Accesories

Electromagnetic Shieldings

Description

An electromagnetic shielding is a conductive layer installed inside a magnetic resonance room to prevent unwanted radio frequency signals inside the room. These signals can compromise the image quality of the system.

The shielding is built using a modular construction system made from wood beams and copper foil, allowing for last minute changes. This system is used for all walls, ceiling and oor. Additional special parts are provided to cover all penetrations such as doors, windows, air inlets and others. This electromagnetic shielding has different components and each one has to work according to stringent speci cations de ned by the system that goes in it

Characteristics

Doors are the most sensitive part of all the RF parts of a electromagnetic shielding. For this reason, it is very important to put special care on its functions and characteristics Also the door is the only moving part of an electromagnetic shielding and is the main source of acoustic noise leakage of the room.

Doors

Standard Size

1200mm x 2080mm Open Area

Floor level

15mm above nished oor

Electrical contact system

Finger strips (copper beryllium)

Opening Force resistance

10kgs

Warranty

2 years (excluding ngers)

Garantía

2 años (excluye fingers)

Frame Material

Stainless Steel

Door Material

Copper over Chipboard with acoustic materials

Door Switch

Mechanical

Rw Standard Door

25db

Rw Acoustic Door

35db

Windows are key for visual contact between patient and technologist because it allows to control anxiety and to verify exam procedure in general. At the same time, windows have to comply with the requirements of the systems and to keep shielding integrity.