Discussion Chain ID: 171519
================================================================================

User asked: What is the correct unit of a RI detector used for HPLC analysis?

I analyze sugar compounds in soft drinks with HPLC using refractive index detection (RID) on Agilent 1260 Infinity Binary LC ([System User Guide (PDF)](https://www.agilent.com/cs/library/usermanuals/public/G1312-90303_BinaryLC_USR_EN.pdf.pdf)). I am not sure which unit to put next to the $y$-axis of the chromatogram. Is it RIU, μRIU, or something entirely different? [](https://i.sstatic.net/e4r6S.jpg)

Discussion:
Comment: The [detector](https://en.m.wikipedia.org/wiki/Differential_refractometer) is measuring change in refractive index of the solution.
Comment: Yes, this is for the UV detection. Did you check the model and make of the detector? Is it https://www.agilent.com/en/product/liquid-chromatography/hplc-components-accessories/hplc-detectors/1260-infinity-ii-refractive-index-detector

Answer 1 (7 votes):
Good question. Now that you have mentioned the instrument name (Agilent 1260 LC), it is easy to locate Agilent's "Refractive Index Detector Agilent InfinityLab LC Series User Manual." It is always good to mention the entire set-up. Some detector display microvoltages on the y-axis on UV detectors. Intuitively, you can see refractive index can not be negative (you have a negative dead time dip) and refractive index values cannot range in thousands. The signal is definitely not absolute refractive index (_n_). It is basically $\Delta n$, and in the jargon of detector, it is "Refractive Index Units". Note that refractive index has no units. The reason for asking for the model was to check what the manual says about the output and what is the scaling factor. It is nano-refractive index units, and the detector is calibrted with respect to a certain concentration of a glucose solution vs. pure water. On page 19, the manual states, > When sample elutes from the column into the sample cell the refractive index of the cell contents changes. The change in refractive index deflects the light beam as it passes through the flow cell resulting in an unequal amount of light falling on each diode. The change in current from the diodes that this causes is amplified and used to produce the calibrated detector signal. This signal expressed, as **nano Refractive Index Units (nRIU),** corresponds to the difference between the refractive index of sample in the sample cell and the mobile phase in the reference cell. Secondary note: You have overloaded peak shapes, they should not front. Dilute the sample.

Discussion on this answer:
Comment: Thank you so much for taking a look at my issue! So, the unit of this specific RI detector is nano Refractive Index Units (nRIU), did I understand that correctly? (I'm not a native English speaker and this is just my high school project)
Comment: Yes, it is plotting the difference of refractive index of mobile phase and your analyte multiplied by 10$^9$.


================================================================================

