Discussion Chain ID: 119039
================================================================================

User asked: Raman Spectrum of Water

This is the Raman spectrum of water. There is $\pu{1635 cm-1}$ Raman peak corresponding to $\ce{HOH}$ and $\pu{3410 cm-1}$ Raman peak corresponding to $\ce{OH}$. But why is there general nonzero background in intensity? What caused it? This is taken using 532nm wavelength raman system: [](https://i.sstatic.net/6KCkc.jpg) This is taken using 785nm wavelength raman system up to 2750 cm-1 only (versus 4000 cm-1 of the first spectrum): [](https://i.sstatic.net/QpiVR.jpg) In the above picture. Why is the background dimishing to zero towards 2750 cm-1. My guess is that as wavelength gets higher, water absorbs more. But can it absorb the noise too?

Discussion:
Comment: All detectors have noise. All spectrometers aren’t perfect. There is always background in every measurement.
Comment: You mean for very expensive advanced raman spectrometers, the noise would be less and less? You mean the above background is simply noise and not any characteristics of water such as fluorescence?
Comment: You need to keep the spectrometer and the detector separate in your mind. It is a quite broad question to determine all the possible noise sources in the whole system. But, consider a high resolution spectrometer- you only pass a very narrow frequency range to the detector. So, not many photons. But the dark counts are whatever they are, so your signal to noise is low.
Comment: The above photo was sample shown to me by manufacturer taken using 532nm laser and visible wavelength detector. Are you saying if 785nm is used, the noise would even be more??
Comment: See (added) 2nd picture using 785nm spectrum of water. Why is the background dimishing to zero towards 2750 cm-1. My guess is that as wavelength gets higher, water absorbs more. But can it absorb the noise too? What do you think cause it to go down to zero?
Comment: You just put side by side pictures taken with different source and detector. Ergo the absolute values at y do not tell info. Fluorescence is present in Raman spectra and is difficult to remove as for it requires the sample to be in, it has no physical blank to run a background.
Comment: porphyrin, water becomes more absorbent as wavelength is increased. So water can absorb IR more than visible. So in the 2nd picture.. As wavelength increased, it's absorbing the fluorescence? why do you think this theory is wrong? The 2nd device used the First Defender RMX https://www.thermofisher.com/order/catalog/product/FIRSTDEFENDERRMX the seller gave me sample picture of water spectra taken direct 3mm from water without passing through any container.
Comment: 535nm principally scatters a factor 4.7 more than 785, not only Raman scattering but also elastic Raleigh scattering, which is one of your major noise sources. Also the second raman with the NIR source is perhaps a modern FT Raman. And the NIR source makes less flourescence, which is the worst noise source of all in most cases.
Comment: @porphyrin: 3000 1/cm from 785 nm is about **1** micron (1026.8 nm).
Comment: @Samzun That 2nd device is a handheld detector which is not designed to output high-quality spectra. It isn't a general-purpose spectrometer. It's used to detect specific compounds, probably by pattern matching the data to a reference library. So it just has to be good enough for that purpose. You have to make a lot of compromises to shrink a spectrometer to that size, and the loss of intensity towards 3000 cm-1 may well be one of those compromises.
Comment: It's been a while since I have run any Raman spectra but the constant noise level is bothersome. I suspect less than optimal instrument conditions and too much stray light in the spectrum.

Answer 1 (3 votes):
This answer is based on the educated guess that the green excitation Raman spectrum was acquired with a **dispersive** Raman instrument*. > But why is there general nonzero background in intensity? What caused it? **For any real-life instrumentation, you should never assume to measure 0 signal for 0 analyte concentration** (see also [Does a calibration curve of absorbance vs concentration need to pass exactly through the origin?](https://chemistry.stackexchange.com/q/110337/160)) **nor expect the baseline to be at zero unless corrected for** , even if there really were no Raman scattering outside those two bands, the instrument and/or sample preparation procedures (electronics, optics, masking, cross-sensitivity, ...) will often lead to some non-zero signal. The CCD in the spectrograph has a so-called dark current. So the raw signal you observe is dark current plus photon signal. Dark current is influenced by settings of the CCD (you trade off sensitivity and dark current) and by temperature. The constant part of the dark current shouldn't trouble you too much: you can measure and then subtract it (which is in fact what spectrometer software does where you see the baseline at zero). However, as subtracting spectra sums their noise** (error propagation). As Raman measurements typcially suffer from low signal-to-noise-ratio, you may want to avoid this, so Raman spectrometer software usually either outputs raw spectra or offers you the choice whether you want to subtract the dark spectra. For SNR of Raman spectra, you'll have to take into account shot noise (Photon noise): the noise is directly proportional to the signal intensity, and that includes optical background such as fluorescence or elastically scattered or reflected light as well as your Raman signal. (And then come the other sources of noise such as flicker noise and the various sources of noise related to your application). For a thorough discussion of the noise properties of Raman spectrometers, there's a whole chaper of [McCreery: Raman Spectroscopy for Chemical Analysis, ISBN: 978-0-471-23187-5](https://www.wiley.com/en-us/Raman+Spectroscopy+for+Chemical+Analysis-p-9780471231875) on the topic (for FT instruments, see [Zhao & McCreery: Multichannel FT-Raman Spectroscopy: Noise Analysis and Performance Assessment, ApplSpec, 1687 - 1697, 1997.](https://doi.org/10.1366/0003702971939361)) * * * **About the 785 measurement in comparison:** The green excitation y-axis looks like raw counts (the data itself would reveal more: raw counts are usually integer numbers). In contrast the red spectrum's y-axis goes from 0 to slightly above 1.0. Which means that _some_ transformations have been done (which is impossible to say from the data alone - you'll have to talk to the manufacturer what they do). > water absorption Absorbance of in the range below 1100 nm is < 0.2/cm. So yes, you loose about 1/3 of the intensity through 5 mm water and back - but you can avoid that by focusing not as deep into the water. What hurts your signal more is that the quantum efficiency down there is pretty awful for many CCDs ([around 10 %](https://andor.oxinst.com/learning/view/article/low-dark-current-deep-depletion-\(ldc-dd\)-technology), there's a reason why their spectral range ends in the mid of the -CH stretching vibrations). * * * * A rather safe guess, as FT instruments for such short wavelength AFAIK exist at the moment only as "home-built" instruments developed in some research facilities - whoever uses one of those is going to be fully aware that they are using something special. ** more precisely: the variances of independent random variables are additive

Discussion on this answer:
Comment: Do you think I should get this 785nm model https://www.shopstellarnet.com/785nm-preconfigured-raman-spectrometer-system/ or this 532nm model http://enspectr.com/portfolio/enspectr-r532/ ? I just want to buy one for learning molecular structure.. which would you buy if you were just a beginner?
Comment: btw.. when you said "absorbance of in the range below 1100 nm is < 0.2/cm". Did you mean by below is like 900nm or below is like 1200nm? I was asking water absorbance between 800nm to 1050nm. how many percentage is the absorbance through 5mm of water using 785nm laser? I want to study water hence can't get 532nm off my mind. Thanks so much.
Comment: cbeleites, the Stellar model has 499mW laser but at 785nm. The Enspecr has 30mW laser only at 532nm. I know 532nm has 5 times intensity than 785nm. But it's like equal to 30mWx5= 150mW laser only. What substance that you need the full 499mW on the 785nm.. which still has 3 times the laser power compared to 150mW (30mW x 5). I guess this will be my deciding factor which of them to get. Also need to know absorbance of water between 785 to 1015nm. Is it like 15% only or 1/3? Thanks!
Comment: 900 nm is below 1100 nm (so all your Raman range).
Comment: You cannot conclude too much from the 500 mW vs. 15 mW: λ⁴ dependency is one thing, quantum efficiency and the losses in the optical path are another thing. You'll have to measure to find out how well the instruments do.
Comment: But for visible light 532nm laser raman spectrometer, there is no losses (or almost insignificant absorbance) via the water since it's not IR, right? I want to study effect of 1% solution in water which the 532nm system claims to do.
Comment: Samzun: From what you have written so far, the question which Raman spectrometer is best for you cannot be answered. I also have the impression that you need a better understanding of the instrumentation basics and tradeoffs here. I feel that thisis leaving the region of what is generally useful to the public. As I do advise (and teach) professionally on these topics, you can contact me (see profile) if you are interested in getting professional help with the decisions for purchasing an instrument. Otherwise, reading & understanding the literature I've linked and working through a list of the
Comment: ...questions that need to be answered in order to get to a good decision (both in my answers and in the comments of others!) should get you a good understanding of what to look for.

Answer 2 (2 votes):
**A background on Raman:** Samzun (OP), buried in your query might be the taught belief that a Raman scatter spectrum is independent of the excitation wavelength, and this is mostly true, except for a scatter intensity that scales as (1/wavelength)^4 of the excitation. This wavelength-independence makes Raman useful for chemical species identification, and being able to chose the excitation wavelength makes it amenable for various carrier media, including water, for the analyte under observation. So while the ideal (theoretical) Raman spectrum is fixed, various pieces of your optics will _multiply_ their signature onto the spectrum. As noted in _cbeleites'_ response, a **dispersive grating** can add its signature. Unlike a simple prism, a grating's signature can be non-monotonic and is often oscillatory, especially in highly efficient holographic blazed gratings. Another significant contribution is the **quantum efficiency** of your detector. I suspect you are using a silicon detector for both your 532nm and 785nm excitation. Silicon's quantum efficiency drops off slowly and as it nears 900nm drops off precipitously to zero at 1050nm. At an excitation of 785nm a lot of the scatter will be at those longer wavelengths. Other pieces of the optics will also add signature to your spectra, e.g. anti-reflection coatings, filters, polarization dependence, etc. A lot of care must be taken to remove these various contributions. As you note, the carrier media (water in your case) can also be an absorber. But in your case for 785nm excitation, you will find that water absorption is not the main contributor, and the detector quantum efficiency is. If you know the signatures of the various contributors, you can divide them out to get the ideal (wavelength-independent) Raman spectrum of your analyte. **Water spectrum:** An additional question that you ask is about the background around and between the HOH and OH vibrational modes. I do not believe this has been properly addressed in the other replies, and unfortunately the answers are buried in academic literature. This additional background of water is also bothersome for the shot noise is adds across the whole spectrum. Polarization dependent Raman will be the easiest way to convince yourself this broad background in water's spectrum is real. The likely answer for this broad background is **hydrogen-bonding** between the water molecules. It loosely bonds multiple water molecules and you then have the vibrational modes of a larger amorphous superstructure. This is very complicated, and so only anecdotal and handwavey discussions are found in literature. This may be unsatisfying, especially to a theoretician, but in your practice it may not matter, e.g. if water's background stays constant and consistent and you know its input variables. Good luck observing!


================================================================================

