Discussion Chain ID: 43385
================================================================================

User asked: Peak Wavelength of Emission Spectrum of a Flame vs. Actual Flame Temperature

I am looking at a lab test of two variants of mixed hydrocarbon gas (57.8% $H_2$, 16.9% $N_2$, 12.4% $CH_4$, 1.11% water vapor, and 11.8% "unknown" gases) burning in oxygen which shows a discontinous emission spectra. The authors are claiming that since their graph shows a peak wavelength at 471 nm, the gas therefore has a peak flame temperature at 6132 K / 5859 °C / 10578 °F according to Wien's Displacement Law, which is absurdly higher than the adiabatic flame temperature of any known hydrocarbon gas. Now, I know that the spikes shown correspond to the $C_2$ and $CH$ Swan Band. Can someone provide a detailed explanation as to why this is not an accurate measurement of flame temperature? I should also mention the graph only ranges from 200 nm to 1100 nm, and that they fitted the emission spectrum to a blackbody curve. Could the peak wavelength actually be further into the infrared spectrum? Is fitting the emission spectrum to a blackbody curve even appropriate for a non-blackbody emitter such as a flame? [](https://i.sstatic.net/QbUfr.jpg) [](https://i.sstatic.net/x9Ud9.png)

Discussion:
Comment: Your skepticism seems very well founded. Wikipedia has "an acetylene/oxygen flame burns at about 3,773 K" https://en.wikipedia.org/wiki/Oxy-fuel_welding_and_cutting // Fitting to a peak in the spectra seems totally weird // Could you provide a reference to the paper?
Comment: http://www.magnegas.com/docs/MG-Flame-report.pdf
Comment: My suspicion is the authors chose an arbitrary peak and fitted it to that point. There is no rationale or equation they used in justifying this action.
Comment: I'm not an expert on measuring flame temperatures, but the method used seems totally wrong. You don't curve fit the whole spectra but rather take background measures (i.e not on a peak) on the high energy side and extrapolate to zero intensity. At least that is what I did for x-ray fluorescence and it worked well.
Comment: Another possible issue is perhaps they are confusing "flame temperature" with "color temperature" due to emission lines which I think is more likely. Just because a blue flame is blue doesn't mean it burns over 6000 K.
Comment: @MaxW I'm having trouble trying to explain that all in my head since I'm no expert on flames myself, but I still find it disingenous and sloppy for a serious laboratory to present data like this.
Comment: I'm not an expert but surely you need to *exclude* strong emission lines before fitting (it doesn't look like they have done this). Even then it seems like a stretch to assume the rest of the emission is black body (which is the only thing an emission curve fitting would make sense for).

Answer 1 (2 votes):
There seems to be a number of problems here. 1. spectrometers need to be calibrated in intensity on a wavelength-by-wavelength basis. Each part of the system has an efficiency which can vary with wavelength, but the grating response tends to be one of the strongest. Here is a random example of a grating efficiency curve from a Thorlabs catalog : [](https://i.sstatic.net/HZm5O.gif) You can see right away that a grating efficiency curve can impose it's shape on the measured spectrum, generating a shape superficially similar to a blackbody, but totally instrumental. Without calibration for efficiency, you just have ADC counts vs wavelength raw data! 2. Fitting of a Plank shape should be done to the spectrum without ignoring the peaks. In other words, a good fitting program will fit the shape you are looking for (Plank) PLUS fit the peaks at the same time. And don't forget to calibrate intensity first (see 1.)! [](https://i.sstatic.net/j0oK4.png) 3. " Is fitting the emission spectrum to a blackbody curve even appropriate for a non-blackbody emitter such as a flame?" Probably not unless you really know what you are doing. Flames are beautiful and complicated. There's thermal emission from gas in addition to peaks, but stronger is the blackbody-like emission from hot soot particles. Since there will be a distribution in temperatures within the 3D volume of a flame, you have a spectrum recorded from a significant fraction of the flame volume will include a variety of Plank spectra corresponding to different temperatures. So the answer is: you have to do fitting, remove peaks by fitting them simultaneously, and more importantly, you must calibrate the spectrometer intensity, because grating efficiency curves look strangely Plank-like, althouth they fall to zero much faster above and below their useful wavelength range. Also have to deal with considering the sample volume of the flame and whether or not it's true black-body emission or not. Those "peak" values, are closer to peak efficiency wavelengths of the spectrometer, and not really related to [Wein's displacement law](https://en.wikipedia.org/wiki/Wien%27s_displacement_law) at all!

Answer 2 (0 votes):
I calculated a small table of values using the formula for Wien's law: $$ \lambda_{max,nm} = \dfrac{2898000}{T}$$ where T is the temperature in Kelvin and the wavelength is in nanometers. I can't reconcile my table, the graph, nor the calculated temperature. For the top graph it looks like the cutoff is something like 350nm which would be rough 8250 Kelvin which is much greater than 6132 Kelvin which the authors cited, and much hotter than an acetylene/oxygen flame ($\approx$3773K). T(Kelvin) W(max) in nanometers 2000 1449 2250 1288 2500 1159 2750 1054 3000 966 3250 892 3500 828 3750 773 4000 725 4250 682 4500 644 4750 610 5000 580 5250 552 5500 527 5750 504 6000 483 6250 464 6500 446 6750 429 7000 414 7250 400 7500 386 7750 374 8000 362 8250 351 8500 341 8750 331 9000 322 9250 313 9500 305 9750 297

Discussion on this answer:
Comment: This is Ref. 1 used in the report: http://www.tempe.mi.cnr.it/zizak/tutorial/cairol06-flame-emission.pdf Fig. 7 shows "Emission spectrum of a rich pre-mixed butane-air flame." By the logic of fitting to a blackbody curve, then the peak flame temperature of butane-air becomes 9500 K, which is definitely not the actual flame temperature. I think my assertion over the confusion between "color temperature" and "flame temperature" is probably correct.
Comment: The lack of a control gas in the testing such as butane or acetylene make it difficult to accept their results. They referenced a butane spectrum off Wikipedia and it is nearly identical to the second graph I posted.
Comment: Revised comment: I'd add that 350nm or so is probably getting near to the UV cutoff for air. The oxygen in the air will absorb short wave UV, so UV radiation shorter than 350 would be attenuated quickly by air.
Comment: The authors stated that their spectrometer wasn't sensitive enough to detect emissions in the UV range. Mind you, the Ocean Optics HR2000 spectrometer is the same model I used when I took introductory physics and chemistry to analyze discharge tube spectra. You would think a dedicated optics lab would have access to better equipment. I also found different experimental setups for measuring flame temperatures but none of them involved using a single spectrometer.
Comment: I'm really stumped as to why the first spectra appears to cutoff at about 350nm. About 8200K seems way to hot. If you are trying to look for the minimum wavelength then as you go from visible to UV you'd have to consider the detector optics as well as the air absorption problem.
Comment: So am I, unfortunately. "Combustion Measurements" by Norman Chigler gives several methods of measuring temperature: https://books.google.com/books?id=aZGBBs7moYgC&printsec;=frontcover#v=onepage&q;&f;=false
Comment: Is the background luminosity (not on a spectral "line" of some sort) of a flame really black body radiation?!?


================================================================================

