Athena Philis?Tsimikas|Anna Chang|Lupe Miller
Precision, Accuracy, and User Acceptance of the OneTouch SelectSimple Blood Glucose Monitoring System
Scripps Whittier Diabetes Institute|John Muir Health
onetouch selectsimple blood glucose monitoring system bgms device selfmonitoring blood glucose designed ease use alarms alert subjects low 2069 mgdl 1138 mmolliter high 180239 mgdl 99132 mmolliter high 240600 mgdl 133331 mmolliter blood glucose readingsrepeatability blood intermediate precision aqueous controls examined using blood one donor adjusted different glucose concentrations tested 10 meters 1 teststrip lot system accuracy evaluated blood samples 100 diabetes patients tested 3 teststrip lots compared reference system ysi 2300 stat test user accuracy patients n 156 health care professionals hcps tested subject blood selectsimple twice health care professionals evaluated subject bgms technique 35 day hometesting period users evaluated instructions use responded user acceptance questionnairein repeatability intermediate precision testing selectsimple bgms coefficient variation 5 standard deviation 5 mgdl clinical accuracy study 100 measurements 75 mgdl 42 mmolliter within 15 mgdl 08 mmolliter reference value 996 measurements 75 mgdl 42 mmolliter within 20 patients able use bgms appropriately evaluated easy use acceptance selectsimple bgms within predefined limitsin studies selectsimple bgms met criteria precision system user accuracy easy use well accepted patients