Discussion Chain ID: 20325
================================================================================

User asked: Can you measure plasma glucose with a regular glucose meter?

I'm wondering if plasma glucose can be measured with a regular glucose meter and strips like [this](http://www.adwdiabetes.com/product/free-truebalance-meter-kit_1606.htm?source=GoogleBase&zmam=69792428&zmas=1&zmac=2&zmap=2129200312KIT&gclid=Cj0KEQjwx4yfBRCt2rrAs-P5vtkBEiQAOdFXbZicho2Iysf_AAL2oypo69bLltFs2rS1ETURpNfODNUaAs-e8P8HAQ). I know these meters are normally used to measure whole blood glucose, but can they measure plasma glucose levels as well? Thanks.

Discussion:
Comment: What is blood: Mostly red blood cells, white blood cells and serum. So if you want to divide cells and serum, you should be able to measure the serum glucose levels with equipment like this as well.
Comment: ABSOLUTELY NOT - using those meters for measuring glucose in fluids besides blood is inaccurate and dangerous - you should never do that clinically or experimentally

Answer 1 (2 votes):
Excellent question... **"Can you measure plasma glucose with a blood glucose meter?"** **Absolutely Not**! In my experience, if you try to measure plasma glucose with a whole blood glucose meter the results are highly variable for a number of reasons (mentioned below). Using blood glucose meters to measure plasma glucose is **dangerous**. The meters are calibrated for whole blood and not plasma, which is chemically different and will yield different results. Depending on the clinical situation (or experimental situation), the results can be significantly different and non-linear across a huge spectrum of concentrations. [Here is a nice link from the University of Virginia demonstrating the basics of how one type of these meters works.](http://faculty.virginia.edu/analyticalchemistry/Electrochem%20Gluc/Electrochem.html) There are a number of types of glucose meters commercially available to the public as well as healthcare facilities. Most of these meters use varying solid-state chemistry techniques to drive the production of an electrical current that can be measured by the glucose meter. In other words, when the blood is taken up by the strip in the meter the glucose takes part in a redox chemical reaction that creates a small "electrical current" that is detected by the meter. _The amount of glucose in the sample is proportional to the electrical current produced_. However, there are a number of assumptions that the meters make (which can differ between meters). One of the most clinically significant assumptions is that the meter assumes that the hematocrit (i.e. the percentage of blood volume that is erythrocytes) is **normal** or approximately 45%. If you use these types of meters on plasma (which has no erythrocytes in it), the chemistry is inaccurate because the meter "thinks" all the glucose that it is detecting is actually dispersed within a pool of erythrocytes in a hematocrit of 45%. [There are a number of other issues and important points discussed in this review](http://www.ncbi.nlm.nih.gov/pmc/articles/PMC2769957/). When the clinical trials (e.g. [NICE-SUGAR](http://www.nejm.org/doi/full/10.1056/NEJMoa0810625)) began showing that intensive glucose control was associated with improved mortality, clinical practices changed from keeping blood glucose concentrations in critically ill patients between 150 and 250 mg/dl to a much lower concentrations (i.e. 80-120 mg/dl). As these new guidelines began rolling out, there were a number of subsequent studies that showed that intensive glucose control was actually associated with more frequent hypoglycemia episodes. There were likely a number of issues to explain this, but one of those was that if your meters were not correcting for differences in hematocrit the blood glucose meters would read spuriously higher than what the true measurement was (thus, you could be treating someone with insulin to lower their glucose that was 160 mg/dl - when in true the value was much, much lower).

Discussion on this answer:
Comment: I assume you mean that *using* the measurements is dangerous. There is nothing dangerous about doing the measurement, right?
Comment: Dangerous, if the patient is monitoring their blood glucose concentrations in order to maintain a safe bodily state. I.e. if the pateint has Type 2 diabetes. The incorrect reading could lead to incorrect administration.
Comment: @daniel yes - using the measurement is potentially very dangerous – that's what happened in some of the early intensive glucose control studies. Glucose was being monitored with glucose meters at the bedside and those meters were giving inaccurate results because the meters assume certain physiological parameters like hematocrit that were not normal. The patients were then getting treated with insulin who had normal blood glucose but high "measurements" – and then those patients were becoming hypoglycemic and having adverse events because of insulin they didn't actually need.
Comment: While I agree with the basic conclusion - that glucometers do not reflect *plasma* glucose, I think you're assertations can be misleading. Glucometers can be invaluable in *predicting* plasma glucose. Also, this is a funny phrase "associated with improved mortality". Better death? Should that be "improved mortality rate" or something similar? This language is a bit loose.


================================================================================

