jpg) [](https://i.sstatic.net/5Zupk.jpg) [](https://i.sstatic.net/g4fNs.jpg) [](https://i.sstatic.net/ZbbLp.jpg) For simplicity, I use the less accurate equations on p455, section 9.2. As a slight diversion, the other way to measure [wet & dry bulb temperatures](http://science.waltermack.com/VirtualFieldTrip/whirling.html) is to use a [whirling hygrometer](https://www.aweimagazine.com/article/use-of-whirling-hygrometers-1137/). The wet bulb temperature is always read first because it will start to warm as soon as the hygrometer stops whirling. [](https://i.sstatic.net/oTa44.gif)

Discussion on this answer:
Comment: While screenshots of text are usually discouraged, this is the best possible answer for me at the moment, and I appreciate the speedy response! I'll take a look today and if nobody else does I'll add a summary answer based on what I've implemented, so that those who are visually impaired can have an answer as well. Thanks!


================================================================================

