All subjects underwent a detailed ophthalmologic examination. Axial length was measured using IOL Master (Carl Zeiss Meditec, Dublin, CA, USA) optic biometry. Visual field was assessed with Humphrey II Perimetry (Carl Zeiss Meditec, Dublin, CA, USA) SITA standard central 24-2 test. Tests with fixation loss less than 20%, false negative, and false positive response rates below 33% were considered reliable. Patients with at least two reliable visual field tests were included in the study. MD values were obtained.