Respiratory signals were determined from the projection images of free-breathing mice scanned on the X-RAD system using four methods: the so-called Amsterdam shroud method, a method based on the phase of the Fourier transform, a pixel intensity method, and a center of mass method.  The modified Amsterdam shroud, Fourier transform, and intensity respiratory signal methods are sufficiently accurate to be used for 4D imaging on the X-RAD system and show improvement over the existing center of mass method. 
X-RAD
Respiratory signals were determined from the projection images of free-breathing mice scanned on the X-RAD system using four methods: the so-called Amsterdam shroud method, a method based on the phase of the Fourier transform, a pixel intensity method, and a center of mass method.  The average error (and standard deviation) in phase bin assignment was 0.24 ± 0.08 (0.49 ± 0.11) phase bins for the Fourier transform method, 0.09 ± 0.03 (0.31 ± 0.08) phase bins for the modified Amsterdam shroud method, 0.09 ± 0.02 (0.33 ± 0.07) phase bins for the intensity method, and 0.37 ± 0.10 (0.57 ± 0.08) phase bins for the center of mass method.  Little dependence on ROI size was noted for the modified Amsterdam shroud and intensity methods while the Fourier transform and center of mass methods showed a noticeable dependence on the ROI size.


 The modified Amsterdam shroud, Fourier transform, and intensity respiratory signal methods are sufficiently accurate to be used for 4D imaging on the X-RAD system and show improvement over the existing center of mass method. 
Fourier transform
Respiratory signals were determined from the projection images of free-breathing mice scanned on the X-RAD system using four methods: the so-called Amsterdam shroud method, a method based on the phase of the Fourier transform, a pixel intensity method, and a center of mass method.  The Amsterdam shroud method was modified so the sharp inspiration peaks associated with anesthetized mouse breathing could be detected.  The average error (and standard deviation) in phase bin assignment was 0.24 ± 0.08 (0.49 ± 0.11) phase bins for the Fourier transform method, 0.09 ± 0.03 (0.31 ± 0.08) phase bins for the modified Amsterdam shroud method, 0.09 ± 0.02 (0.33 ± 0.07) phase bins for the intensity method, and 0.37 ± 0.10 (0.57 ± 0.08) phase bins for the center of mass method.  The intensity and modified Amsterdam shroud methods are recommended due to their high accuracy and low dependence on ROI size.
Amsterdam shroud method
Respiratory signals were determined from the projection images of free-breathing mice scanned on the X-RAD system using four methods: the so-called Amsterdam shroud method, a method based on the phase of the Fourier transform, a pixel intensity method, and a center of mass method.  The average error (and standard deviation) in phase bin assignment was 0.24 ± 0.08 (0.49 ± 0.11) phase bins for the Fourier transform method, 0.09 ± 0.03 (0.31 ± 0.08) phase bins for the modified Amsterdam shroud method, 0.09 ± 0.02 (0.33 ± 0.07) phase bins for the intensity method, and 0.37 ± 0.10 (0.57 ± 0.08) phase bins for the center of mass method.  Little dependence on ROI size was noted for the modified Amsterdam shroud and intensity methods while the Fourier transform and center of mass methods showed a noticeable dependence on the ROI size.



Intensity method
Respiratory signals were determined from the projection images of free-breathing mice scanned on the X-RAD system using four methods: the so-called Amsterdam shroud method, a method based on the phase of the Fourier transform, a pixel intensity method, and a center of mass method.  The average error (and standard deviation) in phase bin assignment was 0.24 ± 0.08 (0.49 ± 0.11) phase bins for the Fourier transform method, 0.09 ± 0.03 (0.31 ± 0.08) phase bins for the modified Amsterdam shroud method, 0.09 ± 0.02 (0.33 ± 0.07) phase bins for the intensity method, and 0.37 ± 0.10 (0.57 ± 0.08) phase bins for the center of mass method.  Little dependence on ROI size was noted for the modified Amsterdam shroud and intensity methods while the Fourier transform and center of mass methods showed a noticeable dependence on the ROI size.


 The modified Amsterdam shroud, Fourier transform, and intensity respiratory signal methods are sufficiently accurate to be used for 4D imaging on the X-RAD system and show improvement over the existing center of mass method. 
Center of mass method