Discussion Chain ID: 17742
================================================================================

User asked: Why digest proteins into peptides for Liquid Chromatography - Mass Spectrometry?

Digesting (trypsin or whatever other proteolytic enzyme) proteins generates multiple peptides so the degree of complexity of the sample, at the peptide level, increases a lot. In addition there is the problem of infering the original protein from its constituent peptides. Why is this digestion step needed when you have to go back to protein level ? Is it just technological limitation?

Discussion:
Comment: As you rightly put, it is a technological limitation. Mass spectrometry based studies require proteins to be fragmented into smaller peptides for the ease of analysis.
Comment: It seems though, that intact proteins (even protein complexes) can also be analysed by MS. I just knew that
Comment: See pg no. 33 and 35 of [this pdf](http://wolfson.huji.ac.il/purification/PDF/ReversePhase/VYDAChandbookRPC.pdf). It says that Reversed Phase HPLC can be used both for studying protein digests and intact proteins.
Comment: Yes, LC works on separating large macromolecules, but my guess is that the limitation is on analyzing them by MS. Why can't you just do high throughput proteomics by LC-MS/MS as you would with digested protein mixture but with intact proteins?

Answer 1 (3 votes):
This is going to be a very long answer but to give a short response. You have to consider that MS for peptide detection works on the bases/principle of mass to charge (m/z) to detect an AA molecule, which is then normalised and analysed etc etc (<http://en.wikipedia.org/wiki/Mass_spectrometry>). Once you have the amino acids, then you just look at the order in which they get through and then you have your peptide sequence. Please read about the B, A and Y-type ions although this page is slightly technical (<http://www.matrixscience.com/help/fragmentation_help.html>). If the proteins was not digested then it would be far too big to be analysed since it would be read as one massive blob of m/z, which could be anything! So once you have your spectra from a peptide, you need to compare it to a model spectra, and based on that you predict what the sequence is as fragmentations peaks can be anything. Look at SEQUEST (<http://en.wikipedia.org/wiki/List_of_mass_spectrometry_software>). Now there are tons of correction algorithms that are applied, which you can look up. Now RP-LC is carried out before MS to prevent all the sample rushing into the MS machine all at once as the machine can not cope if too much sample is inserted hence the flow rate is controlled. What I think you are thinking in terms of the set machine limit, is the (digested) peptide fragment that enters the machine, before undergoing electrospray. Now the experts that run the MS machines do set a peptide mass upper and lower limit, for what size/mass of fragment to accept and what size/mass to ignore. Look at this theoretical digest example (<http://prospector.ucsf.edu/prospector/cgi-bin/msform.cgi?form=msdigest>). This happens frequently with trypsin digest as it recognises specific AA before cleavage (<http://en.wikipedia.org/wiki/Trypsin>). Now a days, MS has advanced so much that we have machines such as Orbitrap (<http://en.wikipedia.org/wiki/Orbitrap>), which can cope with peptides, digested with elastase, which cleaves non specifically, which means you are less likely to get peptides that are far too big for analysis, hence increasing the protein coverage. Hope this helps!

Answer 2 (0 votes):
I found the answer to my own question It was formulated in a Nature publication by Mathias Mann (Title: The ABC's (and XYZ's) of peptide sequencing), a pope in Proteomics: > Why are peptides, and not proteins, sequenced? > > After protein purification, the first step is to convert proteins to a set of peptides using a sequence-specific protease. Even though mass spectrometers can measure the mass of intact proteins, there are a number of reasons why peptides, and not proteins, are analysed in proteomics. Proteins can be difficult to handle and might not all be soluble under the same conditions (it should be noted here that many detergents interfere with MS, because they ionize well and are in a huge excess relative to the proteins). In addition, the sensitivity of the mass spectrometer for proteins is much lower than for peptides, and the protein might be processed and modified such that the combinatorial effect makes determining the masses of the numerous resulting isoforms impossible. Furthermore, it is not easy to predict from the sequence what the mass of a mature, correctly modified protein will be or, conversely, which protein might have given rise to a measured protein mass. Most importantly, if the purpose is to identify the protein, sequence information is needed and the mass spectrometer is most efficient at obtaining sequence information from peptides that are up to approx20 residues long, rather than from whole proteins. Nevertheless, with very specialized equipment, it is becoming possible to derive partial sequence information from intact proteins, which can then be used for identification purposes or the analysis of protein modifications in an approach called 'top-down' protein sequencing


================================================================================

