Discussion Chain ID: 70348
================================================================================

User asked: Using nanodrop for analysing biological samples other than nucleotides

I am a 3rd timer postdoctoral fellow with some experience in molecular biology and biochemistry, but major skills in Zoology and Natural History. I am studying some natural extracts, and isolating compounds. Some biological samples come in limited amounts (e.g. invertebrate haemolymph), hampering analytical methods that rely on larger amounts. For instance, establishing UV-Vis spectra of extracts is a usual non-destructive method for approaching unknown samples and estimating general parameters, done with a spectrophotometer. The microvolume-scaled spectrophotometer system popularly known as 'Nanodrop' has been around labs since almost two decades now. Its use has been usually limited to fast purity & concentration estimations for DNA & RNA in molecular biology labs. I am considering using the nanodrop to estimate parameter of other biological and chemical samples. The manual says most commonly used solvents are compatible with the system. I have however yet found no-one else who has tried using Nanodrop for different applications. Please, anyone here who has experimented using a Nanodrop system with other biological samples and extracts could comment on the experience?

Answer 1 (4 votes):
The Nanodrop is a generic UV-visible spectrophotometer. According to the manufacturer, [the latest model can measure absorbance **from 190 to 850 nm**](https://www.thermofisher.com/order/catalog/product/ND-ONE-W). Its dynamic range is also very good: from about 0.1 to about 60 in absorbance. Therefore, as long as you don't use an incompatible solvent, **you can measure anything that absorbs in this wavelength range**. I use it very often for purified proteins. It also works well for turbid suspensions (like bacterial cultures; reading optical density at 600 nm gives an idea of turbidity). It is easy to use, and of course the low volume requirement is a big advantage. One drawback is that it won't let you fine tune certain parameters like a "real" spectrophotometer would allow (bandwidth, gain, etc.). But you can definitely get decent spectra for characterization of mixtures and concentration estimation of pure samples.

Discussion on this answer:
Comment: 'you can measure anything that absorbs in this wavelength range' -- Yes, I have been trying out for the first time recently. But have you officially done it? My peers seemed revolted as they state a nanodrop should be used **only** for DNA/ RNA/ protein standards. They sounded so dogmatic.
Comment: Yes, I have measured absorption of fluorescent dyes that is nowhere near absorption wavelength of proteins or nucleic acids. Using a Nanodrop only for proteins and nucleic acids seems like a lab policy (in my opinion, too strict and not justified), but it is definitely not a technical limitation of the instrument.

Answer 2 (1 votes):
I used the Nanodrop for the measurement of the absorbance at 600 nm of bacterial cultures, as well as nucleic acid preparations. However, I switched to using another spectrophotometer (Spetrophotometer, Ultrospec 2100 pro, UV/Visible Spectrophotometer, Amersham Biosciences) for my bacterial cultures because I trust it more due to using a cuvette containing 1 ml of bacteria instead of just 1 microlitre of culture.

Discussion on this answer:
Comment: In fact some nanodrop models do have a special pedestal for measuring OD600 of microbial growth. I ought to have mentioned that in my brief intro for the question. My main issue is, as soon as I mentioned to peers I'd be using nanodrop with other biological samples they got revolted, stating 'a nanodrop should be used **only** for DNA/ RNA/ protein standards'. I was wondering why they sounded so dogmatic about it, and how readers would receive if I employed it otherwise.
Comment: Maybe some people measure their sample and retrieve afterwards and you are contaminating the system. They might be concerned that their sample gets contaminated?
Comment: Perhaps. Nanodrop is naturally unstable, and people may blame anything for a weird result. But if samples are soluble in compatible solvents, I see no logical point, and they really declared they believe it is designed solely for nucleotide analysis (which is not what it says in the manual).

Answer 3 (0 votes):
After completing my tests, I have decided to come back to answer my own question. A Nanodrop machine has served my purpose quite efficiently, as far as I can judge. I used have used it to evaluate the degree of purity of a natural alkaloid extract from insects, using synthetic alkaloid analogues as controls. See our published paper discussing the method [here](https://doi.org/10.1016/j.dib.2018.03.101), along with [raw data](http://dx.doi.org/10.17632/cpnscw2gkc.1). We have noticed no alterations nor cross contaminations to colleagues' DNA/RNA samples running in parallel, highlighting on the fact that I properly cleaned the equipment between use. Therefore, yes: I recommend using the Nanodrop as a cheap & quick method to scan biological samples, provided chemicals and solvents used in cleaning the pedestal are compatible with the manual. Hope this discussion help others dare!

Discussion on this answer:
Comment: @WYSIWYG I reckon we did what you describe, no? On using synthetic compounds as standards and doing a wavelength scan?
Comment: Sorry. I did not see it. Also, if you use the "UV-Vis" mode (I have nanodrop 1000), you can choose two wavelengths (in other assays, the wavelengths are fixed). I assume you used that.


================================================================================

