Discussion Chain ID: 178090
================================================================================

User asked: UV-Vis Absorption Spectra climbing at lower wavelengths

I have run a sample of Pyrene in Ethanol and got a UV Vis absorbance spectra that starts climbing as it reaches the lower wavelengths. I am using quartz cuvettes and was wondering what could be the cause of this rise at the shorter wavelengths. [](https://i.sstatic.net/JuZF6.png) Here is the literature spectra. [](https://i.sstatic.net/59SmZ.png) Here is my measured spectra.

Discussion:
Comment: Please add more instrumental details. Is it a single beam instrument or a double beam instrument. What was your blank?
Comment: the solvent and quartz of your cell (depending on type of quartz) will start absorbing at shorter wavelengths. You can minimise this to some extent using a matched blank cell.
Comment: @AChem it is a double beam instrument, however on this test run I did not use a blank
Comment: You should not use air as a blank.

Answer 1 (1 votes):
If you had given the reference for the literature spectra, I would be able to say for certain, but it's possible that they didn't use a quartz cuvette. According to [this website](https://qvarz.com/quartz-cuvettes-for-uv-vis-ir-spectroscopy/), quartz starts absorbing below ~200 nm. Looks like it depends on the type of quartz, and brand of cuvette, you've used (from [this link](https://airekacells.com/blog/cuvette-guide)): [](https://i.sstatic.net/568UH.jpg)

Discussion on this answer:
Comment: At wavelengths shorter than $200$ nm, everything absorbs the light, even glass, water or air. Absorption spectrum at these wavelengths must be determined in a vacuum.
Comment: @Maurice it didn’t occur to me how high that energy was (I’m used to working in wavenumbers)… but yeah, that makes sense


================================================================================

