New pulse oximetry detection based on the light absorbance ratio as determined from amplitude modulation indexes in the time and frequency domains.
The Pandemic COVID-19 situation, a pulse Oximetry is significant to detect a varying blood oxygen saturation of a patient who needed the device to operate with continuous, rapid, high accuracy, and immune of moving artifacts. In this article, three main schemes for low-complexity pulse oximetry detection are proposed. In the first scheme, the light absorbance ratio (  ) is obtained by separating the red and infrared photoplethysmography (PPG) amplitude modulation (AM) signals from the frequency-division multiplexing (FDM) signal with two different bandpass filters (BPFs), determining the ratio of modulation index of red and infrared PPG AM signals. In the second scheme, the output PPG AM signals for the red and infrared light wavelengths from the BPFs are transformed into the frequency domain such that the AC components of both PPG AM signals are the magnitudes of the highest peaks in their respective sidebands, while the DC components are the magnitude of their carrier frequencies; then, the  ratio of the red PPG AM signal is divided by the  ratio of the infrared PPG AM signal is  . In the last scheme, the FDM signal is transformed into the frequency domain without being passed through any BPF, and  is obtained in the same way as in the same second scheme. Experimental results obtained by using the first scheme have an average error of about 0.7138%, for the second and the last scheme have an average error of about 1%, and all the methods agree with the corresponding mathematical model.