generally of a clear plastic material and which contain the infant on the upper surface 24 generally the guards 26 are removable and or releasable for complete access to the infant frame 10 includes upper and lower cross members 28 and 30 respectively joining a pair of vertical struts 32 and which vertical struts 32 may provide a means of support for other structural parts such as a shelf 34 mounted on the upper cross member 28 may be a control module 35 for containing the various electrical controls to operate the care center in addition a heater 36 is mounted to the upper cross member 28 as will be noted the location of the heater 36 is such to be above the infant bed 22 the heater is focused so as to provide a footprint on and around the infant to optimize the amount of heat directed upon the infant various types of focusable heaters are available for such application examples of which may be a calrod focused heater of about 500 600 watts or a corrugated foil heater preferably the latter is of a linear length such that the footprint of heat at the infant bed 22 is generally rectangular typically the heater 36 is about 18 to 24 inches in length extending outwardly cantilever fashion from the cross member 28 and will contain therein the calrod resistance heater that is enclosed within a glass tube also within the heater 36 is a parabolic metal reflector that redirects the infrared radiation emanating in all directions from the calrod resistance heater downwardly towards the infant bed 22 the parabolic reflector and calrod heater.