Titel: Artificial Left Ventricle

Authors:  Saeed Ranjbar , Ph.D, Tohid Emami Meybodi , MD, , Mahmood emami Meybodi , MD

## Research institute:

- 1Modarres Hospital, Institute of Cardiovascular Research, Shahid Beheshti University of Medical Science, Tehran, Iran.
- 2Shahid sadoughi university of medical sciences, Yazd, Iran

## *Corresponding Author: Saeed Ranjbar

The full postal address of the corresponding author:

Modarres Hospital, Institute of Cardiovascular Research, Shahid Beheshti University of Medical Science, Tehran, Iran

E-mail: sranjbar@ipm.ir

Tel/FAX: +9821 22083106

## Artificial Left Ventricle

Saeed Ranjbar , Ph.D, Tohid Emami Meybodi , MD, , Mahmood emami Meybodi , MD

## Abstract

This Artificial left ventricle is based on a simple conic assumption shape for  left  ventricle  where  its  motion  is  made  by  attached  compressed elastic tubes to its walls which are regarded to electrical points at each nodal  .This  compressed  tubes  are  playing  the  role  of  myofibers  in  the myocardium of the left ventricle. These elastic tubes have helical shapes and are transacting on these helical bands dynamically. At this invention we give an algorithm of this artificial left ventricle construction that of course  the  effect  of  the  blood  flow  in  LV  is  observed  with  making beneficiary  used  of  sensors  to  obtain  this  effecting,  something  like  to lifegates  problem.  The  main  problem  is  to  evaluate  powers  that  are interacted  between  elastic  body  (left  ventricle)  and  fluid  (blood).  The main goal of this invention is to show that artificial heart is not just a pump,  but  mechanical  modeling  of  LV  wall  and  its  interaction  with blood in it (blood movement modeling) can introduce an artificial heart closed to natural heart.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-12d57b84-baa8-40c7-9eba-65012a84a417.jpg)

These bands initiate from the posterior-basal region of the heart continues through the left ventricle free wall, reaches the septum, loops around the apex, ascends, and ends to the superioranteriorsection of the left ventricle.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-193e1a0a-a079-42c5-90a6-d94d5e6be763.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-6c3433b2-939e-4028-9fc3-3f469ad838b2.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-68cb06d7-cf12-4ec8-973c-a10939f76002.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-cac7238a-5d95-401a-92e3-8fe25ce1ec95.jpg)

Polymer motion by power effecting where are applied on micro actuators that have been situated at nodal points on elastic tubes.

FIGURE 6

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-a6333f3e-84bc-46ae-a5d8-c552132e0fbf.jpg)

Figure 7

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-5fd6d9df-562d-42a3-93e3-0c0e7cd0cbe9.jpg)

Figure 8

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-73385aed-9109-4fde-8cb4-0ff45181ddbc.jpg)

FIGURE 9

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-04db7d8a-cbe8-4a2c-9b32-dfd5c114e1d4.jpg)

Having used sensors at each nodal point, blood force on the LV wall is considered.

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-7aa3ce86-a70c-43cf-b426-478e71d1a30d.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-9ba30f0c-dbb5-4158-822c-17aca9092f50.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-d30c5394-fb8c-4a09-9fc2-c963f407b591.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-b0105620-bc33-4524-ade6-61778335d982.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-8b50bd84-8f85-411e-ac43-445bd119c411.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-7956a0d1-7aeb-4410-8caa-6b44ee74cd6f.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-c61ff6fc-2342-4e6f-a3f3-93183204b3f2.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-707705c6-df1e-47cd-8ba3-4cb66eecd14c.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-eaad17cf-b9cb-46e2-9a88-605981eb0d2e.jpg)

FIGURE 12

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-60d8785b-45e1-4b19-828a-7577f069b61d.jpg)

FIGURE 13

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-72c8dc2d-401e-4fd0-aedd-3a475c0dfed3.jpg)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-c67a2400-5980-4a2e-8a48-235238de83de.jpg)

Figure 14

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-89ab225d-c10f-4e2b-9ae2-4468cad19358.jpg)

## Background of the invention

All recent artificial left ventricles are just miniature pumps. The pump Electric coils  cause a  rotor with  an  embedded magnet to spin. Fins on the rotor push Oxygen-rich blood (red arrows) from the heart through a tube in to the aorta. On October 27, 2008, French professor and leading heart  transplant  specialist  Alain  F.  Carpentier  announced  that  a  fully implantable artificial heart will be ready for clinical trial by 2011 and for alternative transplant in 2013. It was developed and will be manufactured by him, Biomedical firm Carmat, and venture capital firm Truffle. The  prototype  uses  electronic sensors and  is made  from chemically  treated  animal  tissues,  called  "biomaterials",  or  a  "pseudoskin"  of  biosynthetic,  microporous  materials.  Another  US  team  with  a prototype called 2005 MagScrew Total Artificial Heart, including Japan and South Korea researchers are racing to produce similar projects. But at  this  presentation  the  main  goal  is  to  advance  recent  attempts  on artificial left ventricles by these point of views where using compressed elastic tubes with helical shapes toward microprous materials which are applied  to  nodal  points  of  these  elastic  tubes.  At  this  invention  these elastic tubes are replaced with the fibers of the myocardium in the left ventricle  where  this  is  needed  to  find  a  good  understanding  of  these fibers not only statically but also dynamically at the same time. A fiber in the myocardium means to follow the curve of the motion of a muscle volume sample in the myocardium from the end of diastole to the end of systole. Characterization of local and global contractile activities in the myocardium is essential for a better understanding of cardiac form and function. The spatial distribution of regions that contribute the most to cardiac  function  plays  an  important  role  in  defining  the  pumping parameters  of  the  myocardium  like  ejection  fraction  and  dynamic aspects such as twisting and untwisting. In general Myocardial shortening  and  lengthening,  tangent  to  the  wall  ,  and  ventricular  wall thickening are important parameters that characterize the regional contribution within the myocardium to the global function of the heart.

We  calculated  these  parameters  using  vector  bundles  of  myocardium deformation,  which  were  captured  through  the  displacement  encoding with stimulated echoes in 50 normal volunteers. High spatial resolution of the acquired data revealed transmural changes of wall thickening and tangential shortening with high fidelity in beating hearts during a cycle. By filtering myocardium regions that showed a tangential shortening of &lt;  0.23,  we  were  able  to  identify  the  complete  or  a  portion  of  a macrostructure composed of connected regions in the form of a helical bundle  within  the  left  ventricle  mass.  In  this  study,  we  present  a represetative  case  that  shows  the  complete  morophology  of  a  helical myocardial band due to a famous hypothesis about fibers transaction in the LV as well as cases that present ascending and descending portions of the helical myocardial band.

A  better  understanding  the  structure  and  function  of  the  heart continues to challenge modern  medicine and physiology. Despite the  remarkable  advances  in  the  understanding  of  myocardial  cell function  and  its  complex  molecularstructure,  the  role  of  cardiac sarcomeres  and  myofibers  in  generating  contractile  forces  in  the normal heart baffles us at all levels. A better understanding of the spatial  distribution  of  regions  with  elevated  contractile  dynamics within  the  myocardium  mass  would  perhaps  help  to  resolve  this issue.  As  it  can  also  be  observed  in  the  morophology  of  the  LV, our hypothesis is that if myofibers of the left ventricle form a gross macroscopic structure in the form of a band, its importance in the dynamics of the heart  must be revealed in characterized regional and global function of the myocardium. Based on our hypothesis, such an anatomic structure should act like a physical pathway for the maximum transmission of systolic contractile force and should facilitate the spatial coordination of relaxation and diastolic recoil. The effective of this approach depends on the proper indentification of the cardiac regions that exhibit higher contractile activity and the nature of their linkage and inner connectivity. In this presentation, we implement an approach that enable us to map regions  of  intense  and  directional  contractile  activities  within myocardium to examine the nature of gross morophology suggests that a band in the form of elleptical characterizes the most intense contractile activities within the ventricular mass.

The  other  assessment  to  gain  a  necessary  understanding  of  the local contractility, in contrast with chemical and electrical reasons of  the  cardiac  muscle  contraction  would  be  followed  by  a  new notion of the intrinsic power of myocytes resulting in the sufficient energy  to  the  cardiac  motion  and  deformation.  If  we  follow myofiber's  transactions  from  the  end  diastole  to  the  end  systole, they would show us helical curves that intrinsic power is made to move  myofibers  on  these  bands.  The  other  values must  be calculated is the local power  of  the fluid (blood) which  is transfered to the myocardium.

It  would therefore be desirable to design a method and system to fully applying myofiber transactions in the myocardium and their interactions with the fluid through the LV for this assistant devise where it has the central role in the cardiac function.

## Summary

This invention includes a method and system that is applied to an elastic conic shape where here is a miniature of the left ventricle.

In fact our elastic material of the left ventricle is a plastic modeling of a simple mathematical modeling  of the left ventricle in compared with its physical modeling.

The  motion  and  deformation  of  the  LV  are  forced  by  studying  of  the behaviors of fibers in LV, statically and dynamically. Our evidence not only in cardiac imaging but also in the mathematical modeling show that the most of these fibers move on helical bands. Here these fibers in the myocardium  are  replaced  with  compressed  elastic  tubes  where  are deformed to helical shapes. For each part of LV wall, they set somehow in the myocardium where their nodal points pass from the LV free wall, the mid of the septum, the apex, the mid of the lateral and the base of the anterior respectively.

Electrical sensors which are made by microprous materials are used at these  nodal  points  to  push  elastic  tubes  on  their  helical  curves.  The powers  which  are  recharged  by  these  microprouses  save  from  the evaluation of the power of the local myocardium at our LV mathematical modeling. And by these sensors and having used evaluated  powers  of  the  blood  when  comes  in  to  the  LV,  we  can observe effecting of the blood to body movement of the left ventricle.

## Brief Description of the Drawings

The drawings illustrate one preferred embodiment presently contemplated for carrying out the invention.

In the drawings:

- FIG. 1 is  a  general  flowchart  of  a  regenerative  artificial  left  ventricle where states what models are involved to pose a good structure of an artificial left ventricle near real left ventricle.
- FIG. 2 is a flowchart of the mathematical modeling of the left ventricle function  not  only  statically  but  dynamically  at  the  same  time  where data  are  evaluated  by  electrocardiogram  data  in  compare  with  the other cardiac imaging tools.
- FIG. 3 is a  flowchart  of  the  fluid  (the  blood)  dynamic  as  a  nonnewtonian  fluid  in  the  left  ventricle  where  is  studied  as  fluid-wall interaction. samples for solving partial differential equations come from Echo ( as one of tools in cardiac imaging by ultrasound method).
- FIG. 4 and 5 are flowcharts of computer and polymer modeling at this invention.
- FIG. 6 shows  a  simple  mathematical  conic  assumption  of  the  left ventricle  at  the  end  of  diastole  where  is  constructible  by  an  elastic polymer.
- FIG. 7 is a mesh file of this conic shape at the Matlab software.
- FIG. 8 shows  nodal  points  in  a  compressed  elastic  twisty  tube.  Micro actuators are applied at these points to transfer enough energy that we have explicitly computed these values of energy.
- FIG. 9 is  a  scheme  of  using  sensors  at  nodal  points  to  evaluate  the effect of the blood though the mitral valve and left ventricle.
- FIG. 10 is  a  short-axis  view  of  the  left  ventricle  toward  landmarks  of experimental samples.
- FIG. 11 is the diagrams of power local values of the myocardium where have been tested on 50 normal cases by Vivid 7 Echo.
- FIG. 12 is a scheme of a muscle volume sample toward a myofiber on it in the myocardium at the end of diastole.
- FIG. 13 is  the same muscle volume sample after passing 0.2s from the end of diastole.
- FIG. 14 is a scheme of a muscle volume sample in the myocardium with a  distribution  of  its  fibers  in  it.  And  a  scheme  of  three  vectors  of deformations where is attached to this volume sample.

## Detailed Description of the prefferd of embodiment:

Referring  to  FIG. 1 ,  the  major  components  of  a  regenerative  artificial left  ventricle 1-6 incorporating  the  present  invention  are  shown.  The operation of this regenerative artificial left ventricle is locally designed

Srr mathematically to FIG. 2,  8 and 15 in  compared with its real physical modeling of the left ventricle. Behavior of the left ventricle is basically divided to statical modeling of the LV 9-14 and to dynamical studying of the  LV 16-25. Having  assumed  LV  as  a  conic  shape 8 at  the  end  of diastole  and  look  at  shapes  of  its  sections 9-11 ,  we  can  gain  a  good understanding of the local geometrical shape of the myocardium at the left  ventricle.  These  local  mathematical  shapes  are  attached  to  each muscle volume samples and the main theme is to follow these samples on the curves where they are the fibers of a deformation flat map 1619 . In  FIG. 14 , left  ventricle  can  be  realized  as  a  vector  bundle  of deformation components radial strain longitudinal strain and Circumferential strain that are evaluated by Echo techniques for each landmarks at our samples. And the curves of myofibers transactions are iso-curves of a deformation map which is defined by the following way:

)

![image](https://ac.marcodsn.me/arxiv-markdown-images/1408.0430v1-figure-338cae51-c581-4afa-b71b-d2f0e72d4ad0.jpg)

fibers  (as  an  outputs  of  an  algorithm  where  is  running  co-images  of the deformations map) of this map  have a twisty shape geometrically 12-14 ,  and  fiber  transactions  on  such  these  bands  where  are  on  isocurves  of  the  deformation  map  give  us  a  fiber  modeling  of  the myocardium at the left ventricle 23-25 . The advantage of this modeling is due  to  ultrasound  method  where  is  really  accessible  and our algorithm  as  a  software  can  also  be  applied  to  develop  imaging  in echocardiography at this invention. Since left ventricle function  is not just  an  elastic  body  movement  but  is  a  problem  in  fluid  (blood)-wall interaction. Having used ultrasound method in echocardiography, fluid dynamic in the left ventricle is the major goal in FIG. 2 ,  as  a  fluid-wall interaction  problem  we  are  dealing  to  solve  a  system  of  partial differential equations one by continuum motions 27-34 , and the others are related to Navier-Stocks  equations 35-40 . One  of  the  major investigation is to solve discretely Navier-Stocks equations by experimental electrocardiogram data 41 , two main methods have been applied on electrocardiogram data at this invention 42-43 , that after a manipulating  of  these  data,  they  can  be  considered  as  inputs  of  an algorithm where simulates blood through the left ventricle in LSDYNA software as a fluid dynamical simulation software 45 .  Referring to FIG. 2 and FIG. 3 we give a detailed description of our method to solve of equations by using ultrasound method. One of the main attempts at this invention serves to study an arbitrary piece of the myocardium while is mechanically contracted, from the end-diastole to the end-systole. This finding can  be  posed  to  gain  a  good  understanding  of  the  local contractility,  in  contrast  with  chemical  and  electrical  reasons  of  the cardiac  muscle  contraction.  This  assessment  would  be  followed  by  a new notion of the power of myocytes resulting in the sufficient energy to the cardiac motion and deformation. This power was measured through 50 normal cases for each muscle volume samples in the myocardium at peak  systolic.  Case  data  included  velocity,  displacement,  strain  and strain rate. For instance, passing 0.2s from end diastole for a case in a piece with unit volume of his/her base lateral, 2.72w power is needed to

get 4.5cm/s, 0.83cm, 25% and 1.85 1/s, in velocity , displacement, strain and strain rate respectively. This option is also considered as a remodeling index of the ventricles that as an example of this include due to vavular stenosis or regurgitation, coronary artery disease with associated  decreased  contractility  and  fibrosis  leading  to  ventricular dilatation, genetic abnormalities resulting in hypertrophy, and progressive development of local fibrosis, severe hypertension, conduction abnormalities, etc. in fact this remodeling is a response to a problem with either the muscle itself or the environment in which it has to work and is an attempt to keep on fulfilling the heart's task-circulating the  blood.  However,  since  this  is  an  abnormal  situation  with  inherent mechanical disadvantages, in the long term, this will lead to irreversible damage  to  the  muscle  which  evolves  into  ventricular  dysfunction  and heart  failure.  The  early  detection  and  follow-up  of  change  in  cardiac function and myocardial properties is thus of major importance.

The strain-stress  relationship  is  a  common  concept  used  to  investigate the mechanical properties of materials and has recently been applied to the  analysis  of  cardiac  tissue  properties.  The  concept  of  strain  (ε) corresponds to the deformation of an object as a function of an applied force stress (s). It represents the precentage of change of the unstressed dimension after the application of stress. And hold for both expansion (positive strains) and compression (negative strains). Considering a one dimensional  object.  The  possible  stress  deformations  are  lengthening and shortening. In this case the strain can be expressed as the difference between  the  original  length  ( )  and  the  length  (L)  after  deformation. Normalized  by  the  original  length  ( ).  As  shown  in  the  following equation:

$$\varepsilon = \frac { \left ( \frac { \left ( \frac { \left ( \frac { \left ( \left ( \pi _ { 0 } } { r } } } { r } } { \right ) }$$

As  shown,  the  strain  is  a  dimensionless  parametr  and,  by  convention, positive correspond to lengthening and negative strains to shortening.

In some cases the length of the object is known during the deformation process, in that case we can define the instantaneous strain as:

$$= \frac { L ( t ) - L ( t _ { 0 } ) } { L ( t _ { 0 } ) }$$

Where (L(t)) is the length at a given instant (t) and (L( )) is the original length.

The strain rate is measurement of the rate of deformation and corresponds  to  the  velocity  of  the  deformation  process.  Taking  into account equation (1) and (2) the latter definition, the instantaneous strain rate is expressed as the temporal strain derivative:

$$\dot { \varepsilon } ( t ) = \frac { d \varepsilon ( t ) } { d t } = \frac { d L ( t ) } { d t. L ( t ) } = \frac { L ^ { ^ { \prime } } ( t ) } { L ( t ) }$$

With being  the  rate  of  deformation  and  L(t)  the  instantaneous length.  The  strain  rate  units are . Compared  with  the  rate  of deformation unites m. . As a conclusion, both strain rate and strain are closely related and they can be derived one from each other as we will discuss later.

Different  approaches  have  been  proposed  to  calculate  the  strain  and strain rate parameters; some preliminary studies have been carried out to evaluate their precision and clinical usefulness. Two main methods have been  proposed:  the  Crosscorrelation  method  and  the  velocity  gradient method.  The  Cross-correlation  method  is  based  on  the  principles  of elastography where two consecutive radiofrequency signals are compared  to  extract  information  about  the  tissue  elasticity  properties. Considering two consecutive radiofrequency signals applied to a tissue under  deformation,  the resultant backscatter signals received have similar patterns, except  for  a temporal  shift  related to the actual deformation.  Cross-correlation  analysis  provides  the  temporal  shift  or delay introduced due to the object motion. From this analysis, different parameters such as the change in distance, local motion, velocities, etc., can  be  derived.  The  principal  limitation  of  this  technique  is  its  high computational  cost  and  the  need  of  very  high  temporal  resolution  to avoid noisy estimates. That limits its application to M-mode acquisition. On  the  other  hand,  Fleming  et  al  and  Uematsu  et  al  .  introduced  the concept of myocardial velocity gradient as an indicator of local contraction and relaxation. This concept is directly related to the already formulated strain rate parameter (3) under the assumption of linear and uniform strain (homogeneous, isotropic and incompressible material) as can be shown in the following expression:

$$\dot { \epsilon } ( \mathbf t ) = \frac { d \varepsilon ( t ) } { d t } = \frac { d L ( t ) } { d t. L ( t ) } = \frac { L ^ { ( } ( t ) } { L ( t ) } \\ ( 4 )$$

where and are  the  local  instantaneous  velocities  at  two myocardial points separated an L t ( ) distance. This formulation allows to compute  myocardial  strain  rate  as  the  spatial  gradient  of  myocardial velocities,  which  can  be  obtained  by  Doppler  echocardiography.  Since the  computational  load  is  not  high  it  can  be  implemented  as  a  postprocessing  step  after  acquiring  Doppler  Tissue  images  or  in  real-time from digitally stored tissue velocity information . The axial natural strain component can be calculated from the strain rate curve time-integration expressed by:

$$\dot { \varepsilon } ( t ) d t$$

When  using  myocardial  motion  and  deformation  to  assess  (dys-) function,  it  is  important  to  understand  the  relation  between  intrinsic power  and  the  resulting  motion  and  deformation.  In  fact  by  roughly speaking, how  much  mechanical  power  is  needed  to  make  these dynamical  deformations  for  each  cardiac  muscle  volume?  This  matter would  give  the  best  information  of  cardiac  function.  In  summary,  the main  factors  influencing  to  the  cardiac  function  mechanically  are:  1) Regional myocardial motion and deformation. 2) Intrinsic power, i.e. the sufficient  energy  force  developed  by  myocardium,  resulting  in  motion and deformation. Having applied ViVid7 setting, for 50 normal cases, we got velocity, displacement, strain and strain rate at peak systolic, for each  part  of  the  myocardium,  i.e.  at  the  base,  Middle  and  Apical  of Septal, Lateral, Anterior and inferior FIG. 10 .

And  using  these  data  and  applying  them  in  the  "Power  formula",  we could compute intrinsic power values at peak systolic, for each local part of  the  myocardium  where  these  values  have  been  presented  at  the diagrams  in  FIG. 11 ,  for  instance,  the  value  of  this  new  option  at  the base  septal  for  a  case  is  6.5w  that  means,  6.5w  is  needed  to  make  a necessary motion and deformation of this part at the peak systolic. And with removing the general motion of the heart, this value for that case at the  base  septal  is  less  than  6.5w  where  this  states,  tethering  helps  that piece of myocardium to handle the matter.

We pick a cardiac muscle volume sample, for instance from the basal septum and follow-up its motion and deformation during the end diastole to the end systole FIG 12 .

Passing time" t" after the end diastole, radial strain rate for the fiber AB is:

$$s t r a i n \ r a t e = V _ { B }$$

Let are displacement and velocity of the above muscle volume element at time t.

By classical mechanic we set :

$$D _ { t } + L _ { A B } ( t ) - L ( t _ { 0 } ) = \frac { 1 } { 2 } \, a _ { t } t ^ { 2 } + W _ { t } \mathbf t$$

is the time and is the length of the fiber AB respectively at the end diastole.

Having used the 1-D deformation definition we rewrite the above formula by the following way:

$$\frac { D _ { t } } { t } + \frac { L _ { A B } ( t ) } { t } - \frac { L ( t _ { 0 } ) } { t } = \frac { 1 } { 2 } \, a _ { t } t \,.$$

By radial strain rate and strain of the fiber AB at time "t" we have:

$$\frac { D _ { t } } { t } + \ s t r a i n \ r a t e ( t ). ( \ s t r a i n ( t ) + 1 ) L ( t _ { 0 } ) = \frac { 1 } { 2 } a _ { t } t + W _ { 1 }$$

Thus we can reformulate in below:

$$a _ { t } = 2 \frac { D _ { t } } { t ^ { 2 } } + 2 s t r a i n \, r a t e ( t ). \frac { ( s t r c ) } { - \frac { 2 W _ { t } } { j } }$$

Now if µ be the density and Volume(t) (8) be the volume respectively of our muscle volume sample after the contraction during the time "t". the power which is needed to result in this motion and deformation at time "t" would be FIG. 13 :

$$& P ( t ) \\ & = \mu. \text{Volume} ( t ). a _ { t }. \, D _ { t } / _ { 1 }$$

Decreased  or  increased  in  power,  for  a  muscle  volume  sample  can  be caused  a maker  of  myocardial  ischemia,  LV  dysfunction,  or  LV hypertrophy FIG. 4, 59 . Inasmuch as intrinsic power has simultaneously been  contributed  by  some  codependent  factors  such  as  motion  and deformation FIG. 2, 18,19 , and 22 , it is realized as a remodeling index to keep up normal conditions. For instance failure to achieve adequate wall thickening (at radial deformation) and lengthening/shortening (at longitudinal  deformation)  of  a  fiber  can  be  made  by  power.  Having increased in wall thickening of a piece of left ventricle, motivate us to inhibit widely dilate with a hypertrophic wall, by a reduction in power of that part. Particularly when power tends to zero for a muscle volume, it may be a factor that shows myofibers of that section, have died. So it would  be  a  criterion,  where  we  can  make  use  of  stern  cell  method  to improve cardiac? FIG. 1 and 5

On the other hand, this power shows its influence on red blood cell flow by change in volume FIG. 3, 27 , In fact when a unit volume of red blood cell is dealing with a piece of ventricles, has a simple description of its change in volume, i.e. it can calculate by:

∆V(t)= Local change in volume of RBC with unit volume from end diastole to end systole , where

## ∆V(t)= (Radial deformation(t))x(Longitudinal deformation(t)) x Displacement(t)                                                 (8)

From this point of view, one can locally compute myocardial pressure which  is  a  very  valuable  measure  for  many  normal  and  pathological conditions  of the heart, particularly right ventricular function  and interrelationship  of  the  right  and  left  heart.  This  pressure  is  directly measured by the following way:

## Local myocardial pressure (t) =

.

## (9)

If  we  follow  myofiber's  transactions  from  the  end  diastole  to  the  end systole,  they  would  show  us  curves  FIG. 2,  23 ,  that  intrinsic  power  is made to move myofibers on these bands.

Referring to FIG. 4 ,  the  present  invention  includes  algorithms  that  are applied to softwares 50 and 53 to follow a muscle volume sample in the myocardium  and  its  interaction  with  a  red  blood  cell  within  the  LV contractions. Experimental samples are evaluated by electrocardiogram data and mathematical formulations will be also applied on these data to disclose  images  of  a  fiber  modeling  of  the  left  ventricle  motions  and blood  through  it.  Referring  to  a  ventricular  disease  in  terms  of  its electrocardiogram  data,  we  can  also  do  the  same  work  to  give  a  fiber modeling for this disease 59 , that it would basically be hard to diagnosis it by echo techniques. By a technical manipulating of electrocardiogram data  therein  as  inputs  of  our  algorithms  we  can  do  cardiac  surgeries virtually 53 and 60 .  Having made the beneficiary used of fiber modeling of  the  LV  where  had  been  modeled  by  echocardiography  tool  and referring to FIG. 4 , and 51, 52 we can make a program to virtual reality in  cardiac  surgery  and  follow  new  techniques  and  repairs  for  cardiac surgical tasks.

Due to FIG. 5 , the best mathematical modeling at this invention was the main goal where it can be applied on an elastic polymer for an artificial left ventricle 64 and 65 . In fact dynamical properties of this polymer are made  by  representing  elastic  tubes  (as  myofibers  in  the  left  ventricle) where  have  been  equipped  to  micro  actuators  in  their  nodal  points  to charge enough powers 66-69 .

The  present  study  has  been  described  in  terms  of  a  mathematical modeling that all mathematical formulations are stated from ultrasound methods in compared with the other cardiac imaging tools.

Disclosure:

There is no disclosure .

## References:

- 1Saeed Ranjbar ,  Mersedeh Karvandi. System and method for modeling left ventricle of heart. US Patent Number: 8,414,490.
- 2Saeed Ranjbar , Mersedeh Karvandi  Solution Navier-stocks equations of the blood as a . non-newtonian fluid in the left ventricle. US Patent number: 8805663
Saeed Ranjbar Tohid Emami Meybodi Mahmood Emami Meybodi
Artificial Left Ventricle
This Artificial left ventricle is based on a simple conic assumption shape
for left ventricle where its motion is made by attached compressed elastic
tubes to its walls which are regarded to electrical points at each nodal .This
compressed tubes are playing the role of myofibers in the myocardium of the
left ventricle. These elastic tubes have helical shapes and are transacting on
these helical bands dynamically. At this invention we give an algorithm of this
artificial left ventricle construction that of course the effect of the blood
flow in LV is observed with making beneficiary used of sensors to obtain this
effecting, something like to lifegates problem. The main problem is to evaluate
powers that are interacted between elastic body (left ventricle) and fluid
(blood). The main goal of this invention is to show that artificial heart is
not just a pump, but mechanical modeling of LV wall and its interaction with
blood in it (blood movement modeling) can introduce an artificial heart closed
to natural heart.