Discussion Chain ID: 101238
================================================================================

User asked: Unknown Issue with qPCR Standard

I am currently doing the ground work for a SELEX experiment and I am attempting to verify my standards for the starting library. I ran a reaction with samples containing 50, 5, 0.5, and 0.05ng of my starting library and the samples for the 5, 0.5, and 0.05 appear to amplify as expected, but whenever I run a sample containing 50ng of my starter library, the curves start around -2000 RFU and peak at 0 RFU. Has anyone seen this behavior before? EDIT: Added annotated graph **Method:** SYBR Green **System:** Bio-rad CFX384 **Software:** Bro-rad CFX Manager [](https://i.sstatic.net/7wGh6.png)

Discussion:
Comment: Can you please label your curves to make it clear which is which? Also, how are you computing RFU? Getting negative fluorescence usually indicates a process failure of some sort.
Comment: @jakebeal My apologies, I should have annotated it from the start. I've updated the post to include the annotated graph and more information. I am reading RFU with SYBR Green on a Bio-Rad CFX384. I've performed this assay several times with fresh dilutions of library DNA and the results seem to be the same each time.
Comment: I'm not too familiar with SELEX, but just looking at the curves, it seems like a baseline normalization issue. The Bio-Rad software that I use automatically normalizes amplification curves to a baseline average from the first few cycles. Since your 50ng wells were hot from the start, it looks like it took the saturated endpoint fluorescence as baseline, and normalized the rest of the curve to that. You can turn normalization off in the software, but probably still won't be able to get a good Cq value for the 50ng curves.
Comment: @MikeyC This does seem like the most likely explanation. The only thing that might make a difference in this regard is that I did run a no-template control which was stable at 0 throughout the run. It is not pictured here because it covered up a portion of the 50ng sample curve. I am new to this system so I am not totally sure how it goes about calculating a baseline. Thank you for your input.
Comment: @jakebeal The no-template control doesn't factor in. The software determines a different baseline for each well, which it subtracts from that well's fluorescence readings throughout the run so they can all lineup at 0 for the first few cycles. It's just not getting a usable baseline from those wells, so it's subtracting the saturated fluorescence instead. You can disable baseline subtraction under the Settings menu in your CFX manager software. Only problem is, you can't set a threshold for Cq determination without baseline subtraction.
Comment: @MikeyC If you're right, then yes, definitely a process failure --- and yet another example of why I never let instrument software handle calibration processes. Can you please turn your comments into an answer so that they can be voted up?


================================================================================

