Discussion Chain ID: 49720
================================================================================

User asked: How to explain two NMR spectra of the (allegedly) same compound with slightly different shifts?

For my synthetic lab work I synthesised a Horner-Wadsworth-Emmons phosphonate; for the sake of the question let’s assume it to be the one presented in the image below. When first synthesising it, I of course recorded NMR data. I performed a Horner-Wadsworth-Emmons reaction using a tiny bit of the phosphonate and kept the rest in the fridge. One month later I retried the HWE and got very bad yields, so I decided to record another NMR spectra. The two spectra themselves look identical and the signals (those that aren’t multiplets) also show the same coupling constants. However, when referencing the spectrum to the chloroform signal ($7.26~\mathrm{ppm}$), _all_ signals are shifted by the constant value of $\approx 0.07~\mathrm{ppm}$. I decided to check the old literature spectrum from 1986, but it was unhelpful: It only reported multiplets such that both of my spectra comfortably fall well within the multiplets’ limits. Considering that the two spectra _look_ identical and that the difference between two signals corresponding to the same hydrogens _is_ identical as are the coupling constants, I want to assume that both spectra show an identical compound. However, the $0.07~\mathrm{ppm}$ shift for everything save chloroform still irritates me. Both spectra were recorded at $500~\mathrm{MHz}$. The parameters of the earlier one state a temperature of $25~\mathrm{^\circ C}$, the later one was apparantly measured at $25.1~\mathrm{^\circ C}$. However, I don’t think that that could explain the shift. **What could explain this shift difference?** (My first thoughts were some changes to the compound that only affect the phosphorus atom, but aside from deoxygenation being chemically unlikely shouldn’t it also cause _random_ chemical shift differences including slightly different coupling constants? In fact, shouldn’t _any_ chemical modification cause the shifts to change in a seemingly random fashion depending on the proximity of hydrogen in question and entity modified?) After checking because I was asked in the comments: The later spectrum (the one that has higher shift values) has broadened peaks. That includes the chloroform peak.

Discussion:
Comment: Do the shifts change for carbon and phosphorous?
Comment: For reference, the later spectrum included a dichloromethane peak that stemmed from the bulb of the pipette I used for pipetting chloroform. The former didn’t, so I wasn’t able to check if dichloromethane was shifted, too. A very minor peak at $\approx -0.02$ or $\approx -0.09$ *was* shifted.
Comment: @Lighthart I didn’t record carbon and phosphorus spectra to be quicker.
Comment: Do the peak widths change at all?
Comment: @Lighthart edited the answer in.
Comment: Maybe the solvent signal is shifted? You have not used an internal standard like TMS?
Comment: @Mart As far as I know today’s NMR relies on the solvent signal for referencing and TMS is almost never added … At least, I hardly ever added any.
Comment: If the peak widths are different, that means the machine's field has drifted a little. And as a result, somewhat wider peaks (or worse shim) may be modestly adjusting the digital calculation of peak maximum (and hence chemical shift). Wider peaks would provide evidence that this is a measurement problem, not a chemical purity issue. That being said, you indicate you are not telling us exactly the compound synthesized, so we can't really judge if 0.1C will impact for reasons of conformational/rotomer changes.
Comment: @Lighthart I think that the first bit is the answer I am looking for and I am relieved that that may be the case. About the compound I actually used, it is very similar but not identical to the one I depicted here. And if I drew it for you, you would probably agree that conformation or rotamer changes would not be a logical conclusion. ;)
Comment: If you're goal is synthesis, I think you just botched your reaction, try it again. If your goal is a rigorous description of the deviation, you've got some number crunching to do.
Comment: @Lighthart the former ;) I do also have another explanation on why the reaction failed but I was so confused at this spectrum today that I was too scared to redo the HWE ;)
Comment: Another possible reason is that the shift is caused because of aggregation. Not unreasonable to happen for a phosphonate but on the other hand the shift you observe is too wide to be explained by aggregation alone? Have a look at this [paper ](http://pubs.acs.org/doi/pdf/10.1021/jm400535b) for example. I only know about this from its use as an assay in med chem so I couldn't help with more specific info.
Comment: How large was the DCM peak? Maybe I missed it, but which direction were the signals shifted: lower or higher ppm than before? Also, did you see any other signals that were new? Maybe some water or an alcohol that could hydrogen bond with some of oxygens here?
Comment: If all non-solvent signals are shifted by exactly the same amount, I would pick a shift of the solvent signal itself as the most likely source. If this were measured in H2O/D2O, I wouldn't be surprised at all, but the CDCl3 chemical shift is not as temperature sensitive as the H2O/D2O one.
Comment: I agree with @MadScientist. The solvent could shift due to presence of methylene chloride if there is a significant amount. Probably have to be a lot, though.
Comment: @Senders The DCM peak is really minute. You could publish the spectrum because it wouldn’t have been seen in normal scroll intensity. The new spectrum had slightly higher ppm values.
Comment: @MadScientist And that’s the reason why I always add methanol to $\ce{D2O}$ NMRs ;)
Comment: That just seems so strange. How could the magnetic properties of one nucleus change without any others changing? Even if the probe had drifted, I would expect *everything* to move, not just the solvent peak.

Answer 1 (2 votes):
That's a pretty small chemical shift difference. Keep in mind that chemical shifts depend on temperature, concentration, pH, and other factors. The concentration difference is the most likely, but I wouldn't rule out T effects. If you're running it in a different batch of CDCl3 (or even the same batch, but it's aged), the pH could be different as CDCl3 becomes acidic over time. If you used identical settings when collecting the spectra (e.g. using a default Proton) then you can measure the signal-to-noise ratio of two identical peaks and get an estimate for how different the concentrations are. You can also simply test how sensitive your chemical shifts are to concentration by making up a more concentrated sample (or diluting your current sample). Or try changing the temperature with the VTC and ramp it up to 25.5 or 26.0 C and see how much the chemical shifts change (referencing everything to CHCl3 for consistency). If your compound were reacting with something, there would be a much more dramatic change in the chemical shifts than 0.07 ppm.

Discussion on this answer:
Comment: Actually I never had a shift difference *as* large … but I never had phosphonates.


================================================================================

