Discussion Chain ID: 81021
================================================================================

User asked: Meaning of Phred score - probability of base calling error

I learned that Phred quality scores are logarithmically linked to error probabilities. For example, if Phred assigns a quality score of 10 to a base, the chances that this base is called incorrectly are 1 in 10. How is it possible to have 10 possiblities when you can only have 4 bases (ACGT) ? When assigning a base, there are only three ways to be wrong to me.

Answer 1 (3 votes):
Phred score is a measure of the probability that a base call is wrong/right. Your understanding of this probability as "_if Phred assigns a quality score of 10 to a base, the chances that this base is called incorrectly are 1 in 10_ ", is not entirely accurate. Quality scores calculation and base calling are specific to the machine. The way an Illumina HiSeq calls a base is different from how PacBio does it. For example, Illumina uses high precision cameras to record the colors generated during sequencing (each color represents a base). Moreover, it does not sequence a single molecule of DNA but a cluster. Now, the machine's base calling software will calculate the reliability of the light signal being interpreted as a base (let's say A). Several parameters are considered in this calculation. > For each base call, a number of quality predictor values are computed. Quality predictor values are observable properties of clusters from which base calls are extracted. These include properties, such as intensity profiles and signal-to-noise ratios, measure various aspects of base call reliability. They have been empirically determined to correlate with the quality of the base call. > > * * * > > [Illumina Technical notes: Understanding Illumina Quality Scores](https://www.illumina.com/content/dam/illumina-marketing/documents/products/technotes/technote_understanding_quality_scores.pdf) Likewise, different machines have different techniques for reading a base signal and computational models to calculate the accuracy of the signal. > How is it possible to have 10 possiblities when you can only have 4 bases (ACGT) ? Why not? You are just calculating what is the probability that a base is not adenine. The minimum probability should be more than 1/4, at or below which the base call wouldn't probably even assign the base as adenine. Imagine it like this: if adenine is represented by blue light and you see that in the spot there is not a pure blue signal but some interference of other colors (noise) which represent 10% of the signal. There can be other parameters too that make the calculated blue signal to deviate from the "ideal signal". The percentage of non-overlap with the perfect signal would be the error.

Answer 2 (1 votes):
> if Phred assigns a quality score of 10 to a base, the chances that this base is called incorrectly are 1 in 10 This does not mean that there are 10 different possibilities for a base. This means that you have 10% chance that the base is different from the one that was called. If the base called is an A with a score of 10, it has 90% chance of really being an A, and 10% chance of being either C, G or T. If we assume that the probabilities of each of the 3 other bases are equal, this would be: | A | C | G | T | -------------+-------+-------+-------+-------+ P(real_base) | 0.9 | 0.033 | 0.033 | 0.033 | However, I'm not sure that the probabilities of C, G and T are really equal: it could well be that, due to the specifics of the sequencing technology, some (read base, real base) pairs are more common than others.


================================================================================

