## Study of the characteristics of GEM detectors for the future FAIR experiment CBM

S. Biswas a, ∗ , A. Abuhoza a,b , U. Frankenfeld , J. Hehner , C. J. Schmidt , H.R. Schmidt , M. Tr¨ ager , S. Colafranceschi , a a a c a d A. Marinov d , A. Sharma d

a

GSI Helmholtzzentrum f¨ ur Schwerionenforschung GmbH, Dar mstadt, Germany-64291 b King Abdulaziz City for Science and Technology (KACST),Riyadh, Saudi Arabia c Eberhard-Karls-Universit¨ at, T¨bingen, Germany u d Physics Department, CERN - Geneva, Switzerland

## Abstract

Characteristics of triple GEM detector has been studied systematically. The variation of the e ff ective gain and energy resolution of GEM with variation of the applied voltage has been measured with Fe 55 X-ray source for di ff erent gas mixtures and with di ff erent gas flow rates. Long-term test of the GEM has also been performed.

Keywords: FAIR, CBM, Gas Electron Multiplier, Gas gain, Resolution PACS: 29.40.Cs

## 1. Introduction

## 2. Detector descriptions and tests

The Compressed Baryonic Matter (CBM) experiment at the future Facility for Antiproton and Ion Research (FAIR) in Darmstadt, Germany will use proton and heavy ion beams to study matter at extreme conditions [1, 2]. The CBM experiment at FAIR is designed to explore the QCD phase diagram in the region of high baryon densities [3]. This will only be possible with the application of advanced instrumentation, including highly segmented and fast gaseous detectors.

Gas Electron multipliers (GEM) will be used in CBM Muon Chamber (MUCH) located downstream of the Silicon Tracking System (STS) of the CBM experiment along with other sophisticated detectors [4]. The main goal of MUCH is to detect the dimuon signals arising from the decay of the low mass vector mesons and those from the decay of charmonia produced in the heavy ion collisions at FAIR.

In GSI detector laboratory an R&amp;D e ff ort is launched to study the characteristics of GEM detectors for the CBM experiment. The primary goals of this R&amp;D program are: (a) to verify the stability and integrity of the GEM detectors over a period of time, during which a charge density of the order of several Coulomb cm 2 / is accumulated in the detector; (b) to establish the functioning of triple GEM as a precise tracking detector under the extreme condition of the CBM experiment; (c) to study usual parameters e.g., e ffi ciency, rate capability, long term stability, spark probability by varying conditions like temperature, gas composition or radiation dose.

In this article, we would like to present the initial results of characterisation of triple GEM detectors for CBM.

∗

Corresponding author

Email address: S.Biswas@gsi.de (S. Biswas)

A triple GEM detector, consisting of double mask foils, obtained from CERN has been studied systematically. The drift, transfer and induction gaps of the detector are 3 mm, 2 mm and 2 mm respectively. The voltage to the drift plane and individual GEM plates has been applied through a voltage divider chain. Although there is a segmented readout pad the signal in this study was obtained from all the pads summed by a add up board and a single input is fed to a charge sensitive preamplifier. After that a PXI LabVIEW based data acquisition system is used. The gain of the detector has been measured by obtaining the mean position of 5.9 keV peak of Fe 55 X-ray spectrum with Gaussian fitting. A typical spectrum recorded with a Fe 55 source is shown in Fig. 1. The variation of the gain and energy resolution of this detector with that of the applied voltage has been measured for di ff erent gas mixtures and with di ff erent gas flow rates.

Figure 1: Fe 55 X-ray spectrum obtained with triple GEM detector.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-c461c190-0614-45f6-a07e-fa872bc331b7.jpg)

## 3. Results

The detector has been operated with Argon and CO2 in 60 40, 70 30, 80 20 mixing ratios with di / / / ff erent flow rates e.g. 50 ml min, 100 ml min and 200 ml min. / / / The variation of effective gain and energy resolution with that of global voltage ( ∆ V1 +∆ V2 +∆ V3) are shown in Fig. 2 and Fig. 3 respectively for the gas mixture Ar CO2 / in 70 30 with di / ff erent flow rates. It is clear that the gain increases exponentially and the energy resolution improves with the increase of voltage. The gain is relatively higher with the higher gas flow rate. This is due to the less oxygen contents (shown in Fig. 2) during higher flow rate. Same behavior is also observed for the other two gas mixtures.

The detector is operated continuously for five days with Ar CO2 / in 70 30 ratio with 100 ml min flow rate at a global / / voltage 1176 V. The gain and resolution as function of time are shown in Fig. 4 and Fig. 5 respectively.

Figure 2: Gain vs. global voltage.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-32dced2c-a964-45fb-925c-80fd90563d34.jpg)

Figure 3: Resolution vs. global voltage.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-6b24ce9c-6f06-4b6e-b814-bd786fb77e06.jpg)

## 4. Conclusions and outlook

Variation of gain and resolution with voltage for di ff erent gas mixtures and with di ff erent gas flow rates has been measured in GEM detectors. It is clear that oxygen content in the gas has a very important role in GEM performance even in a few ppm level.

In an initial continuous operation of five days the gain fluctuates with time, and resolution varies about 4%. Further investigation is necessary to correlate this variation with other ambient parameters.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-d6d4e339-53a1-4864-8d8e-1b4d1df37e7d.jpg)

Figure 4: Gain as a funcion of period of operation.

Figure 5: Resolution as a funcion of period of operation.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-7edee0b8-2597-405b-b0ad-4fb5d003891c.jpg)

Figure 6: New GEM box designed at GSI.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0075v1-figure-48e31954-7d93-4f7e-8904-535a83a6e659.jpg)

Fabrication of new 10 cm × 10 cm GEM detector with newly designed box (shown in Fig. 6) and their characterisation is also carried out. Further studies are still in progress and will be communicated at a later stage.

## 5. Acknowledgement

We are thankful to Dr. Ingo Fr¨hlich of University of Franko furt, Prof. Dr. Peter Fischer of Institut f¨r Technische In foru matik der Universit¨ at Heidelberg, Prof. Dr. Peter Senger,CBM Spokeperson and Dr. Subhasis Chattopadhyay of VECC, India for their support in course of this work.

## References

- [1] http: // www.gsi.de forschung fair / / experiments CBM / /
- [2] http: // www.fair-center.org /
- [3] CBM Progress report 2008.
- [4] F. Sauli, Nucl. Instr. and Meth. A 386 (1997) 531.
S. Biswas A. Abuhoza U. Frankenfeld J. Hehner C. J. Schmidt H. R. Schmidt M. Traeger S. Colafranceschi A. Marinov A. Sharma
Study of the characteristics of GEM detectors for the future FAIR experiment CBM
Characteristics of triple GEM detector have been studied systematically. The
variation of the effective gain and energy resolution of GEM with variation of
the applied voltage has been measured with Fe55 X-ray source for different gas
mixtures and with different gas flow rates. Long-term test of the GEM has also
been performed.