Discussion Chain ID: 73054
================================================================================

User asked: Slight FTIR band intensity changes across several different copper salt spectra: relevant for characterization?

I have several spectra of different samples which are probably all copper (neutral and/or basic) acetate(s) and which are very similar. I present them overlayed and (some of them) stacked: [](https://i.sstatic.net/k3Qn2.png) [](https://i.sstatic.net/VYThx.png) My question is: can I assume all the samples are qualitatively identical or do subtle/slight intensity differences in the same bands mean a qualitative difference? These are all spectra made with a pellet of KBr and I intend a qualitative characterisation to compare with literature, not a quantitative one.

Discussion:
Comment: Chemists prefer IR spectra with (remnant) %transmittance along the ordinate axis, and with one (or even two) brakes in the abscissa at about 2200 1/cm (or even a twice at about 1000 1/cm) instead of a straight linear axis; it renders the spectrum more legible. In most cases, the software of the spectrometer offers this and it is only a few clicks away. Is it a transmittance spectrum with a pellet of KBr, or a neat film/ATR? Do you intend a qualitative characterisation to compare with literature, or a quantitative one? Do the traces represent several batches of one (what?) intended product?

Answer 1 (1 votes):
(Answered after your edit of your question:) _Some_ variation in the location of the absorption bands (_vs._ 1/cm) and their intensities (_vs._ remnant transmittance) is unavoidable due to the interaction of your analyte (here: your copper salt/salts) with the matrix (here: pellet of KBr) compared to recording a neat film of your sample. The spectra depicted by the trace in magenta and blue seem to be a little bit different from the others, _perhaps_ due to some humidity (KCl exposed to air, or a sample not completely dried, or both, ...). On first view of the current representation, the other look very similar, compared with each other. * As your data represent infrared spectroscopy, figure out how to change the representation such that your ordinate is in remnant %transmittance instead of absorption. Let the spectrometer's software do this for you. For a visual inspection of the data, a split in the abscissa, too, for example around 2000 1/cm eases the visual inspection _a lot_ \-- equally something typically offered by the software of the spectrometer. (On good instruments, both changes do not alter the raw data collected by the spectrometer.) For example: [](https://i.sstatic.net/l4FJz.png) ([source](https://www2.chemistry.msu.edu/faculty/reusch/virttxtjml/Spectrpy/InfraRed/infrared.htm), the picture was edited by addition of horizontal lines) * Determine the barycentre of your absorption bands. Likely, as you start with IR spectroscopy, a deviation of _about_ 1/cm is not an issue (ask your instructor). In other words, if the peak pic by the software _suggests_ 1837.54 1/cm, a report like 1838 1/cm likely is good enough (ask your instructor). * For each _important_ absorption band, have a look about the shape, note if it is a particular broad absorption band -- in your example, around 3350 1/cm, for example. * Beside location, an other important information provided by IR spectroscopy is the _intensity_ of the absorption band. In the representation of %T, imagine three equal parts between the baseline on top, and the most intense absorption band on the bottom. A bit like the three horizontal blue dotted lines added to the picture above. The absorbtion bands in the first strip are then (semiquantitatively) classified as of _weak_ , the ones reaching till the second strip of _medium_ , and the ones reaching down to the third strip of _strong_ intensity. (As an example, the edited spectrum above contains an absorption band of medium intensity at about 3500 1/cm, and two strong ones at about 1700 and 1500 1/cm.) * _Compare_ localisation and absorption band intensity of the spectral data recorded by you, with data found in a reference (textbook, journal article, data base). Are all _important_ absorption bands found in your spectrum mentioned in the literature reference? Do you find all _important_ absorption bands described in the literature reference in your spectrum, too? Are there some _important_ bands in addition or missing? * Of course you may compare sample #1 with sample #2, with sample #3, etc., too. Only by growing experience you will learn what _important_ absorption bands in infrared spectroscopy are. Because -- in contrast to other spectroscopic techniques -- it is rather the exception than the general case that you may correlate each absorption band with a structural element in the structure of your compound. For future reference I suggest to record the IR data either already in the %T mode, or (second best) to convert them into this form. This allows you to check rapidly if even the most intense absorption band does not hit a transmittance of 0% (such an occurence were a rapid indication of a sample with to high concentration of X in your pellet of KBr.)


================================================================================

