Discussion Chain ID: 163086
================================================================================

User asked: What makes carbon dioxide gas "sticky" in pipes?

There's some perceived wisdom where I work that when flowing $\ce{CO2}$ gas in pipes it acts somewhat "sticky", in the sense that it lingers around longer and takes more $\ce{N2}$ or air to purge out than, say, methane or $\ce{N2}$. Unfortunately, no one here seems to know the physical or chemical basis for this observed result. I do have some quantitative data through various flow sensors and IR gas sensors that show it really does stay around longer than $\ce{N2}$ or $\ce{CH4}$, so I don't think it's due to bad observation. As you can imagine, googling "why is $\ce{CO2}$ sticky" is not very helpful. If anyone can explain to me why this happens, I'd be very grateful, or even just point me in the right direction to do my own research. * * * ### EDIT: Thanks for your responses so far. Here's some more info on our gas system: we buy our $\ce{CO2}$ in large bottles at ~400 psi. We then feed it from outside through a pressure regulator that regulates it down to 2 bar (~29 psi) absolute pressure. All the pipes until this point are stainless steel. It's then fed through a proportional valve and the pressure is held at 1030 mBar over a set of flow and gas sensors, then exhausted to atmosphere. The pipes in this section are a mix of Polyurethane tubes, and two proprietary rubber-like materials called Tygon and Viton tubing. These are held together with nylon fittings. All of this is at ambient temperature. In these pipes, only $\ce{N2}$, $\ce{CO2}$, and $\ce{CH4}$ flow. Here are some graphs I collected of the system: [](https://i.sstatic.net/vnypW.png) In this graph, the green & brown lines are each gas being flowed in the system. The two blueish lines are a rough analog of how much $\ce{CO2}$ is detected in the system. As you can see, even after 5 mins of methane and N2 purging, and continuous intermittent $\ce{N2}$ purges, $\ce{CO2}$ still creeps back into the system when no gas is flowing. Here's a Graph of Gas Flow [](https://i.sstatic.net/vyRVR.png) The [Flow In] sensor is just after the 1030 Prop valve, and the [Flow Out] sensor is just before the exhaust. They are both identical Wheatstone bridge thermal flow sensors. As you can see, there's a marked increase in flow loss (Flow In/Flow Out) from 2% to about 7% when CO2 is flowing, and some very strange transient spikes when I switch gasses to/from $\ce{CO2}$. That's most of the data I've collected. Anecdotally, I've also noticed that the $\ce{CO2}$ valve is significantly harder to turn than the $\ce{N2}$ or $\ce{CH4}$ valves. I hope this helps with the problem, and thank you again for your help so far.

Discussion:
Comment: There are some cross-country CO2 pipelines ( like 18" diameter). They pose unusual fracture concerns for the steel pipe. While following the technology , I never encountered "sticky" considerations. Operating pressures are above the critical point.
Comment: @Andrew I have added some more supporting information. Let me know what you think.

Answer 1 (28 votes):
$\ce{CO2}$ solubility in water is about 100 times higher than other gasses. So if there is any moisture on the walls, $\ce{CO2}$ is much more likely to stay there. $\ce{CO2}$ is also much more chemically active, forming lots of compounds through its carbonic acid. Some of these compounds are reversible, can absorb and emit $\ce{CO2}$ as its partial pressure or temperature changes. There is likely water or mineral contamination in your system.

Discussion on this answer:
Comment: Indeed, unless you can pump out the lines to less than 1E-6 Torr, you have water adsorbed on the walls of your piping.
Comment: @JonCuster baking the lines helps too, but I suspect the OP's system isn't set up for that and it would be harder than getting a turbopump on there. The materials may not be suitable for proper heating anyway. Even with a pump it would be much quicker with some heat.
Comment: Also with a vacuum pump and voids separated by long runs, a process of pump out, N2 purge, pump out etc. can be more effective than simple pumping (He purge in my case as N2 is one thing I'm trying to get rid of to avoid it freezing). And optimising changeover processes so ambient air never gets in
Comment: @ChrisH - Pump/purge cycles work wonders too... So many tricks of the trade out there, and so few places to learn them anymore... I just queried an editor at the Journal of Vacuum Science and Technology if they would be interested in a vacuum technique article based on some work we did on a project. They did say OK, but that there was no identified area editor for those anymore...
Comment: @JonCuster a varied engineering/science career has left me well placed to learn on the job - just as well as there's no other way. But we need the time and space to fiddle about without causing big problems and that opportunity seems hard to find these days
Comment: @JonCuster Thanks for the info, looking at my data again water adsorption does seem to fit. Do you have any idea how the water might have gotten into the system in the first place? Also, as you suspected I'm not confident my system is suitable for either vacuum pumping or heating. I have flexible pipes which might collapse in under a vacuum, and a lot of my pipework is intertwined with non heat-proof components. Do you have any advice on what techniques might work here, or where i might look for this kind of information?
Comment: @AlexanderJohnson circulate a gas over dessicant to remove the moisture if heating and vacuum are not an option. No need to purge the gas during this, if dessicant absorbs the water, circulation is enough. Once water is absorbed from the system, make sure feed gas is dry in the future. Also could separste CO2 and other gasses as much as possible, for example by using separate dessicant chambers for them. When water is removed CO2 will behave more similsrly to other gasses and less purging will be needed.
Comment: @AlexanderJohnson you might want to read a little about vacuum systems and what's required to get to ultrahigh vacuum. Any time a surface is exposed to atmosphere with water in it, the water will stick to the walls, and slowly outgas from them until it reaches equilibrium and the partial pressure of water in the gas is at the vapour pressure of water at the relevant temperature. Especially bad is if the system is below the dew point when this happens, then you get mass condensation of water, so there's oodles of water to outgas even if you start flushing the vapour out somehow
Comment: @AlexanderJohnson (I'm not saying you need to get to ultrahigh vacuum, I'm just saying the principles might help you.) Does the CO2 you're getting come with a purity specification for water and other contaminants? Simply flushing the system by flowing through a dry gas (nitrogen is usually cheapest I think) for a while after any time it's exposed to atmosphere might help, and this will work better at higher temperatures even if it's only by 10 or 20 K (look at a curve of H2O vapour pressure vs temperature)


================================================================================

