Discussion Chain ID: 75241
================================================================================

User asked: Compare concentrations using LC/MS

Suppose I run a reaction and then pass the results through LC/MS to find the conversion. One idea could be to compare the peaks of the reactants to the peak of the product. But: * MS peak will depend on the levels of ionization of each of the compounds * UV spectra depends on the molar extinction coefficient of the compounds So it looks like I can't compare the peaks. Is this right and if so what are the possible ways of finding the conversion?

Discussion:
Comment: You;re completely right, its one of the 'issues' with trying to monitor reactions by LCMS. If its a reaction you plan on running a lot, its not too much effort to measure the UV absorption and then calibrate the LC results, but failing that, GCMS or NMR are both viable options
Comment: Even in GC one has to calculate something akin to absorption factors which I don’t know the name of right now.

Answer 1 (1 votes):
You can indeed find the _conversion_ (not the absolute concentrations), even without calibrating the system. At time t = 0, take an _exactly measured volume_ of reaction mixture (e.g. 1 mL) and dilute it in a volumetric flask with water and/or acetonitrile to an _exact volume_ (e.g. 100 mL). Inject an _exact volume_ (e.g. 50 uL) of the diluted solution in the LCMS and note down the (UV-based) peak area of the limiting reagent. This is your ${A_0 = \epsilon \cdot C_0 \cdot k}$, where ${C_0}$ is the concentration of your limiting reagent at this time, and ${k}$ is a constant related to the sample size and dilutions. At any time t, do exactly the same as above, and you get an area ${A_t = \epsilon \cdot C_t \cdot k}$. Under these conditions, ${k}$ is the same, and of course so is ${\epsilon}$. As you know, the conversion is ${y = 1 - \frac {C_t}{C_0} = 1 - \frac {A_t}{A_0} }$.

Discussion on this answer:
Comment: Could you describe what A and ϵ are? And if this is a well-known equation, could you please name it and/or give the references?
Comment: It's based on [Beer-Lambert](https://en.wikipedia.org/wiki/Beer%E2%80%93Lambert_law). ${A}$ is the the LCMS peak area from a UV detector, which is what I wrote, and is proportional to [absorbance](https://en.wikipedia.org/wiki/Absorbance). ${\epsilon}$ is the molar extinction coefficient. Not sure why my answer got a-1 mark, considering that it answers the question (how to measure conversion by LCMS).
Comment: Thanks! I don't think we'll be able to do the measurements at t=0 in our case, but still this clears some things up for me.
Comment: In some special cases you need no t=0 measurement. E.g. if your reaction is 1st-order (or at least pseudo-1st order), plotting ${log(A_t)}$ vs t should give you a straight line, from which you can extrapolate ${log(A_0)}$. Ordinary reactions (zero order, 2nd order, etc) should all be amenable to such treatment. With different functions of course.


================================================================================

