The elastic constriction drives the liquid through the tubing and eventually through a flow restrictor out into the patient connection. The pressure is consistent until near the end of the infusion, when there will usually be a pressure spike resulting in a higher flow rate. The accuracy of the flow rate is controlled by a flow restrictor (glass capillary or steel cannula) that is molded into the tubing of the system or placed within the elastomeric reservoir.