Discussion Chain ID: 158511
================================================================================

User asked: Two different definitions for chemical shift (NMR)

Looking online I seem to have found 2 different definitions for chemical shift $\delta$ (in the context of NMR). The first definition uses the chemical shift of the reference in both the numerator and denominator: $$\delta = \frac{\nu - \nu_{ref}}{\nu_{ref}}$$ This can be found on the wikipedia page here: <https://en.wikipedia.org/wiki/Chemical_shift> The second uses the frequency of the _spectrometer_ as the denominator, i.e. $$ \delta = \frac{\nu - \nu_{ref}}{\nu_{spec}}$$ This can be found, for example, here: <https://chem.libretexts.org/Courses/Athabasca_University/Chemistry_350%3A_Organic_Chemistry_I/13%3A_Structure_Determination-_Nuclear_Magnetic_Resonance_Spectroscopy/13.03%3A_Chemical_Shifts> Now, I am aware that the "frequency" of the spectrometer is really just an approximate frequency at which protons tend to resonate in that field, and so these numbers would be roughly similar. My question is, is there an actual difference between the two (which is largely ignored because they give similar values) or are they somehow identical? Anyways, thanks in advance.

Answer 1 (9 votes):
First, let's be careful about what the quantities refer to. The _reference frequency_ , of course, refers to the frequency of a reference compound. In the case of $\ce{^1H}$ or $\ce{^{13}C}$, the reference frequency is given by the methyl groups of tetramethylsilane. The spectrometer frequency, however, could be considered slightly ambiguous. It may either refer to the _carrier frequency_ , which is placed in the middle of the spectral window. If your spectrum runs from $\pu{0 ppm}$ to $\pu{10 ppm}$, then the carrier frequency is $\pu{5 ppm}$, or whatever that corresponds to in terms of $\pu{MHz}$. On the other hand, the _spectrometer frequency_ itself usually refers to the frequency that corresponds to $\pu{0 ppm}$. (In TopSpin, these two parameters are respectively called `SFO1` and `BF1`; [this link](http://www2.chem.uic.edu/nmr/downloads/bruker/en-US/html/Avance%20Beginners%20Guide/en-US/18014398881161483.html) contains more explanation about these parameters.) Given that the reference frequency is _defined_ to be at $\pu{0 ppm}$, it follows that the reference frequency is actually the same thing as the spectrometer frequency, which—in a strict sense—makes this question trivial to answer: it can be either one, since they're both the same thing.* Officially, the definition is with respect to the reference. As per the [IUPAC Gold Book](https://doi.org/10.1351/goldbook.C01036): > **chemical shift** , $\delta$ (in NMR) > > The fractional variation of the resonance frequency of a nucleus in nuclear magnetic resonance (NMR) spectroscopy in consequence of its magnetic environment. The chemical shift of a nucleus, $\delta$, is expressed as a ratio involving its frequency, $\nu_\text{cpd}$, relative to that of a standard, $\nu_\text{ref}$, and defined as: > > $$\delta = \frac{\nu_\text{cpd} - \nu_\text{ref}}{\nu_\text{ref}}$$ The above equation assumes that the numerator is measured in $\pu{Hz}$, the denominator in $\pu{MHz}$, and the chemical shift in $\pu{ppm}$. (This implicit convention is a bit annoying, but when actually doing NMR spectroscopy it comes quite naturally.) Why not the carrier frequency, though? The carrier frequency is also proportional to the field strength, so it has no bearing on the argument that the chemical shift should be independent of field strength. I'd argue that it doesn't make sense to use that because that's a parameter that you can change, by varying the centre of the spectral window. If you acquire two different spectra with different spectral windows, it doesn't make sense that the chemical shifts should differ between the two of them. On top of that, we can motivate this choice by comparing the chemical shift formula against the expression for the _relative error_ of a measurement: the relative error $\eta$ is given by $$\eta = \left| \frac{x - x^*}{x^*} \right|$$ where $x$ is the measured value and $x^*$ is the true value. Note that the denominator contains the quantity that we are trying to compare it against. In the case of the chemical shift, it is not an error that we are trying to measure, but rather a "relative deviation" from the reference frequency. It is logically consistent, therefore, that the denominator contains the quantity that we are trying to compare against. * * * **Footnote** * Philosophically, however, they're different things. For a given magnetic field strength, the frequency of TMS is a physical constant. That constant is the reference frequency. However, when calibrating a spectrometer, you'll need to determine what that reference frequency is, and then set the spectrometer frequency to that number. If you don't determine it accurately, then the spectrometer frequency will be off (and the resulting spectra will be [shifted by a little bit](https://i.sstatic.net/DFa6H.png)). Using an analogy: let's return to 2018, before the redefinition of the kilogram. At that time, the _reference kilogram_ was defined by [a block of metal](https://en.wikipedia.org/wiki/International_Prototype_of_the_Kilogram); so you could say that the _reference mass_ was the mass of that block. This is a quantity that is a fixed constant, and can't be changed. On the other hand, if I had to calibrate a new scale, the _scale mass_ might be said to be the mass that makes the scale show $\pu{1 kg}$. It follows that the scale mass _should_ be the same as the reference mass, for a well-calibrated scale. However, for a poorly-calibrated scale, this may differ.

Discussion on this answer:
Comment: So if you have a signal at 1.000000 ppm, and you choose the wrong denominator, it would come out as 0.999995 ppm ?
Comment: @orthocresol The 2001 recommendations in [PAC](https://doi.org/10.1351/pac200173111795) recall that the 1972 rules included a factor $\times 10^6$ for proton NMR (eq. 4), then *define* $\delta$ as in the first equation by the OP (eq. 5) as more general (and open to ppb «for some isotopic effects») and show one *to calculate* $\delta$ in ppm in [Hz] (numerator) and [MHz] denominator. [IUPAC 2008](https://doi.org/10.1351/pac200880010059) then *only* shows the first equation of the OP (eq. 1) in lines of «"ppm" is exchangeable with $\pu{\times 10^{-6}}$ [...] as is "%" with $\times 0.01$».
Comment: @Buttonwood Thanks for that. Just to be clear, do you just mean to clarify the missing factor of $10^6$ (which I admit, I didn't even think about!), or there something more than that?
Comment: @KarstenTheis I actually have some data that illustrates this: https://i.sstatic.net/DFa6H.png The red spectrum was divided by a slightly larger value and the blue spectrum was divided by a slightly smaller value (this took place before/after recalibrating a spectrometer frequency). The difference between the peaks is ca. 0.0005 ppm.
Comment: @orthocresol Preparing an answer, I encountered this Hz/MHz division in one of the equations. Maybe this is a reason why we find the computation of $\delta$[ppm] with division over spectrometer frequency (in MHz) $\times{} 10^6$. But it is not about an error in your answer I'm fine with.
Comment: @KarstenTheis Hmm, thinking about it a bit more, I suspect the numerator plays a larger role. I reason about it this way: the denominator is a large, positive number anyway, so hardly matters whether it's 699.90 or 699.95 or 700.00. The numerator, however, refers to a frequency difference (i.e. difference of two large numbers) which can range from anywhere between, let's say, 0 Hz to 20000 Hz. Shifting the reference frequency (or recalibrating) will affect the denominator by a tiny fraction, but the numerator may shift by quite a bit.


================================================================================

