Whats the number of times the letter . appears in:
Measuring AC voltage harmonic form of 10mkV to 1000V.
The frequency range from 20Hz to 100kHz.
Basic measurement uncertainty ± 0,035% (to 1B), ± 0,05% (10, 100, 1000B), ± 0,06% (up to 100 mV), ± 0,2% (to 10 mV).
Bx. Resistance. (1 ± 0,1) MW.
Bx. 100 pF capacitance.
493h98h480mm dimensions, weight 10kg.
Voltmeters V3-60 for measuring AC voltage are in stock. The devices are ready for shipment.
AC voltmeter digital V3-60 is designed for measurement of AC voltage harmonic form by srednevypryamlennogo transformation in the laboratory and shop floor. The reading device and output DC voltage of the device is equal to or proportional to the rms value of the measured voltage.
AC voltmeter digital V3-60 have a digital display of the measurement result and the analog output.
AC voltmeter B3-60 interface unit to the B3-60 (hereinafter - the device with a block interface) provides a connection to the shared channel (CPC).
mains voltage (220 ± 22) V, (50 ± 0,5) Hz and harmonic content up to 5%.
Measuring range of the instrument voltages up to 1000V 10mkV overlapping ranges with upper limits of 10mV, 100mV, 1V, 10V, 100V, 1000V.
The normal frequency range of the measured device alternating voltages from 400 Hz to 10 kHz, 10 mV, except for sub-band, for which the normal range of frequencies from 400 Hz to 2 MHz. Operating frequency range from 20 Hz to 100 kHz.
Prince - nominal static characteristic conversion unit, equal to 100, 10, 1, 0.1, 0.01, 0.001 for sub-bands on the upper limits of 10, 100 mV, 1, 10, 100, 1000B, respectively; δ-unit conversion error in% .
The error of the static characteristic of the transformation (the output voltage error) parting voltmeter B3-60 AC does not exceed the maximum permissible error of the instrument.
Uk - the final value assigned to the sub-band measurements; Un - the instrument reading in B.
Error of the instrument AC voltmeter digital V3-60 in the working frequency range does not exceed the values ​​calculated by the formulas given in Table 1.
Additional instrument error caused by changes in ambient temperature from normal to any temperature within the operating temperature range and in a normal working Fly over the frequency does not exceed the maximum permissible error of the instrument for every 10 ° C change in temperature.
Un - the reading of B.
Error of the instrument AC voltmeter digital V3-60 high humidity within the operating conditions of use does not exceed the limit value of 2.5 permissible basic error.
Short-term volatility readings for 10 minutes is equal to no more than 0.1 of the maximum permissible error of the respective sub-band measurement, provided that during this period of normal conditions.
Ripple of the output voltage of a device is less than 3 mV.
Settling time readings from the filing of the input voltage and the transition of one sub-band to another is less than 10 seconds.
The input impedance of the instrument is (1 ± 0,1) MW.
Input capacitance is 100 pF. Capacity of the supplied cable does not exceed 80 pF.
The output impedance of the analog output device is (1 ± 0,1) k.
AC voltmeter digital V3-60 retains its specifications within the standards set specifications, after exposure for 1 minute test voltage of 10 V at sub-bands of 10, 100 mV, 1 V and 1100 V subband-10V, 100V, 1000V.
AC voltmeter digital V3-60 retains its specifications within the standards set specifications, when powered from the mains AC voltage (220 ± 22), frequency (50 ± 0,5) Hz.
The power consumed by the device from the network at the rated voltage does not exceed 34 VA.
Power consumed by the interface unit at the rated voltage does not exceed 16 VA.
The device allows for continuous operation in the operating conditions for 8 hours per day, while maintaining its specifications within the standards set specifications.
Provided the normal patterns of semiconductor devices, electric components within the norms, standards and specifications for them.
Note. Continuous operation time does not include the operating mode setting time unit.
AC voltmeter digital V3-60 provides its specifications within the standards set by TU, under the influence of the amplitude of the ac magnetic field frequency equal to the power supply, voltage up to 8 A / M on the sub-bands of 10 and 100 mV and strength up to 80 A / m at remaining subbands.
AC voltmeter digital V3-60 enables remote control (RC) and access to the digital printing unit (CPU) in a parallel BCD.
Now the device interface unit provides the interface functions listed in Table 2.
weight of the device and the interface unit with storage compartment does not exceed 50 kg.
over-sized shipping container size does not exceed 502x1084x788 mm.
V3-60 voltmeter image.
V3-60 front view.
V3-60 side view.
V3-60 rear view.
V3-60 bottom view.
user
I found 56 "."s in the provided passage.
assistant