herein means any change in position or orientation of the fluid source whether by sliding pivoting rotating or bending the mount which causes a significant redirection of the fluid spray for example this movement may be by the practitioner bending the tube 42 of the mount 40 of the embodiment of fig1 5 rotating or pivoting the ball member 142 of the embodiment of fig6 or sliding the sliding member 242 of the embodiment of fig7 10 if a refillable fluid source such as a syringe is used the practitioner can refill or replace it with a full fluid reservoir this can be accomplished by the practitioner decoupling the needle support portion of the fluid source typically by unscrewing it from the fluid reservoir refilling or replacing it with a full reservoir and recoupling it to the retained needle support portion as the fluid ejecting portion of the fluid source is preferably retained by the cup member of the splashguard the practitioner can avoid re sticking the splashguard when refilling or replenishing the fluid source this feature can reduce the likelihood of needle stick injury to the practitioner after irrigation is complete the wound splashguard is removed from the wound and the fluid source is disengaged from the splashguard preferably in the emergency room setting both the fluid source and splashguard are made of relatively inexpensive plastic resins and both may be disposed of by the practitioner however in the surgical suite setting it may be desirable to have a reusable fluid source and splashguard that are made of.