The instrument has a maximum capacity of 100 Kgf.
The accuracy of the instrument is ±1%.
The instrument offers a test speed of 12.5±2.5 mm/min.
The instrument requires a power supply of 220 volts Single phase AC with a frequency of 50 Hz.
The dimensions of the device are 60 cm X 50 cm X 125 mm (L X W X H).
The approximate net weight of the instrument is 93 kg, and gross weight of the device is 110 kg when packed.
The instrument complies with IS 7063-2 standard.
A digital readout is provided with the instrument for the easy and accurate reading of test results.
An S type load cell is incorporated in the machine for accurate electronic force measurement.
There is a program lock provided with the machine for prevention of accidental resetting of Peak load indicator.
The instrument has an electromechanical loading system.
The instrument is operated through sophisticated electronic controls for precision and ease of operation.