Discussion Chain ID: 169318
================================================================================

User asked: What is the standard industrial method for measuring caffeine content in food and drinks?

There are a lot of questions here looking for simple methods to measure caffeine content (or extract it), including one of mine here and on the [_Coffee_ site](https://coffee.stackexchange.com/tour): * [Home science optical absorption test for approximate caffeine quantity in coffee?](https://chemistry.stackexchange.com/q/70793/16035) * [Are there any ways that coffee consumers can measure the caffeine content of what they are drinking by themselves?](https://coffee.stackexchange.com/q/6073/9096) But here I'd like to ask something different: **Question:** Is there a standard industrial method for measuring caffeine content in food and drinks? For example, if there were government requirements to post the amount of caffeine in a product there may be some standard way that government advised that it be done (i.e., "this is the way we're going to do it to verify what you report."). While I don't know of any such requirements, that doesn't mean there aren't any.

Answer 1 (14 votes):
The answer is fortunately rather simple. Use a C18 reversed phase column and a UV detector in an HPLC system. One would make a calibration curve by injecting various concentrations into the column. Now, if an unknown sample of a drink is injected, and one sees a peak at the same retention time in the sample chromatogram, you know it is caffeine. By interpolating the area, we know its concentration. When an HPLC method is developed, it is ensured that no other component elutes at the same time. This is achieved by using gradients of mobile phase composition. If there is a _legal_ issue and one has to provide a 100% foolproof test, then one has to use a mass spectrometer along with a UV detector to confirm the identity of the peak at various locations, i.e., one ensures that the peak is pure. See the examples in Agilent's note "Determination of Caffeine in Coffee Products According to DIN 20481 by Edgar Naegele from where this figure is taken: [](https://i.sstatic.net/cDF27.png)

Discussion on this answer:
Comment: I could see series caffeine aficionados laying out $50k for an HPLC. How else do you tune in the espresso machine for that perfect cup?
Comment: [Determination of Caffeine in Coffee Products According to DIN 20481](https://www.agilent.com/cs/library/applications/5991-2851EN.pdf) (also [here](https://lcms.labrulez.com/paper/11709)) compare figures 4 & 5\. *Nice!*
Comment: "DIN"="Deutsches Institut für Normung" - the German ISO member. DIN norms are pretty much the requested "standard industrial method".
Comment: @JonCuster or maybe a group of them can start a local neighborhood HPLC co-op?
Comment: @uhoh, Mini portable and complete HPLCs of the size of a brief case are available now. I see no problem in running caffeine stds and samples.
Comment: @AChem that's great! I'm afraid that the compactness in volume won't translate to compactness in cost - buying one may require a brief case full of Starbucks gift cards or equivalent. So the cost sharing motivation for the co-op idea can be translated to a traveling "coffee inspector" who makes their rounds on a weekly basis on a motorcycle using an Uber Eats or Food Panda on-demand model. *I like it! I'm gonna be rich!*
Comment: @uhoh We used to keep a 5 pound bottle of caffeine citrate next to the always on lab break room coffee pot. The sign on it, due to me, was “Two scoops per pot, to combat the Santa menace.” So you could just start with decaf and add caf, to desired level of wiredness. (We took the big bottle of caffeine citrate away after a few weeks, because someone might have not understood and actually added the two scoops!)
Comment: @EdV Not a chemist can you explain lol
Comment: @AzorAhai-him- My comment had a typo: “Santa” should be “Sanka”, a brand of decaf coffee. In a large research group, coffee “powers” progress and decaf would be pretty much useless if you want to facilitate productivity. Our group, run by a famous chemist (not me!), typically had around 25 grad students, around 8 post-docs/sabbatical leave folks, and we had undergrads doing short research projects, though we never made a list of them because they came and went semester by semester. So the caffeine citrate was just kind of a joke, with a little point: get stuff done!
Comment: @EdV Why can't you add the two scoops?
Comment: @AzorAhai-him- Two scoops would be *way* too much and could have very bad medical consequences!
Comment: @EdV Ah, lol. .
Comment: for non-chemists like myself this [video](https://www.youtube.com/watch?v=IFJOEJ98jVs) and [lab writeup](https://analyticalchem.community.uaf.edu/wp-content/uploads/sites/313/2013/04/Lab-7-Caffeine-Extraction-by-HPLC-Lab-text.pdf) titled "Lab 7: Caffeine Quantification by HPLC" are quite helpful.


================================================================================

