exchanging H2O in buffer for D2O - pH vs pD? I have TBS pH 7.4 buffer that I am exchanging all the water out for $\ce{D2O}$. How do I find out what the pD is after all the water is gone? I understand that pH meters don't work for deuterium, and it seems like deuterium also changes the spectral properties of the chromophores in pH paper (or in this case pD paper)

Also, if I want to best mimic the alkaline conditions in $\ce{D2O}$ as $\ce{H2O}$, should pD final = pH initial?
The articles by Glasoe,[1] Covington et. al.,[2] and (much more recently) Krężel and Bal[3] suggest in their intros that pD values can be obtained from a pH calibrated electrode by adding a correction constant of about $0.4$. This was determined by comparing the measured pH when the same amount of acid or base was dissolved in $\ce{H2O}$ and $\ce{D2O}$.
References:

P. F. Glasoe, Citation Classic - use of glass electrodes to measure acidities in deuterium-oxide. Current Contents/ Physical Chemical &amp; Earth Sciences 1979, 16, 12-12. A1979HZ34800001 (pdf) Original paper: P. K. Glasoe, F. A. Long, J. Phys. Chem. 1960, 64,:188-190. Doi:10.1021/j100830a521
A. K. Covington, M. Paabo, R. A. Robinson, R. G. Bates, Anal. Chem. 1968, 40 (4), 700–706. DOI: 10.1021/ac60260a013
A. Krȩżel, W. Bal, J. Inorg. Biochem. 2004, 98 (1), 161-166. DOI: 10.1016/j.jinorgbio.2003.10.001 Available at researchgate.net.