Discussion Chain ID: 139564
================================================================================

User asked: Calculating the concentration of product in a solution based on the peak area of HPLC

Please help me resolving the following: > A sample solution, $\pu{0.24mg/mL}$ Compound A, is exposed to heat to form Compound B. From HPLC data report I have peak area for **Compound A** and **Compound B**. The STD concentration is 0.3 Compound A. The concentration of Compound A in sample solution is $\pu{0.20mg/mL}$ **Compound A** calculated using STD Compound A information. Which is the correct way to calculate concentration of **Compound B**? My advisor in the project I'm participating insist I must calculate the concentration of Compound B since is a degradation of Compound A in term of the amount lost of Compound A is the amount gained by Compound B (she call it a indirect method for calculating the concentration). I'm at lost! Originally I performed the calculation using a STD Compound A at the LOQ level of the impurity: $$\text{[Compound B]} = \frac{\text{Area Compound B} \times \text{[STD Compound A]}}{\text{Area STD Compound A}}$$ The new approach is calculate the [compound A] using a STD at the level of the sample and then indirectly calculate the [Compound B].

Discussion:
Comment: There is no way! You need to construct calibration curve for compound B or have a single standard injection for compound B.
Comment: @M. Farooq: I agree with you about calibration curves. Also, may need a calibration curve for compound B, if two compounds are structurally different (based on detecting method, usually an UV detector).
Comment: I think my advisor bold approach is based in Coumpond B is the enantiomer of Compound A (Doxycycline and 4-EPI-Doxycycline).
Comment: Okay, then his assumption is valid. The UV spectrum of enantiomers is **identical**.

Answer 1 (3 votes):
Your advisor is making a pretty dangerous assumption (although sometimes pharma people do it). The assumption is that the impurity is structurally similar to the main compound and its UV-Vis spectrum is almost similar i.e., their extinction coefficients are the same. The idea in your formula is that the calibration curve for A will be same as B. Well that might be a first order approximation when the impurity is known to be structurally similar. If B is a known compound, you should buy a separate standard, and make a calibration curve.


================================================================================

