3. Temporarily lock gonio way d (figure 3-2).
5. Try moving the sliding way horizontally to its maximal traveling end and adjust the roller retainer to its central position.
6. Install a dial gauge at the side of the sliding way base level.
9. Move the sliding way to the central position and slightly lock the adjustment screw at the central position (figure 3-7).
10. Repeat operations from (7) to (9) until there is no gap in horizontal direction. Caution against applying excessive pre-stressed loads.
11. Once there is no gap in the horizontal direction, the dial deflection on the gauge is minimal and stable when the sliding way is moved horizontally; at this time carry out final calibration of the pre-stressed loads by repeating operations from (7) to (9) using the torque force recommended for locking screws.