Discussion Chain ID: 109816
================================================================================

User asked: what the solution of this problem, negative absorbance values in UV-vis spectrum

In my project, I take the UV-visible spectrum of reaction between pyridinium chlorochromate and N-acetylcysteine. In some wavelengths, the absorbance value is negative for example it is (-0.006, -0.002, -0.003){negative but very small}. Also when I take the spectrum of pyridinium chlorochromate alone, there are absorbance values in negative {negative but very small}. The blank is NaClO4 solution. what are the reasons for these negative values? And can I negligible these values? Thank you...

Discussion:
Comment: As they say in Russian, _If you see a "buffalo" sign on an elephant's cage, do not trust your eyes._ Same thing here. These values can't be. They are experimental errors. We may muse about the reasons, but the most important reason is that **all experiments have errors**. Bear with it.
Comment: Could be the sample (the absorption of soluted pyridinium chlorochormate against pure water different to the one of soluted NaClO4 against pure water) in your region of interest. Could be instrument's design (two beam / simultaneous vs. consecutive measurement on single beam), (in)stability of the light source (e.g., age of the lamps), integration time (nm/min sweep rate), cell (glas/polymer). Balance the intensity of bands of interest vs. random fluctuations each recording has, e.g. by varying sample concentrations / cell length or average multiple spectra "recorded" (rather than "taken").

Answer 1 (2 votes):
The negative values can arise due to two major reasons: You did not mention that whether you are using a single beam instrument or a double beam instrument. In a double beam instrument, absorbance of the sample and the blank is continuously measured. In a single beam spectrophotometer, A(blank) is recorded first and later subtracted. In each case, the output value at each wavelength is: Abs (measured) = A(sample)- A (blank) If at any region of the spectrum A(blank) is larger you get slightly negative absorbance values. This could be due to several reasons including refractive index differences. Under ideal cases, the cuvet which contains the blank is same as cuvet containing the sample or both of them should be optically matched i.e. they should show _identical_ absorbance readings at given wavelengths. Another reason to see random negative values is that noise is a part of _any_ instrumental analysis. Eliminating all sources of noises, you still get stuck with shot noise, which originates due to statistical counting of photons. That is the very nature of detection especially in spectroscopic measurements- perhaps the ultimate limit in analytical sciences.


================================================================================

