Discussion Chain ID: 47771
================================================================================

User asked: Is it possible for a liquid chemical to have a pH value lower than its aqueous solution?

During the QC check of one of the products that we make as a company, the product had a $\mathrm{pH}$ of $10.44$ on one $\mathrm{pH}$ meter and $10.20$ on the other. (The product is a liquid, so it's straightforward to immerse $\mathrm{pH}$ probes to take the measurements.) For QC of the product, it is required to make a $1\%$ solution using demineralised water and to take a $\mathrm{pH}$ reading. The first $\mathrm{pH}$ meter read $11.72$ for this $1\%$ solution, and the second meter read $11.41$. How is it possible for the raw product's $\mathrm{pH}$ to be lower than the $1\%$ solution's $\mathrm{pH}$? The product is supposed to be basic and not acidic at all.

Discussion:
Comment: pH is a property of solutions. Bronsted-Lowry bases work by removing H+ from the solvent. If there is no H+ to take, the base cannot act like a base. Is your chemical a liquid when pure?
Comment: Yes it is a liquid. It is used as a high pH boiler water treatment chemical. The previous batches of this product have always had a pH value greater than 12 as a raw product. This is the first time in 11 years of making this product that this has happened.

Answer 1 (2 votes):
The concept of pH tends to break down when water is not greatly in excess of the solute (e.g. if your product is a highly concentrated amine base). The apparent pH of your product, when measured using a pH meter or an indicator, may be relatively low (seemingly less basic) if the solution has a less polar environment than that of pure water--because ionization will be less favorable. If no water is present at all, the product might still give a pH reading with a meter or indicator, but the term "pH" in this case would be irrelevant. It always helps to talk specifics with chemistry, so I am curious as to what the ingredients of the product are.

Discussion on this answer:
Comment: If the product has no appreciable amount of water, then dunking the pH electrode in it could ruin the electrode. The electrode works by having a hydrated layer of silica on the glass bulb.
Comment: Worst of all, the electrode would still continue to produce some readings (seemingly sensible, but in fact random).
Comment: Unfortunately the formula is proprietary so I cannot just post it here..... If you gave me a way to contact you on a personal level then I could divulge the ingredients but not the specific recipe for the product.
Comment: @DylanLeSar Keep your company secrets...the principles apply broadly in any case.
Comment: @iad22agp thank you though for the understanding and the answer provided. I have also brought it to out technical consultant's attention and he seems to agree with you. A person never stops learning ;)


================================================================================

