ADE shoots a droplet from a source well upward to an inverted receiving plate positioned above the source plate. Liquids ejected from the source are captured by dry plates due to surface tension. For larger volumes, multiple droplets can be rapidly ejected from the source (typically 100 to 500 droplets/sec) to the destination with the coefficient of variation typically <4% over a volume range of two orders of magnitude.[5]