Discussion Chain ID: 177435
================================================================================

User asked: What does it mean if a calibration curve regression doesn't cross (0,0)

I have been taking many lab courses in my second year of university and I have come across the technique of creating a calibration curve quite a bit. One thing I noticed in a lot of my plots is that many of them do not intersect at the origin (0,0) when I make a linear regression. [](https://i.sstatic.net/OwAkI.png) This is one of my more recent plots from an MP-AES experiment we did, and the part that I cannot wrap my head around is that the blank solution for this calibration showed basically zero intensity. But we see a completely different spike in signal in the 2ppm trial than the normal sensitivity of the trial that the slope shows. My questions are, what exactly does it mean that my calibration curve doesn't start at (0,0), and, why is it different from the sensitivity (normal slope of the points)? I have additionally gotten solutions that get negative concentrations when I try to use external standards (plugging the solution intensity into the equation and solving backwards) and I am wondering what kinds of errors could account for something like this happening in terms of external standards.

Discussion:
Comment: Well, you made error(s), most likely. Who knows what exactly.
Comment: I can understand that as the explanation for the negative concentrations, but what about the fact that many calibration curves don't start at 0,0 when doing external standards? Or did you mean both.
Comment: In the plot you've given, it looks like you might have saturated your detection method. If you do much lower concentrations (maybe 0-1 ppm), do you get a steeper slope?
Comment: We unfortunately only gathered data for a blank and 2 ppm+ solutions. But the part that was confusing is that the reagent blank showed, what is to me at least, an "expected" near 0 absorption at this wavelength.
Comment: Am I correct in assuming you are doing microwave plasma atomic emission spectroscopy and detecting potassium by its emission at, say, 766.5 or 769.9 nm? So a decent blank may well produce negligible emission there. Your graph does not include the blank. Hate to say it, but the data are pretty bad: big error bars, not linear response, and not good R squared. @Andrew may be right, but those big error bars make it harder to test his hypothesis.
Comment: By the way, it is almost always a bad idea to try to force a calibration curve to pass through the origin: https://chemistry.stackexchange.com/a/116224/79678. Doing so eliminates a possibility of learning the full story that the data are telling.
Comment: Yeah, I saw a couple of reports about how forcing it was bad but I was just wondering if there was any other explanation for why my absorbance might be so high. Also yes we did measure it at 769.9nm. I guess if it is mostly error it might be with making the solutions because the trial we did for copper was pretty good and had R2 of like 0.96.
Comment: I also get a calculated LOD of like 49 so I guess yeah something probrably went wrong
Comment: For given data, using linear approximation does not seem to be numerically justified. It seems the data series is curved toward the origin.
Comment: Thank you, after looking at it for a bit I also started to notice that it looks much more like a square root type of function.
Comment: Who is "we"? Are you the only one who ran this lab? Did others obtain similar results? Or did you aggregate the results and this is what is shown here? How did you determine the uncertainty in the individual measurements (what is the source of the error bars?)? What is your experimental protocol? Is this an instrument that is in good shape, well maintained? Who did the calibration, and when? Some of these qs might be difficult to answer, but more information will lead to a better answer.

Answer 1 (2 votes):
> I saw a couple of reports about how forcing it was bad but I was just wondering if there was any other explanation for why my absorbance might be so high. You are doing emission spectroscopy (AES= atomic emission spectroscopy) not atomic absorbance. Most well prepared calibration curves, or and carefully prepared standards have a very small intercept. What was the composition of the blank? > My question is what exactly does it mean that my calibration curve doesnt start at 0,0 and why it is diffrent from the sensitivity (normal slope of the points). I have additionally gotten solutions that get negative concentrations when I try to use external standards (plugging the solution intensity into the equation and solving backwards) and I am wondering what kinds of errors could account for something like this happening in terms of external standards. Calibration curves and most of the experimental analytical chemistry follows the GIGO principle, i.e., if the standards are not prepared with _utmost_ care, the sample analysis is a useless exercise. The only absolute way to test the accuracy is to use a reference material with a known concentration of potassium. If the experimental calibration curve produces the correct result, it means everything is working alright. If it produces wrong numbers and they do not match the reference, it means the experiment has serious errors. Negative concentrations simply imply the standard preparation was erroneous. Recheck all calculations, purity of the potassium salt, and of course, the blank. I also suspect ionization issues, the calibration curve is nowhere linear. Potassium ionizes very easily.

Discussion on this answer:
Comment: Thank you for your comment, In that particular lab I think we were also comparing it to FAES and I suspected that it might have to do with the temperatures that an MP-AES can reach and how that may cause loss of signal because it might strip electrons. That was my first time using one of these machines so now I really have an appreciation for the precision that we need in sample preparation. Thank you again for the responce.


================================================================================

