extracardiac shunts ( ecs ) are commonly used for the palliation of patients with congenital heart disease ( chd ) who are not candidates for early complete repair .
after placement of a palliative ecs , a period of months may be required to achieve sufficient somatic growth , pulmonary artery growth , and fall in pulmonary vascular resistance to allow candidacy for definitive intracardiac repair . during that time
clotting factors and platelets used at the time of surgical placement of ecs predispose to early layering of thrombus within the lumen of ecs , which potentially leads to cellular ingrowth and a higher propensity toward occlusion .
histopathologic studies have shown that early thrombus deposition may lead to luminal ingrowth consisting of fibrous tissue . in a nonrandomized observational study of infants with systemic to pulmonary ecs ,
in another study , occlusion > 50% was noted in 23% of ecs examined at the time of elective take down .
efforts to maintain ecs patency have the potential to limit morbidity and mortality in chd .
imageguided high mechanical index ( mi ) impulses from a modified diagnostic ultrasound ( us ) system during a systemic microbubble ( mb ) infusion dissolve intravascular thrombi without the need for fibrinolytic therapy .
cavitation and resultant shear forces that lead to a microstreaming phenomenon are proposed mechanisms for sonothrombolysis .
contrast mbs serve as nuclei for cavitation , lowering the peak negative pressure threshold required to induce cavitation , thus enhancing the thrombolytic effects of us . recently ,
the effectiveness of mbenhanced us lysis has been shown in dissolving thrombi that formed on central venous catheters in vivo .
however , no studies to date have examined whether this technique could prevent thrombus formation in vessels or shunts at higher risk for this .
based on these studies of thrombi dissolution , we hypothesized that the microstreaming induced by mb cavitation may be an effective method to prevent the early thrombus accumulation ( thromboprophylaxis ) on surgically placed grafts that lead to subsequent intimal hyperplasia within the graft and resultant occlusion .
accordingly , the specific aim of this study was to examine the effectiveness of thromboprophylaxis using usguided mb treatments in arteriovenous ( av ) , and arterioarterial ( aa ) shunts placed in a porcine model .
the study was approved by the institutional animal care and use committee of the university of nebraska medical center and was in compliance with the standards in the guide for the care and use of laboratory animals .
the pig was fasted overnight and preanesthetized with an intramuscular mixture of telazol ( 4.4 mg / kg ) , ketamine ( 2.2 mg / kg ) , and xylazine ( 2.2 mg / kg ) .
intramuscular atropine ( 0.05 mg / kg ) was used to dry oraltracheal secretions and prevent bradycardia during intubation . following placement of a venous line in the lateral or medial auricular vein ,
the pig was intubated and isoflurane inhalation anesthesia ( induction at 4% , maintained at 1.0% to 1.8% ) administered .
the animal was placed on a ventilator at a volume of 11 cc / kg of air and at a rate of 15 breaths / minute .
after the pig was placed under anesthesia and after preparation of the surgical site , a midline cervical incision was made . through the incision ,
two 4mm diameter , 2.5 cm long , heparinbonded expanded polytetrafluoroethylene shunts ( propaten vascular graft ; gore medical ) were implanted : one interposition within the left carotid ( arterioarterial , aa ) , and one sidetoside av shunt from the right carotid artery to the right internal jugular vein .
this shunt type is commonly used in standard of care clinical practice in humans . for
the interposition shunt , each vessel was clamped proximally and distally and a 2.5 cm segment of vessel was excised . each anastomosis was created with running 60 polypropylene suture ( prolene ; ethicon ) .
both anastomoses were completed before releasing the vessel clamps , ie , the graft was not clamped . for the av shunt ,
4 clamps were applied  to the artery and the vein both proximally and distally  and all were removed when both anastomoses were completed .
the grafts were observed until hemostasis , and then the incision was closed with a subcutaneous and a skin suture layer ( figure 1 ) .
representative examples of surgically implanted rightsided arteriovenous ( av ) shunt from the right carotid artery to the right internal jugular vein , and left sided arterioarterial ( aa ) interposition shunt within the left carotid artery ( black arrows ) .
these mb have a diameter of 1.00.1 m , and concentration of 1.5 to 3.010/ml .
the mb infusion was prepared by diluting 2 ml of the mrx801 in 100 ml of 0.9% saline and infused at a rate of 1.0 ml / min .
a commercially available us system ( ie33 ; philips medical systems ) and diagnostic transducer ( s51 ; philips ) were used .
the us system had low mi contrast sensitive imaging pulse sequences ( power modulation ) that were applied in between high mi impulses to assist in the detection of mb .
the 3 randomized groups tested were : ( 1 ) 2d intermittent guided high mi with intravenous mb ( group 1 ) ; ( 2 ) 2dguided high mi us with intravenous 0.9% saline ( group 2 ) ; and ( 3 ) controls that received no treatment ( group 3 ) .
randomized transcutaneous treatment groups ( 1 or 2 ) and controls were assigned immediately after surgical placement of ecs . the first treatment regimen ( in groups 1 and 2 )
was begun within 2 hours of placement of the ecs , because it is anticipated that this is when significant initial platelet and fibrin deposition occurs .
thirtyminute treatments were given separately to each side ( av shunt on the right side , aa shunt on the left ) .
the mb or 0.9% saline were infused through a peripheral vein . in group 1 ,
the application of brief high mi impulses ( 1.1 mi ; pulse duration 20 s ; frame rate 20 hz ) was guided to allow replenishment of mb within the region of interest .
this frequency and pulse duration for therapeutic us application was chosen based on previous in vitro work from our laboratory .
pulmonary hypertension , mediated by the vasoconstrictor thromboxane has been reported in pigs after exposure to intravenous mb .
this reaction is presumed to be secondary to speciesrelated pulmonary intravascular macrophages ; the macrophage phagocytotic process resulting in the release of thromboxane causing pulmonary hypertension .
ketorolac 60 mg and methylprednisolone 40 mg to prevent pulmonary hypertensive responses to mb were administered intravenously in groups 1 and 2 prior to each treatment .
oxygen saturation by pulse oximetry , arterial blood pressure , and 3lead electrocardiograms were measured before , during , and for 30 minutes after us and mb therapy .
the shunts were imaged using color doppler us ( 4vc1 vascular transducer ; siemens ) before and after each treatment ( figure 2 ) .
a single experienced echocardiographer ( jw ) recorded doppler data as good , reduced , or absent based on color doppler information obtained from within the shunt .
settings of low 2dimensional gain , high color gain , small color overlay , zoom mode , and a single focal zone were used to maximize the sampling rate . as the doppler angles were inconsistent between the implanted shunts ,
the arteriovenous ( left panels ) and arterioarterial ( right panels ) shunts are demonstrated ( asterisks ) on 2dimensional and color doppler imaging , and during thromboprophylactic application of microbubbles and high mechanical index pulse sequences to insonify each shunt .
aa indicates arterioarterial ; av , arteriovenous . each animal was sacrificed at the completion of 4 treatments in groups 1 and 2 , and at a similar time point in group 3 .
each ecs was removed and placed in 10% neutral buffered formalin immediately , and processed according to standard tissue processing techniques .
the shunts were embedded in paraffin wax , sectioned , stained with hematoxylin and eosin for microscopy and reviewed by a pathologist ( sjr ) .
digital images of each section were taken through the iscan coreo slide scanner ( ventana medical systems inc ) at 20 magnification .
images of histopathologic sections from the explanted ecs were reviewed by the pathologist ( sjr ) , and under the pathologist 's direction , a blinded experienced observer ( jra ) performed quantitation of the inlumen residue in each shunt using ventana image viewer 3.1 software .
the area inside the endothelial cell lining of the tunica intima was included in the measurement .
areas calculated for each shunt section were added up to quantify thrombus accumulation ( figure 3 ) .
a , section of an av shunt in a group 1 animal with minimal intimal thickening and no luminal thrombus .
b , section of an av shunt with a mixture of blood , platelets and fibrin partially occluding its lumen .
the luminal composition was a mixture of platelets and fibrin with some endothelial cell ingrowth .
in addition , mononuclear cells , collagen and areas of calcification were seen in some of the sections that were occluded .
therefore , median thrombus burden was reported for us with mb , us with saline , and control groups , and the kruskalwallis test was applied to assess differences among these groups .
the student 's t test was used for comparison of hemodynamic measurements in treatment groups 1 and 2 . to assess the interobserver agreement ,
histopathologic thrombus burden measurements were repeated in 10 randomly chosen aa and av shunts by a second blinded observer .
blandaltman plots were derived to identify possible bias ( mean divergence ) and the limits of agreement ( 2 standard deviation of the divergence ) .
statistical analysis was performed using commercially available software ( minitab version 16.1 ; minitab inc ) .
the study was approved by the institutional animal care and use committee of the university of nebraska medical center and was in compliance with the standards in the guide for the care and use of laboratory animals .
the pig was fasted overnight and preanesthetized with an intramuscular mixture of telazol ( 4.4 mg / kg ) , ketamine ( 2.2 mg / kg ) , and xylazine ( 2.2 mg / kg ) .
intramuscular atropine ( 0.05 mg / kg ) was used to dry oraltracheal secretions and prevent bradycardia during intubation . following placement of a venous line in the lateral or medial auricular vein ,
the pig was intubated and isoflurane inhalation anesthesia ( induction at 4% , maintained at 1.0% to 1.8% ) administered .
the animal was placed on a ventilator at a volume of 11 cc / kg of air and at a rate of 15 breaths / minute .
after the pig was placed under anesthesia and after preparation of the surgical site , a midline cervical incision was made . through the incision ,
two 4mm diameter , 2.5 cm long , heparinbonded expanded polytetrafluoroethylene shunts ( propaten vascular graft ; gore medical ) were implanted : one interposition within the left carotid ( arterioarterial , aa ) , and one sidetoside av shunt from the right carotid artery to the right internal jugular vein .
this shunt type is commonly used in standard of care clinical practice in humans . for
the interposition shunt , each vessel was clamped proximally and distally and a 2.5 cm segment of vessel was excised . each anastomosis was created with running 60 polypropylene suture ( prolene ; ethicon ) .
both anastomoses were completed before releasing the vessel clamps , ie , the graft was not clamped . for the av shunt ,
4 clamps were applied  to the artery and the vein both proximally and distally  and all were removed when both anastomoses were completed .
the grafts were observed until hemostasis , and then the incision was closed with a subcutaneous and a skin suture layer ( figure 1 ) .
representative examples of surgically implanted rightsided arteriovenous ( av ) shunt from the right carotid artery to the right internal jugular vein , and left sided arterioarterial ( aa ) interposition shunt within the left carotid artery ( black arrows ) .
these mb have a diameter of 1.00.1 m , and concentration of 1.5 to 3.010/ml .
the mb infusion was prepared by diluting 2 ml of the mrx801 in 100 ml of 0.9% saline and infused at a rate of 1.0 ml / min .
a commercially available us system ( ie33 ; philips medical systems ) and diagnostic transducer ( s51 ; philips ) were used .
the us system had low mi contrast sensitive imaging pulse sequences ( power modulation ) that were applied in between high mi impulses to assist in the detection of mb .
the 3 randomized groups tested were : ( 1 ) 2d intermittent guided high mi with intravenous mb ( group 1 ) ; ( 2 ) 2dguided high mi us with intravenous 0.9% saline ( group 2 ) ; and ( 3 ) controls that received no treatment ( group 3 ) .
randomized transcutaneous treatment groups ( 1 or 2 ) and controls were assigned immediately after surgical placement of ecs . the first treatment regimen ( in groups 1 and 2 )
was begun within 2 hours of placement of the ecs , because it is anticipated that this is when significant initial platelet and fibrin deposition occurs .
thirtyminute treatments were given separately to each side ( av shunt on the right side , aa shunt on the left ) .
the mb or 0.9% saline were infused through a peripheral vein . in group 1 ,
the application of brief high mi impulses ( 1.1 mi ; pulse duration 20 s ; frame rate 20 hz ) was guided to allow replenishment of mb within the region of interest .
this frequency and pulse duration for therapeutic us application was chosen based on previous in vitro work from our laboratory .
pulmonary hypertension , mediated by the vasoconstrictor thromboxane has been reported in pigs after exposure to intravenous mb .
this reaction is presumed to be secondary to speciesrelated pulmonary intravascular macrophages ; the macrophage phagocytotic process resulting in the release of thromboxane causing pulmonary hypertension .
ketorolac 60 mg and methylprednisolone 40 mg to prevent pulmonary hypertensive responses to mb were administered intravenously in groups 1 and 2 prior to each treatment .
oxygen saturation by pulse oximetry , arterial blood pressure , and 3lead electrocardiograms were measured before , during , and for 30 minutes after us and mb therapy .
the shunts were imaged using color doppler us ( 4vc1 vascular transducer ; siemens ) before and after each treatment ( figure 2 ) .
a single experienced echocardiographer ( jw ) recorded doppler data as good , reduced , or absent based on color doppler information obtained from within the shunt .
settings of low 2dimensional gain , high color gain , small color overlay , zoom mode , and a single focal zone were used to maximize the sampling rate . as the doppler angles were inconsistent between the implanted shunts ,
the arteriovenous ( left panels ) and arterioarterial ( right panels ) shunts are demonstrated ( asterisks ) on 2dimensional and color doppler imaging , and during thromboprophylactic application of microbubbles and high mechanical index pulse sequences to insonify each shunt .
each animal was sacrificed at the completion of 4 treatments in groups 1 and 2 , and at a similar time point in group 3 .
each ecs was removed and placed in 10% neutral buffered formalin immediately , and processed according to standard tissue processing techniques .
the shunts were embedded in paraffin wax , sectioned , stained with hematoxylin and eosin for microscopy and reviewed by a pathologist ( sjr ) .
digital images of each section were taken through the iscan coreo slide scanner ( ventana medical systems inc ) at 20 magnification .
images of histopathologic sections from the explanted ecs were reviewed by the pathologist ( sjr ) , and under the pathologist 's direction , a blinded experienced observer ( jra ) performed quantitation of the inlumen residue in each shunt using ventana image viewer 3.1 software .
the area inside the endothelial cell lining of the tunica intima was included in the measurement .
areas calculated for each shunt section were added up to quantify thrombus accumulation ( figure 3 ) .
representative examples of residual thrombi within av shunt . a , section of an av shunt in a group 1 animal with minimal intimal thickening and no luminal thrombus .
b , section of an av shunt with a mixture of blood , platelets and fibrin partially occluding its lumen .
the luminal composition was a mixture of platelets and fibrin with some endothelial cell ingrowth .
in addition , mononuclear cells , collagen and areas of calcification were seen in some of the sections that were occluded .
therefore , median thrombus burden was reported for us with mb , us with saline , and control groups , and the kruskalwallis test was applied to assess differences among these groups .
the student 's t test was used for comparison of hemodynamic measurements in treatment groups 1 and 2 . to assess the interobserver agreement ,
histopathologic thrombus burden measurements were repeated in 10 randomly chosen aa and av shunts by a second blinded observer .
blandaltman plots were derived to identify possible bias ( mean divergence ) and the limits of agreement ( 2 standard deviation of the divergence ) .
statistical analysis was performed using commercially available software ( minitab version 16.1 ; minitab inc ) .
a total of 18 pigs had a placement of 36 ecs ( 12 pigs for thromboprophylaxis protocols and 6 controls ) .
table 1 shows the hemodynamic measurements in all animals prior to , during , and after the treatments summarized for all 4 treatments . there were no significant differences in hemodynamic parameters between groups 1 and 2 with any of the randomized therapies .
color doppler assessments of the shunts showed reduced or absent color doppler flow during 5 of 24 treatments in 2 group 1 pigs : absent flow was noted during 2 treatments in 1 and reduced flow was seen for 3 treatments in the second pig . among the group 2 pigs , absent or reduced flow
these included 6 instances of absent flow in 2 pigs and 2 instances of reduced flow in a third pig .
these color doppler flow changes in both groups were mostly observed during the third and fourth treatments .
hemodynamic indices compared between groups 1 and 2 for all treatments act indicates activated clotting time ; dbp , diastolic blood pressure ; hr , heart rate ; mb , microbubble ; sbp , systolic blood pressure ; spo2 , systemic oxygen saturation ; us indicates ultrasound . in the untreated animals ( group 3 ) , the median thrombus burden in av shunts was 5.10 mm compared with 4.05 mm in aa shunts ( p=0.199 ) . in the av shunts , group 1
had less median thrombus burden ( 2.5 mm ) in the av shunts compared with group 2 ( 5.6 mm ) and group 3 ( 7.5 mm ) pigs ( p=0.006 ; table 2 ) . among the explanted av shunts ,
no total lumen occlusions were observed in group 1 , whereas one av shunt total occlusion was seen in groups 2 and 3 .
differences in thrombus burden for explanted shunts compared between groups aa indicates arterioarterial ; av , arteriovenous ; mb , microbubble ; us indicates ultrasound .
the luminal composition was predominantly a mixture of organized blood , platelets , and fibrin ; collagen , mononuclear cells , and areas of calcification were also seen in the sections that were occluded ( figure 3 ) .
sections of av shunt histology in pigs treated with guided us with mbs showed reductions in thrombus formation along the entire circumference of the vessel . no difference in median thrombus burden
was seen among the 3 groups for aa shunts ( 4.05 mm in group 1 , 4.25 mm in group 2 , 4.50 mm in group 3 , p=0.895 ) .
blandaltman analysis showed good interobserver agreement for thrombus burden measurements as represented in figure 4 .
blandaltman analysis of interobserver agreement of residual thrombus burden in 10 randomly chosen aa and av shunts .
in the untreated animals ( group 3 ) , the median thrombus burden in av shunts was 5.10 mm compared with 4.05 mm in aa shunts ( p=0.199 ) . in the av shunts ,
group 1 had less median thrombus burden ( 2.5 mm ) in the av shunts compared with group 2 ( 5.6 mm ) and group 3 ( 7.5 mm ) pigs ( p=0.006 ; table 2 ) . among the explanted av shunts ,
no total lumen occlusions were observed in group 1 , whereas one av shunt total occlusion was seen in groups 2 and 3 .
differences in thrombus burden for explanted shunts compared between groups aa indicates arterioarterial ; av , arteriovenous ; mb , microbubble ; us indicates ultrasound .
the luminal composition was predominantly a mixture of organized blood , platelets , and fibrin ; collagen , mononuclear cells , and areas of calcification were also seen in the sections that were occluded ( figure 3 ) .
sections of av shunt histology in pigs treated with guided us with mbs showed reductions in thrombus formation along the entire circumference of the vessel . no difference in median thrombus burden
was seen among the 3 groups for aa shunts ( 4.05 mm in group 1 , 4.25 mm in group 2 , 4.50 mm in group 3 , p=0.895 ) .
blandaltman analysis showed good interobserver agreement for thrombus burden measurements as represented in figure 4 .
blandaltman analysis of interobserver agreement of residual thrombus burden in 10 randomly chosen aa and av shunts .
the clinical implications of thrombosis after surgical procedures in chd , including systemic to pulmonary shunts , bidirectional cavopulmonary shunts , and fontan palliation are significant .
thrombotic lumen occlusion is the usual cause for ecs failure ; this development leads to significant acute mortality and morbidity in chd .
no published randomized controlled trials are available to guide the thromboprophylactic management of patients with ecs , so use of anticoagulation is a subject of debate . although antiplatelet therapy has been studied extensively in adults , evidence for its benefit in infants and young children is limited .
while the use of aspirin was associated with a reduction in ecs thrombosis in some studies , others have failed to demonstrate a beneficial effect . in theory
, aspirin alone may be insufficient to prevent arterial thrombosis because it inhibits only one pathway ( cyclooxygenase ) of platelet activation , and would have minimal effect in treating thrombi that would be expected in ecs .
moreover , the optimal pediatric dose of aspirin for inhibition of platelet aggregation is unknown .
aspirin is associated with reye syndrome in infants and children , and may cause bleeding complications , particularly in the presence of an underlying hemostatic defect or with concurrent thrombolytic therapy .
the use of fibrinolytic drugs has not been established in children , and carries significant risk of major hemorrhage .
attempted recanalization of acutely thrombosed ecs by cathetermediated thrombus disruption or rheolytic thrombectomy has been met with some success .
subacute ecs occlusion develops from neointimal proliferation in addition to thrombus formation within the shunt , so recanalization is more difficult in such cases . therefore , treatments designed to locally prevent thrombus formation / reduce thrombus burden and maintain ecs patency would be ideal , and avoid these systemic complications .
mbs coupled with us have been used to dissolve thrombi without thrombolytic agents , and also have been shown to enhance the effectiveness of exogenous tissue plasminogen activator . in in vivo models , successful dissolution of arterial and venous thrombi is achievable with guided high mi impulses delivered from a diagnostic us transducer .
the advantage of this approach is that it permits simultaneous low mi mb sensitive imaging pulse sequence schemes to detect the mb with the same transducer , and allows one to control the region in which therapy is desired .
the present study describes the application of this treatment for thromboprophylaxis of ecs in vivo .
an imageguided approach ensured the presence of mb near the shunt when the high mi impulses were applied .
the arteriovenous connection generates an intravascular pressure differential , similar to the blalockthomastaussig shunt or sano shunt used for chd palliation .
these shunts are subject to intimal proliferation due to shear forces of turbulent blood flow , and subsequent thrombus deposition on the irregular surface where blood flow slows .
this was presumably because of the ( 1 ) higher baseline flow rates and consequently lower predisposition for stasisrelated thrombus formation , and ( 2 ) the relatively short duration ( 4 weeks ) of placement .
we speculate that the interposition shunts within the carotid artery ( aa ) had improved flow compared to the sideside connected av shunts . for the av shunts
, there is the possibility of narrowing at the sidetoside anastomoses and turbulence of flow due to pressure differences between the vessels , which may have led to increased thrombogenesis .
the safety and success of this noninvasive thromboprophylactic application in our porcine model increase the potential for translation to av shunts in the human . besides chd , a potential therapeutic application of av shunt thromboprophylaxis is in patients with endstage renal disease where the use of such shunts is very common
. the application may also be particularly advantageous in infants and children because of the proximity of vascular structures to allow improved penetration of transcutaneously applied us impulses .
moreover , recent clinical trials have indicated that us contrast agents are safe and effective in children and young adults .
comparisons of efficacy between usmb thromboprophylaxis and pharmacologic ( antiplatelet or anticoagulant ) prophylaxis require further study . the number of animals studied was relatively small and the duration of placement of shunts prior to explantation and histopathologic examination was 4 weeks . considering that av shunts in the human are placed for substantially longer periods of time , whether the same results could be extrapolated for longer durations / older shunts
the purpose of this study was to explore the feasibility of ultrasound and microbubble prophylaxis , but further work is needed to explore this treatment approach in shunts placed for longer periods of time .
evaluation for the possibility of soft tissue injury due to bioeffects associated with highmi impulses from the transducer requires further investigation .
there is potential for unwanted us mechanical bioeffects at mi>1.0 , however , these are likely to be minimal because the us application used in the study was with a commercially available diagnostic transducer , and the highmi therapeutic pulses were significantly lower than the food and drug administration recommended mi limit for clinical use .
the mbs used in this study have been successfully used in human trials as adjuvant for arterial sonothrombolysis .
although these mbs are not yet commercially available , they are similar in shell composition and mb size to commercially available lipid encapsulated mbs . finally , these results would only be applicable to av shunts .
other palliative connections in chd are also subject to thrombotic occlusion , and the role of thromboprophylaxis with us and mb must be explored in such settings , where attenuation of the us beam may play a role in achieving adequate inertial cavitation of the mb .
the number of animals studied was relatively small and the duration of placement of shunts prior to explantation and histopathologic examination was 4 weeks . considering that av shunts in the human are placed for substantially longer periods of time , whether the same results could be extrapolated for longer durations / older shunts
the purpose of this study was to explore the feasibility of ultrasound and microbubble prophylaxis , but further work is needed to explore this treatment approach in shunts placed for longer periods of time .
evaluation for the possibility of soft tissue injury due to bioeffects associated with highmi impulses from the transducer requires further investigation .
there is potential for unwanted us mechanical bioeffects at mi>1.0 , however , these are likely to be minimal because the us application used in the study was with a commercially available diagnostic transducer , and the highmi therapeutic pulses were significantly lower than the food and drug administration recommended mi limit for clinical use .
the mbs used in this study have been successfully used in human trials as adjuvant for arterial sonothrombolysis .
although these mbs are not yet commercially available , they are similar in shell composition and mb size to commercially available lipid encapsulated mbs . finally , these results would only be applicable to av shunts .
other palliative connections in chd are also subject to thrombotic occlusion , and the role of thromboprophylaxis with us and mb must be explored in such settings , where attenuation of the us beam may play a role in achieving adequate inertial cavitation of the mb .
transcutaneous us and intravenous mb are capable of decreasing thrombus burden in arteriovenous shunts without the need for antiplatelet agents , and may be a method of preventing progressive occlusion of a highflow , small diameter expanded polytetrafluoroethylene tube graft such as the modified blalock shunt or the sano shunt .
we appreciate the expert technical assistance of elizabeth stolze , lucas drvol , and gretchen fry of the joint cardiovascular research laboratory of the university of nebraska medical center .
 backgroundpalliative shunts in congenital heart disease patients are vulnerable to thrombotic occlusion 
 . high mechanical index ( mi ) impulses from a modified diagnostic ultrasound ( us ) transducer during a systemic microbubble ( mb ) infusion have been used to dissolve intravascular thrombi without anticoagulation , and we sought to determine whether this technique could be used prophylactically to reduce thrombus burden and prevent occlusion of surgically placed extracardiac shunts.methods and resultsheparinbonded eptfe tubular vascular shunts of 4 mm2.5 cm ( propaten ; w.l gore ) were surgically placed in 18 pigs : a rightsided sidetoside arteriovenous ( av , carotidjugular ) shunt , and a leftsided arterioarterial ( aa , carotidcarotid ) interposition shunt in each animal . after shunt implantation , animals were randomly assigned to one of 3 groups . 
 transcutaneous , weekly 30minute treatments ( total of 4 treatments ) of either guided high mi us+mb ( group 1 ; n=6 ) using a 3% mrx801 mb infusion , or us alone ( group 2 ; n=6 ) were given separately to each shunt . 
 the third group of 6 pigs received no treatments . 
 the shunts were explanted after 4 weeks and analyzed by histopathology to quantify luminal thrombus area ( mm2 ) for the length of each shunt . 
 no pigs received antiplatelet agents or anticoagulants during the treatment period . 
 the median overall thrombus burden in the 3 groups for av shunts was 5.10 mm2 compared with 4.05 mm2 in aa ( p=0.199 ) . 
 group 1 pigs had significantly less thrombus burden in the av shunts ( median 2.5 mm2 ) compared with group 2 ( median 5.6 mm2 ) and group 3 ( median 7.5 mm2 ) pigs ( p=0.006 ) 
 . no difference in thrombus burden was seen between groups for aa shunts.conclusiontranscutaneous us with intravenous mb is capable of preventing thrombus accumulation in arteriovenous shunts without the need for antiplatelet agents , and may be a method of preventing progressive occlusion of palliative shunts . 
Introduction
Methods
Porcine Models of Extracardiac Shunts
Ultrasound and Microbubble Technology
Ultrasound and Microbubble Thromboprophylaxis
Explantation of Shunts and Analysis of Thrombus Burden
Statistical Analysis
Results
Thrombus Accumulation in the Different Treatment Groups
Discussion
Study Limitations
Conclusions
Disclosure
Acknowledgments