![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-33052fb1-1daa-4f03-9ae3-388631ce9e48.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-ba0b4019-8d10-4f94-a203-ab672d211a4f.jpg)

MIGRATE 2018: 211489

## PROPOSED DESIGN FOR SIMULTANEOUS MEASUREMENT OF WALL AND NEAR-WALL TEMPERATURES IN GAS MICROFLOWS

Varun Yeachana 1 , Vikash Kumar , Lucien Baldas , Marcos Rojas-Cardenas , 2 1 1 Christine Barrot , Ryan Enright , Stéphane Colin * 1 2 1

1.Institut Clément Ader (ICA), Université de Toulouse, CNRS, INSA, ISAE-SUPAERO, Mines-Albi,

UPS, Toulouse France

2.Thermal Management Research Group, Efficient Energy Transfer (ηET) Dept., Bell Labs Ireland, stephane.colin@insa-toulouse.fr

## KEY WORDS

Molecular  tagging  thermometry,  gas  microflows,  phosphorescence,  ITO-coated  sapphire  sensor, temperature jump

## ABSTRACT

Gas behavior in systems at microscale has been receiving significant attention from researchers in the last  two  decades  [1-4].  Today,  there  is  an  enhanced  emphasis  on  developing  new  experimental techniques  to  capture  the  local  temperature  profiles  in  gases  at  rarefied  conditions.  The  main underlying  reason  behind  this  focus  is  the  interesting  physics  exhibited  by  gases  at  these  rarefied conditions, especially in the transition regime. There is the onset of local thermodynamic disequilibrium,  which  manifests  as  velocity  slip  and  temperature  jump  [1-4]  at  the  wall.  However, there is limited experimental evidence on understanding these aforementioned phenomena. With the advances  in  experimental  facilities,  it  is  today  possible,  at  least  in  principle,  to  map  the  local temperature profiles in gases at rarefied conditions. Molecular tagging approach is one such technique which has shown the potential to map the temperature profile in low pressure conditions [5].

In molecular tagging approach, a very small percentage of tracer molecules are introduced into the gas of interest, referred as carrier gas. In gas flow studies, the typical tracers employed are acetone and biacetyl. These tracer molecules, assumed to be in equilibrium with the carrier gas, are excited with a source of energy at a specific wavelength, typically a laser. The excited molecules are unstable and tend  to  de-excite  in  a  radiative  and  non-radiative  manner,  which  is  manifested  as  fluorescence  and phosphorescence.  Following  the  deformation  with  time  of  a  tagged  line  permits  to  obtain  the  flow velocity.  In  addition,  the  dependence  of  the  phosphorescence  and  fluorescence  intensity  to  the  gas temperature could also allow to use this technique for local temperature measurements.

The objective of this study is to develop an experimental setup capable of simultaneously mapping the wall  and  fluid  near-wall  temperatures  with  the  final  goal  to  measure  temperature  jump  at  the  wall when rarefied  conditions  are  reached.  The  originality  of  this  setup  shown  in  Figure  1  is  to  couple surface  temperature  measurements  using  an  infrared  camera  with  Molecular  Tagging  Thermometry (MTT) for gas temperature measurements. The bottom wall of the channel will be made of Sapphire substrate  of  650 µ m  thickness  coated  with  a  thin  film  of  Indium  Tin  Oxide  (ITO).  The  average roughness  of  this  ITO  layer  is  about  3  nm.  The  top  wall  of  the  channel  will  be  made  of  SU8  and bonded with the bottom wall with a layer of PDMS. The channel will be filled in with acetone vapor,

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-50a1b301-4904-404f-8a6f-a49d50960403.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-cd0b5a1a-7471-4228-9a0e-e7a5d82b31ac.jpg)

or with a carrier gas seeded with acetone molecules. The channel height h and the pressure of acetone will be chosen in order to achieve a Knudsen number of the order of 0.1 corresponding to the slip or early transition regimes. The ITO layer will be uniformly heated in order to generated within the gas a temperature  gradient  perpendicular  to  the  surface,  that  will  result  in  rarefied  conditions  in  a temperature jump i.e. a difference of temperature between the wall and the gas molecules.

Figure 1: Schematic of the proposed experimental setup

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-05f6399a-ee7f-4c28-86f0-7efa77f7d191.jpg)

Surface temperature measurement of the sapphire wall will be measured with the ITO layer emitting an  IR  signal  recorded  by  a  thermal  camera.  This  is  a  non-intrusive  technique  to  measure  the  wall temperature,  which  has  been  developed  at  Nokia  Bell  Labs  in  Dublin  [6].  For  measuring  the temperature  of  the  molecules  close  to  the  wall,  evanescent  ultraviolet  (UV)  waves  will  be  used  to illuminate a thin layer of fluid near the wall and to excite the acetone tracer molecules present in this layer, as shown in the zoomed view of Figure 1. A pulsed laser sheet of wavelength 310 nm will be subjected to total internal reflection (TIR) at the top end of the bottom wall (in our case a sapphire substrate  with  refractive  index  1.75).  A  sapphire  prism  helps  in  bending  of  the  light  beam  at  the appropriate angle. To satisfy the condition of continuity of electric and magnetic fields at the boundary of reflection, waves of light are transmitted into the medium of lower refractive index (here acetone vapor with a refractive index 1.35) [9]. These waves are referred to as evanescent waves, and they exhibit an exponential decay in intensity with distance. Their presence is felt within an order 100 nm in the material of lower refractive index [7-9]. Therefore, these evanescent waves can be employed to probe the regions closer to the wall.

Evanescent  wave-based  studies  have  been  previously employed  by  researchers, especially in multilayer  Nano-PIV  studies  to  study  the  near-wall  behavior  of  liquid  flows  [7-9].  Also,  some researchers have adopted this technique to map the temperature in liquids in regions which are few hundred nanometers from the wall [9]. In general, these techniques employ fluorescent particles. The challenge of this study will be to obtain near-wall gas temperature measurement  based  on phosphorescence of molecules instead of particles. The phosphorescence is captured by an Intensified relay  optics  (IRO)  attached  to  the  CCD  camera.  The  IRO  not  only  intensifies  the  signal  of

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-e0158e95-0081-4fa4-b8b7-15432d35ab4f.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1908.00621v1-figure-8265a845-91d8-4618-91f1-cdbe98366089.jpg)

photoluminescence, but also provides with an adjustment for the gate i.e. the duration for which the signal  is  captured  (the  CCD  camera  is  always  open,  but  the  signal  is  captured  based  on  the  gate value).The intensity dependence of acetone phosphorescence with temperature has been experimentally demonstrated in our previous work on MTT [5]. Capturing the phosphorescence of the acetone molecules close to the wall will give us an information of the near-wall temperature of acetone vapor.

In  our  previous  study  [5],  phosphorescence  emission  of  acetone  vapor  has  been  studied  at  the excitation wavelength of 310 nm, at various conditions of pressure ( p =  15,000 Pa, 10,000 Pa, 7500 Pa,  5000  Pa,  2500  Pa  and  1000  Pa)  and  conditions  of  temperatures  (20  °C,  34 o C,  and  50 o C). Therefore, these conditions of temperature and pressure will be targeted in the future experiments on the proposed experimental setup. The initial aim of the study will be to carry out experiments without flow  in  the  system.  Subsequently,  experiments  with  flow  in  the  system  will  be  conducted  to  study temperature jump. Based on the difference in temperature values obtained by analyzing both the ITO generated IR signal and the visible phosphorescent signal of acetones molecules within the Knudsen later at the wall, it should be possible to get information on the temperature jump at the wall.

## Acknowledgements

This project has received funding  from the European Union's Framework Programme for Research and Innovation Horizon 2020 (2014-2020) under the Marie Skłodowska-Curie Grant Agreement No. 643095, and from the Fédération de Recherche FERMAT (FR 3089).

## References and Citations

- [1]  Kandlikar, S., Garimella, S., Li, D., Colin, S., &amp; King, M. R. (2005). Heat transfer and fluid flow in minichannels and microchannels . Elsevier.
- [2] Colin,  S.  (2005).  Rarefaction  and  compressibility  effects  on  steady  and  transient  gas  flows  in microchannels. Microfluidics and Nanofluidics , 1 (3), 268-279.
- [3] Agrawal, A. (2011). A comprehensive review on gas flow in microchannels. International Journal of Micro-Nano Scale Transport , 2 (1), 1-40.
- [4] Colin,  S.  (2012).  Gas  microflows  in  the  slip  flow  regime:  a  critical  review  on  convective  heat transfer. Journal of Heat Transfer , 134 (2), 020908.
- [5] Yeachana, V., Fratantonio, D., Rojas-Cárdenas, M., Barrot, C., and Colin, S. (2017). Preliminary investigations on temperature dependency of acetone vapor luminescence for molecular tagging thermometry. Proceedings of 2nd International MIGRATE Workshop, Sofia, Bulgaria p.MIGRATE-153514.
- [6] Kumar, V., Jeffers, N.,  and Enright. R. (2017) Characterization of thin Indium Tin Oxide (ITO) coated sapphire sensor for applications in simultaneous optical imaging and non-intrusive surface temperature  measurement  of  gas  microflows.  Proceedings  of 2nd  International  MIGRATE Workshop, Sofia, Bulgaria p.MIGRATE-152927.
- [7] Li, H. F., &amp; Yoda, M. (2008). Multilayer nano-particle image velocimetry (MnPIV) in microscale Poiseuille flows. Measurement Science and Technology , 19 (7), 075402.
- [8]  Bouzigues,  C.  I.,  Tabeling,  P.,  &amp;  Bocquet,  L.  (2008).  Nanofluidics  in  the  debye  layer  at hydrophilic and hydrophobic surfaces. Physical Review Letters , 101 (11), 114503.
- [9] Yoda, M., &amp; Kim, M. (2013). Studying interfacial transport with evanescent wave-based particle velocimetry and thermometry. Heat Transfer Engineering , 34 (2-3), 101-112.
Varun Yeachana Vikash Kumar Lucien Baldas Marcos Rojas-Cárdenas Christine Barrot Ryan Enright Stéphane Colin
Proposed Design for Simultaneous Measurement of Wall and Near-wall Temperatures in Gas Microflows
Gas behavior in systems at microscale has been receiving significant
attention from researchers in the last two decades [1-4]. Today, there is an
enhanced emphasis on developing new experimental techniques to capture the
local temperature profiles in gases at rarefied conditions. The main underlying
reason behind this focus is the interesting physics exhibited by gases at these
rarefied conditions, especially in the transition regime. There is the onset of
local thermodynamic disequilibrium, which manifests as velocity slip and
temperature jump [1-4] at the wall. However, there is limited experimental
evidence on understanding these aforementioned phenomena. With the advances in
experimental facilities, it is today possible, at least in principle, to map
the local temperature profiles in gases at rarefied conditions. Molecular
tagging approach is one such technique which has shown the potential to map the
temperature profile in low pressure conditions [5]. In molecular tagging
approach, a very small percentage of tracer molecules are introduced into the
gas of interest, referred as carrier gas. In gas flow studies, the typical
tracers employed are acetone and biacetyl. These tracer molecules, assumed to
be in equilibrium with the carrier gas, are excited with a source of energy at
a specific wavelength, typically a laser. The excited molecules are unstable
and tend to de-excite in a radiative and non-radiative manner, which is
manifested as fluorescence and phosphorescence. Following the deformation with
time of a tagged line permits to obtain the flow velocity. In addition, the
dependence of the phosphorescence and fluorescence intensity to the gas
temperature could also allow to use this technique for local temperature
measurements. The objective of this study is to develop an experimental setup
capable of simultaneously mapping the wall and fluid near-wall temperatures
with the final goal to measure temperature jump at the wall when rarefied
conditions are reached. The originality of this setup shown in Figure 1 is to
couple surface temperature measurements using an infrared camera with Molecular
Tagging Thermometry (MTT) for gas temperature measurements. The bottom wall of
the channel will be made of Sapphire substrate of 650 $\mu$m thickness coated
with a thin film of Indium Tin Oxide (ITO). The average roughness of this ITO
layer is about 3 nm. The top wall of the channel will be made of SU8 and bonded
with the bottom wall with a layer of PDMS. The channel will be filled in with
acetone vapor,