Discussion Chain ID: 40119
================================================================================

User asked: Calibrating Motic Camera to Stage Micrometer

Whilst calibrating my Motic Microsope's camera, I am able to calibrate the 10x object lens against the stage micrometer accurately. However when I calibrate (using the Motic 2.0 Plus software) the 40x object lens, it always sets the ratio incorrectly. For example, in the calibration panel, I load a photo of the scale that I took and then I draw the circle so that it measures the pixels both in height and width and then I tell it that the circle is 10 µm. But it doesn't seem to get the ratio correct. When I did this for the 10x lens it worked perfectly. So I took a photo of the scale with the 40x object lens and then went to calibrate that. [](https://i.sstatic.net/Inr1l.png) [](https://i.sstatic.net/imtJP.png) Here you can see how the calibration is performed: you position a circle over the photograph and tell it the size in µm (it's rather large, sorry): [](https://i.sstatic.net/sN4tm.png) Here you can see the measuring line out of whack. It gives me 8 µm for a length that should only be 4 µm. [](https://i.sstatic.net/o7WtL.png) Am I missing something?

Answer 1 (2 votes):
It looks like the shape is an ellipse rather than a circle. The ellipse has major and minor axes, which are described in terms of half-axes (177 and 175 pixels respectively). Is it possible the software is incorrectly using the half-axis measurements, whereas when you draw a perfect circle it will correctly use the diameter? This is consistent with the x and y scale numbers being different from each other. So I would suggest re-calibrating with a perfect circle. Hope this helps.

Discussion on this answer:
Comment: When I calibrate, should I calibrate to the edges of that black box? It's supposed to represent 4µm. If they were tick-marks I would think to measure to the insides... It's kind of confusing. Do I count the tick mark or not? Or do I measure to half of it?
Comment: With my own calibration slide that has a linear scale (like a ruler), I measure from the left side of one tick mark to the left side of another one. I'm not familiar with your calibration slide, so I don't know what the outer lines represent. Do you have any documentation or are there other areas of the slide that are more clearly labeled?
Comment: My Mycologist says that it's supposed to be 10 microns from one tick to the next - whether left to left as you say (which is easy to see - thanks!) or from half to half of the tick. So I tried this by capturing the scale with the 10x lens and calibrated four ticks to be 40 microns. But when I measure from tick mark to tick mark I get 18.10 microns. I'm still missing something.
Comment: My Stage Micrometer has five calibrators: a circle at 1.5mm, a circle at 0.6mm, a scale cross (1 DIV=0.01mm), a tiny circle at 0.15mm and a circle at 0.07mm. Should I be using a particular one of these?
Comment: Okay, I think I've got this. Instead of calibrating by drawing a circle over the scale cross, I used a different calibration setting where I just drew a straight line vertically and a different line horizontally. with my Mycologist's directions that each length between tick marks = 10 µm I am now able to come up with correct dimensions. Thanks for your time and attention! Should I post this as it's own answer, or should you update yours to match so that I can accept a solution?
Comment: Great to hear that you solved it! I'm not a Stack Exchange expert, but I think the solution you found is different from my original idea. So for me, it makes sense for it to be its own answer.

Answer 2 (2 votes):
The way I solved this was by using the _scale line_ method. First, I understand that each division on the stage micrometer is equal to 10 microns. Then, using my photograph of my stage micrometer and the Motic software on my computer, I draw a horizontal line over oh, say, 3 divisions and tell it that this distance (measured in pixels) is equal to 30 microns. Now, using the same method, this is also possible: I found another Windows program called [Meazure](http://www.cthing.com/Meazure.asp) which you can use in the same fashion without any special microsoft software. For this I use XnView to open the photograph of my stage micrometer, set the zoom to 100% and then draw a straight line over oh, say, 3 divisions. Then I see how many pixels long that is. Then I tell Meazure that this many pixels is equal to 30 microns. Now I can measure, in XnView, any image taken by the microscope's camera - I just have to remember to assure the view is exactly the same. The Meazure program is _very_ nice and easy to use, and I actually prefer it, although the Motic software provides labels and such that are useful.


================================================================================

