Discussion Chain ID: 916
================================================================================

User asked: How can you incorporate the hydrophone calibration in Avisoft?

I am stuck in being able to calibrate the avisoft. I have a soundtrap Hf300 and I'm not sure how to insert the data into the avisoft calibration. Any advice? Thanks All the best [](https://i.sstatic.net/TzUgp.png) [](https://i.sstatic.net/eV4vK.png)

Answer 1 (2 votes):
First, input the end-to-end calibration value for whichever mode you recorded your files in. Assuming you did your recordings in 'high gain' mode, then you'd put **-176.4** as our "**reference SPL** " in Avisoft. Then, check out page 19 of the [soundtrap user guide](http://www.oceaninstruments.co.nz/wp-content/uploads/2015/04/ST-User-Guide.pdf). While it does not specify how Avisoft reads in data (and I am unfamiliar with Avisoft), it tells you how to import it into PAMGuard, Matlab, PAMGuide and Audacity, which could be helpful. I would guess that you'd put **1 Vp** as the drop-down selection for "**full-scale range** "... any Avisoft-experienced folks out there?

Discussion on this answer:
Comment: Super thank you!! thats help a lot!


================================================================================

