markdown:
Khoa Học Tự Nhiên
Hóa học - Dầu khí
Tải bản đầy đủ - 0 (trang)
1 Scanning Tunneling Microscopy (STM)
1 Scanning Tunneling Microscopy (STM)
Tải bản đầy đủ - 0trang
50
2
Microscopy – Nanoscopy
Fig. 2.1 Device with a piezoelectric base plate (1) and three feet (2) allowing the sample to be
approached toward the tip which is mounted on a piezoelectric tripod scanner (3). (Reprinted with
permission from [2.20]. © 1994 Cambridge University Press)
Fig. 2.2 Atomically resolved
image of KBr(001) in contact
AFM mode. The small and
large protrusions are
attributed to K+ and Br−
ions, respectively. (Reprinted
with permission from [2.21].
© 2003 American Physical
Society)
(iii) Atomic resolution together with atomic positioning of the tip leads to the
controlled manipulation on the atomic level and the prospects of creating
atomic-scale devices.
Scanning probe microscopes are operated after the following common principles:
A very fine, nanosized probe tip interacts with a specimen surface and is scanned
line by line over this surface. Commonly, the relative movement between tip and
specimen surface is effectuated by piezodrives (see Fig. 2.1). The data registered
by the tip in dependence of the lateral and normal movements are stored in a computer and displayed in a color code on a monitor (see Fig. 2.2). With the scanning
probe microscopy techniques presently available, a wide field of applications in
nanotechnology emerges.
2.1.1 Scanning Units, Electronics, Software
For atomic resolution of a surface, isolation of the microscope against vibrations is
required. Piezoelectric drives can be used for the scanning units operating the 3D tip
2.1
Scanning Tunneling Microscopy (STM)
51
motion (see Fig. 2.1). The bars or tubes with high stiffness offer high resonance frequencies making video scan rates possible without exciting vibrations. This enables
the study of dynamic processes as, e.g., atomic surface diffusion. Furthermore, high
sensitivity, large scan areas, and low thermal drifts are of importance.
By the electronics set-up, a tip bias voltage is supplied giving rise to a tunneling
current which is compared to a preset current level in order to feed the difference
signal into feedback amplifiers in the constant current imaging mode (see below).
The experiments are equipped with computer control and data visualization.
The sensor tip, the most critical component of an STM, should have a small
tip radius and a high-aspect ratio. Sharp tips can be prepared by electrolytical
etching and ion milling [2.20]. Metallic tips can be magnetically modified for highresolution magnetic sensing [2.10]. For chemical sensing on a nanometer scale
carbon nanotube tips have been specifically modified (see Chap. 5.1).
2.1.2 Constant Current Imaging (CCI)
This is a widely used STM mode (see Fig. 2.3). As described earlier [2.20] a
feedback loop adjusts the height of the tip during scanning so that the tunneling
current flowing between tip and specimen is constant. Therefore, the recorded signal Uz Ux , Uy can be translated into a “topography” z (x, y), provided that the
sensitivities of the three piezoelectric drives are known.
In a theoretical treatment of STM the tunneling probability of a 1D contact
can be dealt with in a perturbative time-dependent approach (Bardeen’s transfer
Hamiltonian, see [2.20]). Within Bardeen’s formalism, the tunneling current I can
be evaluated in a first-order time-dependent perturbation theory according to
I=
2π e
f Eμ
1 − f (Eν + eU) − f (Eν + eU) 1 − f Eμ
μ,ν
× Mμν
2
δ Eν − Eμ
Fig. 2.3 Schematics of the constant current mode of STM operation. (Reprinted with permission
from [2.20]. © 1994 Cambridge University Press)
52
2
Microscopy – Nanoscopy
where f (E) is the Fermi function, U is the applied sample bias voltage, Mμν is the
tunneling matrix element between the unperturbed electronic states ψμ of the tip,
and ψν of the sample surface, Eμ (Eν ) is the energy of the state ψμ (ψν ) in the
absence of tunneling, and the delta function describes the conservation of energy
for the case of elastic tunneling. The essential issue is the calculation of the tunneling matrix element Mμν [2.22] which requires explicit expressions for the wave
functions ψμ and ψν of the tip and of the sample surface, respectively. In the simplest model, an s-type tip wave function [2.23] is assumed. Then, the STM images
obtained at low bias voltage in the constant current mode represent contour maps
of constant surface local densities of states (LDOS) at the Fermi energy EF of the
sample surface, provided the s-wave approximation for the tip is justified. In this
model the tunneling current
I ∝ exp (−2κs)
is an exponential function of the distance s between tip and substrate with the decay
rate κ. It should be mentioned that modifications of this model are necessary for the
widely used W or Pt–Ir tips in STM studies because for these materials the density
of states at the Fermi level is dominated by electronic d states rather than s states.
A direct comparison between an experimental constant current scan and a calculated contour of constant local density of states at the Fermi level using the
adatom-on-jellium model was made for the case of xenon atoms adsorbed on a
Ni(110) surface [2.24]. A reasonable agreement appears in Fig. 2.4 because Xe
binds at relatively large distances from metal surfaces with a relatively large size
of the Xe 6s orbital extending further out into the vacuum than the base-surface
wave function.
Fig. 2.4 A comparison of theoretical and experimental normal tip displacement versus lateral tip
displacement for a Xe atom absorbed on a metal surface. (Reprinted with permission from [2.24].
© 1991 American Physical Society)
2.1
Scanning Tunneling Microscopy (STM)
53
The lateral resolution in constant current STM is given by (see [2.20])
L eff = (R + s) /κ
1/2
with the geometrical quantities R (tip radius) and s (distance between sample surface
and front end of the tip) rather than by the wavelength of the tunneling electrons.
This is the typical situation of the so-called near-field microscopes which are operated at distances (1.2–12 nm) between probe and sample that are not so large
compared to the electron wavelength (0.3–1.0 nm). STM facilities are operated
under UHV conditions not only at low temperatures but also at elevated pressures
and temperatures [2.25].
2.1.3 Constant-Height Imaging (CHI)
In constant current imaging the data acquisition rate is limited by the timeconsuming feedback loop. This feedback is switched off in the constant-height
imaging (CHI) mode. In this mode the modulation of the tunneling current with
the full sensitivity of the exponential current dependence on the tip-surface spacing
reflects the atomic-scale topography (see Fig. 2.5). In this mode STM images can
be collected at video rates for the observation of dynamic surface processes such as
Fig. 2.5 Schematic of the
constant-height mode of
STM. (Reprinted with
permission from [2.27].
© 1987 American Institute of
Physics)
54
2
Microscopy – Nanoscopy
diffusion [2.26]. However, the CHI mode is preferably used for zooming to selected
smaller surface areas of sufficient flatness.
2.1.4 Synchrotron Radiation Assisted STM (SRSTM)
for Nanoscale Chemical Imaging
With STM, which conventionally is sensitive to conduction electrons, chemical
information on the composition of a surface can be gained by photo-induced secondary electrons that are produced by core–electron excitation under synchrotron
radiation [2.4]. Since the intensity of the secondary electrons emitted from the surface is proportional to the x-ray absorption probability of core–electrons, one can
obtain the chemical “fingerprint” of an element by measuring with the STM the
secondary electron intensities below and above the characteristic x-ray L absorption
edge of, e.g., Ni or Fe (see Fig. 2.6). By this technique, checkerboard-patterned samples of Fe and Ni stripes on a thick Au film formed on Si (001) can be specifically
detected (Fig. 2.6b, c) with a lateral resolution of ∼ 10 nm.
2.1.5 Studying Bulk Properties and Volume Atomic
Defects by STM
The surface-sensitive STM technique can be used to study bulk properties of a
crystal by exploiting the wave nature of electrons, e.g., in copper and study their
interference patterns on the surface caused by scattering centers in the bulk of the
material [2.28, 2.31]. Most solid-state properties can be characterized by the structure of the Fermi surface – the dividing line between the states occupied by an
Fig. 2.6 (a) Schematic of an Fe and Ni checkerboard pattern as observed by synchrotron radiation
assisted scanning tunneling microscopy (SRSTM). (b) Element-specific image of Fe obtained by
dividing a photo-current image taken at the top of the Fe L3 absorption edge (hν = 706 eV) by that
at the bottom of the edge (698 eV). (c) Element-specific image of Ni obtained around the Ni L3
absorption edge (hν = 852 and 843 eV). (Reprinted with permission from [2.4]. © 2009 American
Physical Society)
2.1
Scanning Tunneling Microscopy (STM)
55
Fig. 2.7 (a) A nearly free-electron gas has a spherical Fermi surface. The blue arrows indicate
the direction of electron propagation at the Fermi surface. (b) In the Fermi surface of Cu certain
directions become preferred due to the non-spherical shape of the Fermi surface. The thick arrows
indicate directions of electron focusing. (c) When a scatterer is present under the surface, the electron wave can be reflected. For a spherical Fermi surface this results in a weak interference pattern
at the surface. (d) When the Fermi surface is not spherical, electron focusing is observed along
certain directions which can give rise to a pronounced interference pattern at the surface [2.31]. (e)
Scanning tunneling micrographs (STM) of four Co atoms below the Cu (111) surface (9 by 9 nm).
The right inset shows (4 by 4 nm) the calculated local density of states (LDOS), whereas the
left inset refers to density-functional theory (DFT) calculations. (Reprinted with permission from
[2.28]. © 2009 AAAS)
electron and the empty states. The simplest model for electrons in a solid is the
nearly-free electron gas with a spherical Fermi surface (Fig. 2.7a). A small deviation from a sphere, such as in Cu (Fig. 2.7b), allows to measure the shape of the
bulk Fermi surface which is most frequently measured by quantum oscillations or
photo-emission spectroscopy and calculated with density-functional theory (DFT)
[2.31]. When an electron is injected from the tip of an STM into a sample, it
propagates as a wave and eventually scatters or arrives back at the tip. This wave
behavior of the electrons in the bulk is not visible in most STM images. The situation changes dramatically when a point defect is incorporated under the surface. A
dramatic increase of the surface interference pattern is observed for Co atoms buried
several layers underneath (Fig. 2.7e). This is ascribed to the shape of the Fermi surface: Along certain spatial directions, the amplitude of the scattered wave decays
very slowly (see arrows in Fig. 2.7b). The electrons are scattered along beams of
electron waves, a phenomenon referred to as electron focusing. When these beams
56
2
Microscopy – Nanoscopy
intersect the surface of the crystal, a strong and characteristically shaped interference pattern is observed containing information about the propagation of electrons
through the bulk of the material, the shape of the Fermi surface, and the characteristics of the scattering potential below the surface. According to calculations [2.28],
separate interference patterns (Fermi surfaces) should be observed for minority spin
versus majority spin electrons in magnetic materials. Kondo resonances of electron scattering at a magnetic atom on a non-magnetic host were studied earlier
[2.29, 2.30].
2.1.6 Radiofrequency STM
STM with an electronic band width as high as 10 MHz has been demonstrated
[2.32]. This allows for high-sensitivity detection of high-frequency mechanical
motion and is expected to be capable of quantum-limited position measurements
(see Sect. 7.2).
2.2 Atomic Force Microscopy (AFM)
The scanning force microscope or atomic force microscope has been invented by
Binnig et al. [2.33]. As shown in Fig. 2.8 a probetip is mounted on a cantilever type
spring. The force interaction between the sample and the tip after approaching each
other causes the cantilever to deflect according to Hooke’s law
F=C×
z
where C denotes the spring constant and z the deflection. The AFM can be operated either in the contact regime or in the non-contact regime where, e.g., small van
der Waals forces may be probed.
The most critical component is the cantilever-type spring with a sharp tip on one
end. For a high sensitivity, a large deflection of a soft spring (with a low spring constant) for a given force is desirable. On the other hand, a high resonance frequency
ω0 = (C/m)1/2
is necessary in order to minimize the influence of mechanical vibrations. Both conditions can be achieved only by keeping the mass, m, and therefore the geometrical
dimensions small so that microfabrication techniques must be employed. Challenges
in receiving atomic resolution and optimization of imaging parameters have been
discussed [2.21].
The applicability of AFM techniques to bulk insulators is of particular importance where electron microscopical, spectroscopical, and STM studies are difficult
because of charging effects.
2.2
Atomic Force Microscopy (AFM)
57
Fig. 2.8 Schematic set-up of an atomic force microscope (AFM) with a four-segment photodiode
for detection of the laser beam deflection due to bending and twisting of the cantilever. (Reprinted
with permission from [2.34]. © 1999 Karlsruher Institut für Technologie)
Mainly optical techniques (see Fig. 2.8) are employed in order to detect the
cantilever deflection. In this set-up a laser beam is reflected from the cantilever
and the cantilever displacement is measured by the detection of the reflected
laser beam using a position-sensitive detector consisting of, e.g., four photoactive segments with eventually sub-Ångström sensitivity. By the four photoactive
cells, vertical and lateral deflections (for, e.g., friction measurements) can be
registered.
2.2.1 Topographic Imaging by AFM in Contact Mode
In this mode, where the tip and the sample are in contact, the interaction force causes
the cantilever to deflect. In the constant force imaging (CFI) mode of AFM which
is analogous to the constant current mode in STM, the cantilever deflection is kept
constant by means of a feedback circuit. The output signal of the feedback loop
Uz is recorded as a function of the (x, y) coordinates and can be translated into
the “topography” z (x, y). As equiforce surfaces are measured by AFM we have to
consider the tip-surface interaction forces in the contact regime.
The invention of the AFM has considerably contributed to an increased interest
in the force picture of quantum mechanical systems. The fundamental relationship
Tài liệu liên quan
Nanoscience the science of the small in physics engineering chemistry biology and medicine
2 Thermal Properties of Nanostructures
4 Giant Magnetoresistance (GMR) and Spintronics
2 Atomic Force Microscopy (AFM)
3 Scanning Near-Field Optical Microscopy (SNOM)
4 Far-Field Optical Microscopy Beyond the Diffraction Limit
5 Magnetic Scanning Probe Techniques
6 Progress in Electron Microscopy
8 Three-Dimensional Atom Probes (3DAPs)
2 Superlattices of Nanocrystals in Two (2D) and Three (3D) Dimensions
4 Nanolayers and Multilayered Systems
Tài liệu mới
1What’s your Exit Strategy?
1What’s the biggest risk?
4Trademarks, Copyright and Industrial Designs
5 Large- Sample Tests in Linear Regression Models
7 Appendix: Test Regressors in Arti cial Regressions
Me = T cơ. R. 106
§4. Lực quán tính và tổng lực, lực tiếp tuyến
§3. quy dẫn khối lượng chuyển động
§1. Mục đích và nội dung
Tìm trên đồ thị Qckmax và tính
Tài liệu bạn tìm kiếm đã sẵn sàng tải về
1 Scanning Tunneling Microscopy (STM)
Nanoscience the science of the small in physics engineering chemistry biology and medicine -0 (trang)
Tải bản đầy đủ ngay(0 tr)
×

Server: Apache


markdown:
## Saigon Hitech thiết kế website Triệt lông hiệu quả
Việc sở hữu website hiện nay không phân biệt cá nhân, cửa hàng bán nhỏ lẻ hay một doanh nghiệp với quy mô đa dạng. Hơn thế, với những thay đổi ngoạn mục về chiến lược thì việc tận dụng một giải pháp mới để nhận được sự quan tâm và sử dụng của khách hàng cũng là điều hiển nhiên. Nhưng trên thực tế thì không phải ai cũng nắm bắt được vấn đề này và xây dựng mục tiêu để thiết kế một website hoàn chỉnh hơn để có cơ hội tiếp cận được khách hàng tiềm năng.
Trước tiên, để
[Saigon Hitech](https://saigonhitech.vn/) thiết kế website Triệt lông hiệu quả hay bất kỳ trang web cho dịch vụ, sản phẩm nào đều cho khách hàng những gợi ý tốt nhất về chuyên môn ứng với lĩnh vực phù hợp để có được kinh nghiệm cao và có thể cho khách hàng tham khảo một số mẫu thiết kế website đơn giản, giao diện đẹp để sớm có thêm một công cụ đắc lực để hỗ trợ kinh doanh với sức cạnh tranh cao so hơn.
Nhờ tông màu hồng nhẹ nhàng làm tông màu chủ đạo chính cho phần giao diện của trang web nên dễ tạo được ấn tượng cho khách hàng trong lĩnh vực làm đẹp. Đồng thời cũng nhờ sự chuyên nghiệp nên tạo được sự tin tưởng, yên tâm cho khách hàng khi tư vấn và quyết định lựa chọn sử dụng dịch vụ. Điều này sẽ tạo điều kiện thuận lợi hơn giữa người nhà cung cấp và người dùng, là sự khởi đầu tốt cho một chiến lược kinh doanh lâu bền trong thời đại khó khăn như hiện nay.
Quý khách hàng có thể liên hệ với đội ngũ chuyên viên tư vấn của Saigon Hitech để được cập nhập thêm những thông tin cần thiết nhằm sớm có cơ hội sử dụng được một dịch vụ
[thiết kế website](https://www.google.com/maps/place/C%C3%B4ng+ty+Thi%E1%BA%BFt+k%E1%BA%BF+website+Saigon+Hitech/@10.787981,106.635604,15z/data=!4m5!3m4!1s0x0:0x77bc8fb1ae5a6d83!8m2!3d10.787981!4d106.635604) chuyên nghiệp để sớm nâng tầm thương hiệu của mình trên thị trường đầy thách thức như hiện nay.
### Bài viết liên quan
[Những ưu điểm của laptop HP Workstation 8770W](https://toilamseo.com/nhung-uu-diem-cua-laptop-hp-workstation-8770w-2-1343.html)(05/02) [Có nên lựa chọn đèn Led Philips downlingt 44080 3.5W?](https://toilamseo.com/co-nen-lua-chon-den-led-philips-downlingt-44080-3-5w-2-1234.html)(13/07) [Saigon Hitech thiết kế website An cung Ngưu Hoàng](https://toilamseo.com/saigon-hitech-thiet-ke-website-an-cung-nguu-hoang-2-1000.html)(30/09) [Website điện máy của Saigon Hitech](https://toilamseo.com/website-dien-may-cua-saigon-hitech-2-980.html)(23/09) [Ổn áp Lioa – giải pháp cho giờ cao điểm](https://toilamseo.com/on-ap-lioa-–-giai-phap-cho-gio-cao-diem-2-809.html)(25/04) [Sử dụng đèn pha led trong phòng ngủ](https://toilamseo.com/su-dung-den-pha-led-trong-phong-ngu-2-765.html)(16/04) [Trở thành nhà tiêu dùng thông thái khi chọn ổn áp Lioa](https://toilamseo.com/tro-thanh-nha-tieu-dung-thong-thai-khi-chon-on-ap-lioa-2-761.html)(14/04) [Nên mua đèn pha led Philips ở đâu?](https://toilamseo.com/nen-mua-den-pha-led-philips-o-dau-2-731.html)(06/04) [Có phải dùng ổn áp lioa thì tốn khá nhiều điện?](https://toilamseo.com/co-phai-dung-on-ap-lioa-thi-ton-kha-nhieu-dien-2-727.html)(06/04) [Tại sao cần dùng đến ổn áp lioa](https://toilamseo.com/tai-sao-can-dung-den-on-ap-lioa-2-718.html)(04/04)

WARC-Type: response

Content-Type: application/http; msgtype=response

WARC-Identified-Payload-Type: application/xhtml+xml

Connection: Upgrade, close

Pragma: no-cache

Cache-Control: no-store, no-cache, must-revalidate, post-check=0, pre-check=0

Vary: Accept-Encoding

Content-Type: text/html


markdown:
# Bắn tàn nhang bằng tia laser giá bao nhiêu?
Bài viết hay nhất >>
[Cách xóa tàn nhang và vết thâm an toàn](http://tritannhanghieuqua.com/cach-xoa-tan-nhang-va-vet-tham-an-toan)
Chào bạn Bảo Ngọc!
Cảm ơn bạn đã tin tưởng và gửi thắc mắc làm đẹp của mình tới hòm thư Thu Cúc. Với câu hỏi “Bắn tàn nhang bằng tia laser giá bao nhiêu?” Chúng tôi xin được giải đáp như sau:
Để điều trị tàn nhang nhiều chị em mặc dù đã thử rất nhiều cách, tìm mọi phương pháp thực hiện ở nhà nhưng vẫn không mấy hiệu quả. Lúc này bắn tia laser để xóa vết tàn nhang là phương pháp cứu cánh cho bạn. Vậy bắn tia laser chữa tàn nhang hết nhiều tiền không?
![ bắn tàn nhang bằng laser hết bao nhiêu](http://tritannhanghieuqua.com/wp-content/uploads/2016/07/tri-tan-nhang-5.jpg)
Trị tàn nhang bằng công nghệ laser Q – Switched là phương pháp an toàn, hiệu quả
Công nghệ Laser Q – Switched là công nghệ điều trị tàn nhang hiện đại, an toàn đã và đang được ứng dụng hiệu quả tại Thu Cúc Clinics. Đây được coi là giải pháp hữu hiệu giúp loại bỏ tàn nhang trên da một cách nhanh chóng với chi phí trị tàn nhang bằng laser cho 1 lần điều trị là 3.500.000 VNĐ.
Tìm hiểu thêm về dịch vụ:
[bao nhiêu tiền một lộ trình trị tàn nhang laser](http://tritannhanghieuqua.com/bao-nhieu-tien-mot-lo-trinh-tri-tan-nhang)
Theo đó, để xác định liệu trình trị liệu phù hợp các chuyên gia sẽ thăm khám hiện trạng thẩm mỹ và đưa ra tư vấn làm đẹp chính xác nhất.
![ Bắn tàn nhang bằng tia laser tại Thu Cúc](http://tritannhanghieuqua.com/wp-content/uploads/2016/07/tri-tan-nhang2.jpg)
Quy trình trị tàn nhang diễn ra nhanh chóng, cho kết quả thẩm mỹ lâu dài
Xem thêm >>
[trị tàn nhang và nám da](http://tritannhanghieuqua.com/cach-phan-biet-nam-tan-nhang-va-doi-moi)
Sau khi làm sạch vùng điều trị, các chuyên gia sẽ tiến hành điều chỉnh bước sóng phù hợp rồi chiếu trực tiếp laser đi sâu vào lớp da bên trong nhằm phá vỡ sự tích tụ sắc tố và lấy đi các sắc tố sậm màu trên da, trả lại làn da căng mịn, tự nhiên như mong muốn.
Quy trình trị tàn nhang tại Thu Cúc Clinics được thực hiện nhanh chóng trong điều kiện vô khuẩn. Bạn có thể về nhà ngay sau đó mà không mất thời gian nghỉ dưỡng tại bệnh viện, không lo ảnh hưởng đến vùng da lân cận. Đặc biệt, bắn tàn nhang bằng laser không chỉ mang lại hiệu quả cao mà còn giúp tăng cường tuần hoàn, kích thích sản sinh tế bào mới, giúp vùng da lão hóa được hồi sinh và duy trì kết quả lâu dài.
![ quy trình trị tàn nhang bằng laser](http://tritannhanghieuqua.com/wp-content/uploads/2016/07/tri-tan-nhang2-3.jpg)
Cơ sở vật chất, thiết bị máy móc hiện đại tại Thu Cúc Clinics
Với quy trình làm đẹp bài bản cộng với mức giá cạnh tranh, Thu Cúc Clinics là địa chỉ thẩm mỹ tin cậy dành cho bạn. Nếu còn băn khoăn gì về “
*bắn tia laser chữa tàn nhang hết nhiều tiền không*” bạn có thể gọi tới số 1900558896 để được tư vấn trực tiếp hoặc xem tại website *tritannhanghieuqua.com* để biết thêm các thông tin khác.
Tìm hiểu kỹ hơn về:
[trị tàn nhang hiệu quả](http://tritannhanghieuqua.com/tri-tan-nhang)
Bác sĩ tư vấn 24/7:
[1900.5588.96](tel:1900558896)

WARC-Identified-Payload-Type: text/html

Connection: close

Vary: Accept-Encoding,User-Agent

Content-Type: text/html; charset=UTF-8


markdown:
![](http://pic.downloadastro.com/extension/pfb_128_128.png)
# .PFB Phần mở rộng tập tin
## Tập tin pfb là gì và làm sao mở nó - Printer Font Binary File
**Mục lục tập tin:** [Tập tin Phông chữ](//vi.downloadastro.com/Tập tin Windows/Tập tin Phông chữ/) **Nhà phát triển:** [Adobe Systems](//vi.downloadastro.com/adobe_systems/)
PFB là tập tin được dịch sang mã nhị phân. Để tập tin này hoạt động cần có một tập tin .PFM kèm theo.
**Cả hai tập tin PFM và PFB đều phải được lưu ở thư mục font trong hệ thống thì mới hoạt động được.**
##
pfb tập tin -phần mềm có thể mở pfb tập tin
Chọn một phần mềm từ danh sách dưới đây để mở, biên tập, tạo, chuyển đổi, chạy hay xem pfb tập tin. Các phần mềm này được sắp xếp theo độ phổ biến và liên quan tới pfb tập tin
[CorelDRAW Graphics Suite 2018 20.1.0.708](//coreldraw_graphics_suite_1.vi.downloadastro.com)
Giải pháp thiết kế và đồ họa
##
Mở pfb tập tin - các chương trình khác
- Autodesk AutoCAD 2014
- FontForge

Server: cloudflare

Access-Control-Allow-Methods: PUT, GET, POST, DELETE, OPTIONS

Access-Control-Allow-Headers: origin, x-requested-with, content-type

Cache-Control: max-age=0, no-cache

Set-Cookie: __cfduid=d11e8986b080aacc6e0a4f6e1a7a04dce1529549504; expires=Fri, 21-Jun-19 02:51:44 GMT; path=/; domain=.downloadastro.com; HttpOnly; Secure


markdown:
# Các khách sạn ở Robertsons Beach
## Tìm khách sạn tại Robertsons Beach
## Bạn định khi nào đến Robertsons Beach?
![ Hotels.com™ Rewards](https://a.cdn-hotels.com/da/assets/s/63.0.8634/images/brands/hcom/hcomrewards/logos/rewards-logo-white-no-moon-vi_VN.png)
Tích lũy 10 đêm, nhận 1 đêm thưởng
Và nhớ tìm Giá Bí mật tại một số khách sạn
Tích lũy 10 đêm, nhận 1 đêm thưởng
Và nhớ tìm Giá Bí mật tại một số khách sạn

Server: Application-Server

X-Hcom-Rendered-With: release

X-Frame-Options: SAMEORIGIN

Content-Security-Policy: default-src 'unsafe-inline' data: about: hcom: blob: callback: chrome-error: *; script-src 'unsafe-eval' 'unsafe-inline' data: about: blob: asset: *; report-uri https://525fade5ca1b22e6c3c9f4504b6b3f68.report-uri.com/r/d/csp/enforce

Set-Cookie: bm_sz=0C569BF07DA1068787E30BEF73E75B12~QAAQG/OPqDLREBRkAQAA16I5IAHOBV+MN/5AEjm1Pf07RD0p/6V7fyqGaPumVgMSKNxlp/SqUbr7zAeK+/wAMKgwWsvDyQdddAHGfPP5wzw+zUJEQO/bBMAwB/Bbvi7BEdtBbtkC4WMUbwDwJHOKw+qVWQEKZF9nslW3kCCRp4FZtIf9SqkBfY2CZ6PMHlE=; Domain=.hotels.com; Path=/; Expires=Thu, 21 Jun 2018 06:43:25 GMT; Max-Age=14400; HttpOnly

X-Content-Type-Options: nosniff

X-UA-Compatible: IE=Edge

Cache-Control: max-age=0, no-cache, no-store

site_id_3: ssl-*.hotels.com - A

Content-Language: vi-VN


markdown:
-
- Tìm kiếm:
**đua xe**
- Tìm kiếm:
**đua xe** Các Kết Quả Tìm Kiếm *(14 chương trình)* **đua xe**Các Kết Quả Tìm Kiếm
#### Lọc theo:
### Các hạng mục
-
[Tất cả](https://vi.softonic.com/s/đua-xe)
-
[Game PC](https://vi.softonic.com/s/đua-xe:t:game-pc)(14)
-
[Game Đua xe](https://vi.softonic.com/s/đua-xe:t:game-Đua-xe)(11)
-
[Tổng hợp](https://vi.softonic.com/s/đua-xe:t:tổng-hợp)(7)
-
[Lướt qua](https://vi.softonic.com/s/đua-xe:t:lướt-qua)(3)
-
[Tiện ích Game](https://vi.softonic.com/s/đua-xe:t:tiện-ích-game)(1)
-
[Tổng hợp](https://vi.softonic.com/s/đua-xe:t:tổng-hợp)(1)
-
[Game nhiều người chơi](https://vi.softonic.com/s/đua-xe:t:game-nhiều-người-chơi)(1)
-
[Xe mô tô](https://vi.softonic.com/s/đua-xe:t:xe-mô-tô)(1)
-
[Các game Đóng Vai](https://vi.softonic.com/s/đua-xe:t:các-game-Đóng-vai)(1)
-
**Need For Speed Underground**
- Mô tả
-
... này là có ba kiểu
**đua**mới - Drag... (rê đít xe) và Sprint ( **đua**nước rút... để khiến cuộc **đua**xe đường phố...
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 2.708.377
-
-
**Need For Speed World**
- Mô tả
-
Đua xe và các cuộc đua với nhiều người chơi
- Giấy phép
- Miễn phí
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 5.890.490
-
-
**Superbike Racers**
- Mô tả
-
Đốt cháy đường đua trong trò chơi đua xe miễn phí này
- Giấy phép
- Miễn phí
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 1.865.489
-
-
**City Racing**
- Mô tả
-
Một trò chơi đua xe tự do miễn phí
- Giấy phép
- Miễn phí
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 2.423.678
-
-
**Audiosurf**
- Mô tả
-
... một game
**đua**tốc độ/âm ... chặng **đua**và câu đố. Âm ... nhiều đường **đua**. Trong... nhạc, câu đố và **đua**tốc tộ....
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 504.617
-
-
**Need for Speed Underground 2**
- Mô tả
-
Tinh chỉnh xe của bạn khiến đối thủ hít khói
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 2.440.988
-
-
**Need for Speed: Most Wanted**
- Mô tả
-
Trở nên nổi tiếng nhất
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 4.471.897
-
-
**Need for Speed Carbon**
- Mô tả
-
Sẵn sàng cho Need For Speed với khả năng tùy biến cao nhất
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 1.875.707
-
-
**World of Warcraft**
- Mô tả
-
... thích chạy
**đua**luyện cấp...
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 1.617.152
-
-
**Bus Driver**
- Mô tả
-
Nếu bạn từng mơ ước được tự mình điều khiển một chiếc xe buýt, hãy thử qua Bus Driver. Bus Driver ...
- Giấy phép
- Phiên bản dùng thử
- Ngôn Ngữ
- Tiếng Anh
-
- Tổng số lượt tải
- 748.168
-

X-Cache: HIT

Server: nginx

Via: 1.1 varnish

Accept-Ranges: bytes

Cache-Control: public, must-revalidate, max-age=300, s-maxage=44086

Content-Security-Policy: upgrade-insecure-requests

Content-Type: text/html; charset=utf-8