Discussion Chain ID: 58918
================================================================================

User asked: How to judge accuracy and precision of measurement devices?

I am unsure about the accuracy and/or precision of measuring instruments. For example, if we have a graduated cylinder, a pipette and a beaker, how and why can we assert them that they can be used with accuracy, precision, or both?

Answer 1 (0 votes):
There are too many measurement devices to give an answer for all types. However, assuming the scope of your question is limited to measuring fluid volume the way this can be done in a professional laboratory is simple. You will need: (1) high purity water (2) a high precision balance which has been recently calibrated (3) a type of container called a weigh boat 1. Tare the weigh boat. This means place the weigh boat on the balance and (if the balance is electronic) press the tare button. This will cause the balance to read zero. (With the old pivot beam style balance another identical weigh boat could have been placed on the other end of the balance to level it. These pivot beam balances are rarely used in modern lab practice as most balances are electronic.) 2. Using the measuring tool in question, measure a certain volume of high purity water. Pour the measured quantity of high purity water into the weigh boat on the electronic balance. The balance should read the mass of the water. Assuming the temperature of the lab is close to 20°C divide the mass of the water by 0.9982 g/ml. This is an accurate way to tell how many ml were actually measured using the tool. 3. Repeat the measurement three times. Three true measurements can be used to determine an average and standard deviation. The difference between what was measured and the average of the three true measurements is related to the accuracy of the measurement. The standard deviation is related to the precision of the measurement. If a pipet is supposed to dispense 10 mL and it actually dispenses 15, 16, and 15 mL then it has high precision (because the numbers are close.) Its accuracy is way off because the average of 15.3 mL is half of an order of magnitude different than 10. It could be used, though it would need to be either calibrated so the measurement is correct or the end numbers adjusted for the inaccuracy. If a pipet is supposed to dispense 10 mL and it actually dispenses 5, 9, and 16 mL then despite being accurate, its precision is off. Because the precision is off, this pipet is too unreliable to be used in scientific practice. Realistically, if a pipet were supposed to dispense 10 mL and it dispenses 10.0010, 10.0000, and 9.9998 mL then it would be suitable for lab work. Because both the accuracy and precision is high, this measurement is said to have high trueness. Trueness -- which is both accuracy and precision -- is desirable in a good measurement. It is always good practice to verify the calibration of any measurement before using the measurement to generate publishable data. Recently published data in high-quality publications is a good source of information to judge if a certain set of data has acceptable trueness.

Answer 2 (-2 votes):
The term precision means two different things. The number 1.23456789 has more "precision" than the number 1.23. That is, there is a mathematical (and computer science) meaning that is different than the one used by physical scientists. In physical science, precision means the property of reproducibility of a value or result. If I am precise, I get the same answer every time (or nearly so). If I am accurate, I get the correct answer. Here is one example: I have a machine which counts people going through the door. Whenever someone walks through, it adds 1.1 to its total. This machine is precise but not very accurate. I modify it and now it adds between 0.95 and 1.05 to the count each time someone walks thru. It is definitely _not_ more precise, but may or may not be more accurate.


================================================================================

