A 75-year-old caucasian male patient with a diagnosis of dilative cardiomyopathy (echocardiographic biplane ejection fraction (EF) 23%) had sinus rhythm with left LBBB ECG (Figure a-b) (QRS with 136 ms) and a profound dyssynergic contraction profile. There was a history of Diabetes Typ II, hypertension and mild renal insufficiency (serum creatinine 1.63 mg/dl). Significant coronary artery disease had been excluded by angiography. Currently, he suffered from New York Heart Association (NYHA) III heart failure status and was admitted for implantation of a CRT – implantable cardioverter defibrillator (ICD). The operative procedure was unremarkable, although some difficulties advancing the right electrode into the right ventricle were reported. Chest X-ray after implantation (Figure a-b) excluded pneumothorax and electrodes appeared well placed. The patient was discharged one day after the procedure when the pacing ECG showed a narrowed QRS complex (Figure c-d) with optimal values of all three electrodes (Table ).\nOne month later on follow-up, the patient reported a marked improvement of his exercise capacity (NYHA II). Electrode values remained stable. An unscheduled echocardiogramm (Philips iE33) revealed a marked reduction of left ventricular dimensions with an improved monoplane ejection fraction (51%). However, through the mitral valve leaflets a thick electrode proceeded into the apex of the left ventricle. An additional transesophageal echocardiographic examination demonstrated the coronary sinus (CS) electrode in place (Figure a) but a malposition of the RV electrode passing the patent foramen ovale (PFO) via an Eustachian valve into the left ventricle (3 b-d).\nAfter studying the literature and informing the patient about possible consequences a revision procedure was performed. The RV electrode was easily removed from the left ventricle but could not be safely placed in the right heart without a sheat. A new electrode was advanced and fixed at the base of the right ventricle. Postprocedure chest x-ray (Figure c-d), ECG (Figure e-f), and measurements (Table ) all documented good results. This time a postprocedure echocardiogramm showed an intact mitral valve, persistent good left performance, and the RV electrode within the right heart.\nThis was a clinical study and no experimental work. No ethical committee had to be involved. Everything happened in the compliance with the Helsinki Declaration.
A 23-year-old female with Ebstein's anomaly of tricuspid valve underwent one and half ventricular repair with tricuspid valve annuloplasty, atrial septal defect closure and cavopulmonary anastamosis of right superior venacava to the right pulmonary artery. She also needed dual chamber pacing for symptomatic bradycardia secondary to sinus nodal and atrioventricular nodal disease. There was no pre-excitation or tachyarrhythmias. Left atrial and left ventricular epicardial leads were connected to a pulse generator that was implanted in epigastric subcutaneous pocket. There was synchronous atrioventricular pacing with acceptable pacing thresholds during the time of implantation. Three months later, new onset symptomatic diaphragmatic pacing was observed from atrial lead. So the pacemaker was reprogrammed to pace ventricle alone. After another two months, diaphragmatic pacing was observed from ventricular lead too. The threshold voltage for ventricular capture was more than that of the diaphragm. The reasons for delayed onset diaphragmatic pacing through both leads almost five months after the epicardial lead placement and pacemaker implantation was not understood. Her symptoms related to bradycardia warranted permanent dual chamber pacing. Since extensive dissections will be needed through the previous surgical scars and adhesions in midline sternotomy and left thoracotomy, a second epicardial lead repositioning was not considered. A minimal incision right anterolateral thoracotomy was made and two introducer sheaths were inserted through two pursestring sutures placed on the right atrial free wall.() Two screw-in endocardial leads were inserted through the sheaths under transesophageal echocardiographic guidance into the right atrial appendage tip and right ventricular apex to achieve atrioventricular pacing. On a follow up of nine months, the patient was asymptomatic, pacing was synchronous and chest X-ray showed both the new and old leads. ()